MALDI-TOF mass spectrometry (MS) is a reliable method for bacteria identification. Some databases used for this purpose lack reference profiles for Brucella species, which is still an important pathogen in wide areas around the world. We report the creation of profiles for MALDI-TOF Biotyper 2.0 database (Bruker Daltonics, Germany) and their usefulness for identifying brucellae from culture plates and blood cultures.
We created MALDI Biotyper 2.0 profiles for type strains belonging to B. melitensis biotypes 1, 2 and 3; B. abortus biotypes 1, 2, 5 and 9; B. suis, B. canis, B ceti and B. pinnipedialis. Then, 131 clinical isolates grown on plate cultures were used in triplicate to check identification. Identification at genus level was always correct, although in most cases the three replicates reported different identification at species level. Simulated blood cultures were performed with type strains belonging to the main human pathogenic species (B. melitensis, B. abortus, B. suis and B. canis), and studied by MALDI-TOF MS in triplicate. Identification at genus level was always correct.
Citation: Ferreira L, Vega Castaño S, Sánchez-Juanes F, González-Cabrero S, Menegotto F, Orduña-Domingo A, et al. (2010) Identification of Brucella by MALDI-TOF Mass Spectrometry. Fast and Reliable Identification from Agar Plates and Blood Cultures. PLoS ONE 5(12): e14235. https://doi.org/10.1371/journal.pone.0014235
Editor: Edgardo Moreno, Universidad Nacional, Costa Rica
Received: June 23, 2010; Accepted: November 15, 2010; Published: December 6, 2010
Copyright: © 2010 Ferreira et al. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Funding: This work was partly supported by a grant from the Consejería de Sanidad, Junta de Castilla y León (Spain) and by the Instituto de Salud Carlos III (Ministerio de Sanidad, Spain) (Ayuda de Infraestructura). The funders had no role in study design, data collection and analysis, decision to publish, or preparation of the manuscript.
Competing interests: The authors have declared that no competing interests exist.
Brucellosis is a zoonosis that remains an important public health problem in wide areas, such as the Mediterranean basin, the north of Africa, Mexico, and Central and South America –. Six species have been described, based on host preferences, metabolism, culture and antigenic features, including the two most recent species (Brucella pinnipedialis and B. ceti), isolated from seals and cetaceans (dolphins and whales), but most human cases remain to be caused by B. melitensis and B. abortus . However, DNA-DNA hybridization shows a high homology between strains, indicating that current species should be rather considered as subspecies corresponding to evolutionary lineages adapted to specific hosts .
Classically, biphasic blood cultures such as the Ruiz-Castañeda method were used to isolate brucellae from blood and bone marrow. Now, most laboratories use continuous-monitoring automated blood culture systems, which can shorten the time to isolation and have been shown to be highly sensitive . Nevertheless, subculture is necessary to identify the microorganism, and brucellae may require 2–3 days to grow on chocolate or blood agar. Rapid automated bacterial identification systems must be interpreted with caution, because brucellae have been misidentified with some of these systems . PCR have shown high sensitivity and specificity, but its use remains infrequent, mainly due to standardization problems .
MALDI-TOF mass spectrometry (MS) has been suggested as a fast and reliable method for bacterial identification , , based on protein profiles characteristic of each microorganism. Databases have been developed that include the main pathogenic microorganisms, thus allowing the use of this method in routine bacterial identification from plate culture. Nevertheless, Brucella has not been still incorporated to some of the main databases available, because of problems derived from their potential bioterrorist use. This is an important problem for the routine use of MALDI-TOF MS for the direct diagnosis of blood cultures in countries where brucellosis is still frequent.
The aim of our study was to identify and differentiate Brucella species by MALDI-TOF MS, combining MALDI-TOF MS with dedicated bioinformatics and statistical methods (database search and pattern-matching algorithm). Initial spectra from three type strains of B. melitensis, five type strains of B. abortus and one type strain of B. suis, B. canis, B. ceti and B. pinnipedialis were used to set up database entries for re-identification of Brucella strains. This database was evaluated with 131 blind-coded Brucella clinical isolates previously identified by conventional methods. We also tested the reliability of this method for identifying brucellae directly from blood cultures, as soon as artificially inoculated blood cultures were reported as positive by a continuous-monitoring automated blood culture system.
Materials and Methods
Sheep blood was used for some experiments, i.e. simulated blood cultures. Since Sheep blood is obtained, as a conventional laboratory product, from commercial sources (Pronadisa Conda, Madrid, Spain), we did not consider any ethics approval to be necessary for this study.
The strains used for generating reference spectra were the following: B. melitensis biotype 1, strain 16M (ATCC 23456); B. melitensis biotype 2, strain 63/9 (ATCC 23457); B. melitensis biotype 3, strain ETHER (ATCC 23458); B. abortus biotype 1, strain 544 (ATCC 23448); B. abortus biotype 1, strain 45/20 (NCTC 11361); B. abortus biotype 2, strain 86/8/59 (ATCC 23449); B. abortus biotype 5, strain B3196 (ATCC 23452); B. abortus biotype 9, strain C68 (ATCC 23455), B. suis (NCTC 10098), B. canis (NCTC 10854), B. ceti (NCTC 12891) and B. pinnipedialis (NCTC 12890) Microorganisms were plated onto chocolate agar plates (bioMérieux, France), and incubated at 37°C for 48 hours. Colonies were used for creating Biotyper 2.0 database profiles.
The same isolates were also spread onto blood agar plates (bioMérieux, France), under the same conditions, to check the score reported by MALDI-TOF for colonies obtained from different culture media.
One hundred and thirty one human clinical isolates were used as blind coded isolates to check the reliability of the Biotyper 2.0 database, once spectra for B. melitensis, B. abortus, B. canis, B. ceti, B. pinnipedialis and B. suis had been created. The clinical isolates were plated onto chocolate agar plates (bioMérieux, France), and incubated at 37°C for 48 hours. Then, colonies were identified by conventional microbiology methods and PCR, according previously described methods , and by MALDI-TOF MS.
Colonies samples preparation for MALDI-TOF MS
Cells of a whole colony were transferred from the plate to a 1.5 mL tube (Eppendorf, Germany) with a pipette tip and mixed thoroughly in 300 µL of water to resuspend the bacterial cells. Then, 900 µL of absolute ethanol was added and the mixture was centrifuged at 15,500 g for 2 min and the supernatant was discarded. The pellet was air-dried at room temperature for 1 hour. Subsequently, 50 µL of formic acid (70% v/v) were added to the pellet and mixed thoroughly by pipetting before the addition of 50 µL of acetonitrile to the mixture. The mixture was centrifuged again at 15,500 g for 2 min. One microliter of the supernatant was placed onto a spot of the steel target and air-dried at room temperature. Each sample was overlaid with 1 µL of matrix solution (saturated solution of HCCA (alpha-cyano-4-hydroxy cinnamic acid) in organic solvent (50% acetonitrile and 2.5% trifluoroacetic acid) and air-dried.
Simulated blood cultures.
The same 3 B. melitensis, 5 B. abortus, 1 B. canis and 1 B. suis type strains used for establishing reference spectra, were used to perform simulated blood cultures. Three to four colonies were suspended into sterile water to reach a concentration around 104 CFU/mL. Then, 4 mL of brucellae suspension were added to 4 mL of blood, and these 8 mL were inoculated into BACTEC Plus + Aerobic/F bottles (Becton Dickinson, NJ, USA), and incubated at 37°C in a BACTEC 9240 device (Becton Dickinson, NJ, USA), until they were reported as positive.
Blood cultures samples preparation for MALDI-TOF MS
Four mL of the positive blood culture were centrifuged at 2,000 g for 30 seconds to remove leucocytes. Supernatant was centrifuged at 15,500 g for 5 minutes to collect bacteria. The pellet was washed once with de-ionized water. Then, the ethanol/formic acid extraction procedure described above was applied.
Measurements were performed on an Autoflex III MALDI-TOF/TOF mass spectrometer (Bruker Daltonics, Leipzig, Germany) equipped with a 200-Hz smartbeam laser. Spectra were recorded in the linear positive mode at a laser frequency of 200 Hz within a mass range from 2,000 to 20,000 Da. The IS1 voltage was 20 kV, the IS2 voltage was maintained at 18.6 kV, the lens voltage was 6 kV, and the extraction delay time was 40 ns.
For each spectrum, 500 laser shots were collected and analyzed (10×50 laser shots from different positions of the target spot). The spectra were calibrated externally using the standard calibrant mixture (Escherichia coli extracts including the additional proteins RNase A and myoglobin, Bruker Daltonics). Calibration masses were as follows: RL36, 4364.3 Da; RS22, 5095.8 Da; RL34, 5380.4 Da; RL33meth, 6254.4 Da, RL32, 6315 Da; RL29, 7273.5 Da; RS19, 10299.1 Da; RNase A, 13682.2 Da; myoglobin, 16952.5 Da.
Spectrum generation and data analysis
For automated data analysis, raw spectra were processed using the MALDI Biotyper 2.0 software (Bruker Daltonics, Leipzig, Germany) at default settings. The software performs normalization, smoothing, baseline subtraction, and peak picking, creating a list of the most significant peaks of the spectrum (m/z values with a given intensity, with the threshold set to a minimum of 1% of the highest peak and a maximum of 100 peaks). To identify unknown bacteria, each peak list generated was matched directly against reference libraries (3,476 species) using the integrated patterns matching algorithm of the Biotyper 2.0 software (Bruker Daltonics, GmbH, Germany). The unknown spectra were compared with a library of reference spectra based on a pattern recognition algorithm using peak position, peak intensity distributions and peak frequencies. Once a spectrum has been generated and captured by the software, the whole identification process is performed automatically, without any user intervention. MALDI-TOF identifications were classified using modified score values proposed by the manufacturer: a score ≥2 indicated species identification; a score between 1.7 and 1.9 indicated genus identification, and a score <1.7 indicated no identification.
For reference library construction, 36 independent spectra were recorded for each bacterial isolate (three independent measurements at twelve different spots each).
Manual and visual estimation of the mass spectra was performed using Flex Analysis 3.0 (Bruker Daltonics GmbH, Germany) performing smoothing and baseline substraction. Checking existence of flatlines, outliers or single spectra with remarkable peaks differing from the other spectra was done, taking into account that mass deviation within the spectra set should be less than 500 ppm. Finally, 20 spectra were selected, removing questionable spectra from the collection. To create peak lists of the spectra, the BioTyper software was used as described above. The 70 independent peaks of a strain were used for automated “main spectrum” generation with default settings of the BioTyper software. Thereby, for each library entry a reference peak list (main spectrum) which contains information about averaged masses, averaged intensities, and relative abundances in the 20 measurements for all characteristic peaks of a given strain was created, so a main spectrum displayed the most reproducible peaks typical for a certain bacterial strain.
Cluster analysis was performed based on comparison of strain-specific main spectra created as described above. The dendrogram was constructed by the statistical toolbox of Matlab 7.1 (MathWorks Inc., USA) integrated in the MALDI Biotyper 2.0 software. The parameter settings were: ‘Distance Measure = Correlation’ and ‘Linkage = average’. The linkage function is normalized according to the distance between 0 (perfect match) and 1000 (no match).
MALDI Biotyper 2.0 database (Bruker Daltonics) used for routine identification of microorganism in clinical microbiology contains 3,476 entries, involving bacteria and fungi groups. Although it contains a wide variety of clinically relevant microorganisms, some important pathogenic microorganisms have not been yet included (e.g. Brucella). This is an important limitation in some countries, where brucellosis is still a relevant infectious problem and it is not infrequent to isolate Brucella species from febrile patients. For this reason, 3 B. melitensis, 5 B. abortus, 1 B. suis, 1 B. canis, 1 B. ceti and 1 B. pinnipedialis type strains were used to generate reference spectra and to extend the MALDI Biotyper database. The visual inspection of these spectra from whole-cell extracts revealed a high similarity among them (Figure 1), with specific peaks commons for all the strains at 2426 Da, 2585 Da, 3757 Da, 4851 Da, 5168 Da, 5870 Da, 6282 Da, 6672 Da, 7042 Da, 7393 Da, and 7511 Da (Table 1). Although some differences can be observed among all the strains, only some strains have discriminating peaks at species level. Thus, peaks at 4244, 4286 and 6854 Da appear only in B. melitensis, but none of these peaks are constant in all the B. melitensis strains. The same happens with peaks at 3899, 6166 and 7358 Da in B. abortus. According these data, there is a peak profile characteristic of Brucella at genus level, but there isn't at species level.
The relative intensity of the ions (a.u., arbitrary units) are shown on the y axis, and the masses (in Da) of the ions are shown on the x axis. The m/z value stands for mass to charge ratio. For a single positive charge, this value corresponds to the molecular weight of the protein.
A cluster analysis with the 12 type strains was performed, using the integrated tools of the MALDI Biotyper 2.0 software package. The resulting dendrogram (Figure 2) shows a close proximity between B. ceti and B. pinnipedialis, and between B. suis and B. canis, and more proximity between these four species than between any of them and B. melitensis and B. abortus. Otherwise, B. melitensis and B. abortus biotypes are extremely close and interrelated, excepting B. abortus biotype 1 strain 544 (ATCC 23448).
Distance is displayed in relative units.
In addition, we tested the influence of different culture conditions. Therefore the different strains were cultivated on blood agar and chocolate agar and comparison between identification and scores obtained did not reveal any significant difference between both culture media in any aspect (data not shown).
Once established the spectra for the Brucella reference strains, 131 blind-coded Brucella strains were matched against the MALDI Biotyper 2.0 database in triplicate, to prove its suitability for routine identification and discrimination of Brucella at the genus and species levels (Table 2). The MALDI-TOF device reported the same identification at species level for the three replicates in 14 out of 17 B. abortus isolates (82.4%), but in only 12 out of 112 B. melitensis isolates (10.7%) and in 1 out of 2 B. suis isolates (50%). Therefore only 20.6% isolates reported the same identification at species level for the three replicates. Nevertheless, in all of them this identification was coincident with conventional and genetic identifications at genus level. In all the clinical isolates tested, the score difference between the ten most probably matched species reported by the MALDI Biotyper software for each isolate was very low and higher than 2 in several of them. As an example, we show in Table 3 the score difference between the ten most probably matched species reported for one B. melitensis clinical isolate. Taking into account that a reliable identification at species level is accepted with scores higher than 2, we can't discriminate between different Brucella species, since different Brucella species give scores higher than 2 in most clinical isolates.
To test the reliability of MALDI-TOF MS for identifying brucellae directly from blood culture, we inoculated blood culture bottles with type strains belonging to the four main human pathogenic species (B. melitensis, B. abortus, B. suis and B. canis). All the blood cultures inoculated were reported as positive by 48 hours of incubation. Three replicates of each positive blood culture were studied using MALDI-TOF MS. All the blood cultures were reported by the MALDI-TOF as Brucella. Nevertheless, in several cases the identification did not match with the organism inoculated in the blood culture at the species level, or different identifications were obtained at the species level among the three replicates of the same blood culture (Table 4).
Brucellosis remains a serious problem in wide areas around the world. Symptoms are nonspecific (fever, malaise, back pain, profuse night sweating). This lack of specificity of symptoms may delay the diagnosis for weeks. Though mortality rate is currently low, it remains a severe disease, and complications such as epididymo-orchitis, arthritis (especially sacroiliitis), and CNS complications are not infrequent. The current taxonomy of this genus is confusing because traditional, phenotype-based classification, and genetic-based classification are used simultaneously . Genetic taxonomic studies have suggested that the 8 currently accepted species, including the 2 marine mammal-associated species, represent a single species , . Nevertheless, recent studies show that traditional species, and the marine mammals-associated species, can be differentiated by outer membrane genes polymorphism, insertion sequences and whole-chromosome preparations , .
Human brucellosis has been described associated to B. melitensis, B. abortus, B. suis and B. canis, among the traditional species. Human cases associated to the newer marine mammals-associated species have also been reported , . Though B. abortus has the broadest geographical distribution, most human cases are now caused by B. melitensis, which usually produces the most severe disease.
Microbiological procedures for Brucella detection and identification have been controversial for years. Blood, and eventually bone marrow, are usually the specimens for diagnosis of human brucellosis. Conventional blood cultures did not yield satisfactory results, and this led to the development of the classic biphasic blood cultures technique proposed by Ruiz-Castañeda. Now, both lyses-centrifugation methods and continuous monitoring blood cultures devices are able to detect Brucella with a good sensitivity .
Early suspicion that an isolate might be a Brucella is important, both for clinical and epidemiological reasons, because of the hazard of laboratory-acquired brucellosis, and for the early detection of a hypothetical bioterrorist attack. Early suspicion may be currently established based on Gram staining, but final identification requires colonies growth on agar plates and performing biochemical or serological tests on these colonies.
For all those reasons, the availability of methods that allow a rapid and reliable Brucella identification both from agar plates and directly from the blood culture bottles, once this is reported as positive by the continuous-monitoring blood culture systems, would be extremely useful. Direct detection methods based on PCR have been described , but these methods are expensive, can be affected by PCR inhibitors present in blood, and do not give any information about the viability of the microorganisms. Thus, they have not reached a wide diffusion by now.
MALDI-TOF MS is an important and increasingly available tool in clinical microbiology laboratories, because it allows a rapid and accurate identification of bacteria , . We have seen that MALDI-TOF MS allows a fast and highly reliable identification of Brucella, at genus level, from colonies growth. Nevertheless, identification at species level is less reliable. Data in Figure 1 and Table 1 show a common profile for all Brucella type strains tested, but no specific peaks are found in any species. This explains the reliability at genus level and the low reliability at species level. Otherwise, the high similarity between all type strains is not surprising, since genetic taxonomic studies suggest that the 6 currently accepted species represent a single species , . Once the protein profiles for B. melitensis, B. abortus, B. suis, B. canis, B. ceti and B. pinnipedialis had been created and included in the database, all the blind-coded Brucella isolates tested were correctly identified, with scores >2. Scores differences between the 10 most probably matched species given by the MALDI Biotyper software for each isolate were extremely low, and replicates of the same protein extract were sometimes identified as different species. In almost all isolates, more than one species matched with score value >2 in the 10 most probably matched species, thus confirming the low discriminating power at species level. The agar on which the microorganism grows does not seem to affect to identification, since scores obtained from blood agar and chocolate agar were similar for all the isolates tested. Previous studies on other microorganisms agree with these results, showing that MALDI-TOF MS protein fingerprints are not significantly influenced by variability in growth conditions –.
MALDI-TOF MS has been previously shown to be able to identify microorganisms directly from blood cultures reported as positive by automatic blood culture processing devices –. In our study, positive blood cultures spiked with brucellae were reported always as Brucella, but species failures were frequent, as had been shown previously in agar plates. Brucella was always identified with scores >2 but, as happens in agar cultures, score differences between the ten most probably matched species reported by the MALDI Biotyper software for each isolate were very low, and species identification had scores >2 in the best and second match, although the species in both cases are different, indicating that the species identification cannot be fully reliable.
Protein profiles found for type strains, show that protein profile similarity does not correlate always with the traditional genus/species/biotype classification. Proteins profile for some species and biotypes may be closer to other species than to other biotypes belonging to the same species, as can be observed in cluster analysis (Figure 2).
In summary, the protein profiles for type strains of B. melitensis biotypes 1, 2 and 3; B. abortus biotypes 1, 2, 5 and 9; B. suis; B. canis; B. ceti and B. pinnipedialis were generated and included in the MALDI Biotyper database. The study of clinical isolates both from agar plate cultures and directly from blood cultures showed that these profiles allow a reliable identification of brucellae to genus level, stating that MALDI-TOF MS is an fast and highly reliable technique for straightforward Brucella identification, both from culture plates and directly from blood culture vials.
Conceived and designed the experiments: LF JMGB JLMB. Performed the experiments: LF SVC FSJ SGC FM. Analyzed the data: LF FSJ JMGB JLMB. Contributed reagents/materials/analysis tools: SGC FM AOD JMGB JLMB. Wrote the paper: LF JMGB JLMB.
- 1. za J, Gudiol F, Pallares R, Viladrich PF, Rufi G, et al. (1992) Treatment of human brucellosis with doxycycline plus rifampicin or doxycycline plus streptomycin. A randomized, double-blind study. Ann Intern Med 17: 25–30.
- 2. Solera J, Rodríguez-Zapata M, Geijo P, Largo J, Paulino J, et al. (1995) Doxycycline–rifampin versus doxycycline–streptomycin in treatment of human brucellosis due to Brucella melitensis. The GECMEI Group. Grupo de Estudio de Castilla-la Mancha de Enfermedades Infecciosas. Antimicrob Agents Chemother 39: 2061–2067.
- 3. Colmenero JD, Reguera JM, Martos F, Sánchez de Mora D, Delgado M, et al. (1996) Complications associated with Brucella melitensis infections: a study of 530 cases. Medicine (Baltimore) 75: 195–211.
- 4. Ariza J, Corredoira J, Pallares R, Villadrich PF, Rufi G, et al. (1995) Characteristics of and risk factors for relapse of brucellosis in humans. Clin Infect Dis 20: 1241–1249.
- 5. Lindquist D, Chu MD, Probert WS (2007) Francisella and Brucella. In: Murray PR, Baron EJ, Jorgensen JH, Landry ML, Pfaller MA, editors. Manual of Clinical Microbiology (9th ed.). Washington, DC: American Society for Microbiology. pp. 815–834.
- 6. Verger JM, Grimont F, Grimont PA, Grayon M (1987) Taxonomy of the genus Brucella. Ann Inst Pasteur Microbiol 138: 235–238.
- 7. Yagupsky P, Peled N, Press J, Abramson O, Abu-Rashid M (2003) Comparison of BACTEC 9240 Peds Plus medium and isolator 1.5 microbial tube for detection of Brucella melitensis from blood cultures. J Clin Microbiol 35: 1382–1384.
- 8. Elsaghir AAF, James EA (2003) Misidentification of Brucella melitensis as Ochrobactrum anthropi by API 20NE. J Med Microbiol 52: 441–442.
- 9. Queipo-Ortuño MI, Colmenero JD, Baeza G, Morata P (2005) Comparison between LightCycler Real-Time Polymerase Chain Reaction (PCR) assay with serum and PCR-enzyme-linked immunosorbent assay with whole blood samples for the diagnosis of human brucellosis. Clin Infect Dis 40: 260–264.
- 10. Seng P, Drancourt M, Gouriet F, La Scola B, Fournier PE, et al. (2009) Ongoing revolution in bacteriology: routine identification of bacteria by matrix-assisted laser desorption ionization time-of-flight mass spectrometry. Clin Infect Dis 49: 543–551.
- 11. Ferreira L, Vega S, Sánchez-Juanes F, González M, Herrero A, et al. (2010) Identifying bacteria using a matrix-assisted laser desorption ionization time-of-flight (MALDI-TOF) mass spectrometer. Comparison with routine methods used in clinical microbiology laboratories. Enferm Infecc Microbiol Clin 28: 492–497.
- 12. Garcia-Yoldi D, Marin CM, de Miguel MJ, Muñoz PM, Vizmanos JL, et al. (2006) Multiplex PCR assay for the identification and differentiation of all Brucella species and the vaccine strains Brucella abortus S19 and RB51 and Brucella melitensis. Rev Clin Chem 52: 779–781.
- 13. Bricker BJ, Ewalt DR, MacMillan AP, Foster G, Brew S (2000) Molecular characterization of Brucella strains isolated from marine mammals. J Clin Microbiol 38: 1258–1262.
- 14. Vizcaíno N, Cloeckaert A, Verger J, Grayon M, Fernández-Lago L (2000) DNA polymorphism in the genus Brucella. Microbes Infect 2: 1089–1100.
- 15. Brew SD, Perrett LL, Stack JA, MacMillan AP, Staunton NJ (1999) Human exposure to Brucella recovered from a sea mammal. Vet Rec 144: 483.
- 16. Sohn AH, Probert WS, Glaser CA, Gupta N, Bollen AW, et al. (2003) Human neurobrucellosis with intracerebral granuloma caused by a marine mammal Brucella spp. Emerg Infect Dis 9: 485–488.
- 17. Navarro E, Casao MA, Solera J (2004) Diagnosis of human brucelosis using PCR. Expert Rev Mol Diagn 4: 115–123.
- 18. Seibold E, Maier T, Kostrzewa M, Zeman E, Splettstoesser W (2010) Identification of Francisella tularensis by Whole-Cell Matrix-Assisted Laser Desorption Ionization–Time of Flight Mass Spectrometry: Fast, Reliable, Robust, and Cost-Effective Differentiation on Species and Subspecies Levels. J Clin Microbiol 48: 1061–1069.
- 19. Barbuddhe SB, Maier T, Schwarz G, Kostrzewa M, Hof H, et al. (2008) Rapid identification and typing of listeria species by matrix-assisted laser desorption ionization-time of flight mass spectrometry. Appl Environ Microbiol 74: 5402–5407.
- 20. Mellmann A, Cloud J, Maier T, Keckevoet U, Ramminger I, et al. (2008) Evaluation of matrix-assisted laser desorption ionization-time-of-flight mass spectrometry in comparison to 16S rRNA gene sequencing for species identification of nonfermenting bacteria. J Clin Microbiol 46: 1946–1954.
- 21. Ferreira L, Sánchez-Juanes F, Guerra IP, García García MI, Sánchez JE, et al. (2010) Microorganisms direct identification from blood culture by MALDI-TOF mass spectrometry. Clin Microbiol Infect Apr 28. [Epub ahead of print].
- 22. Christner M, Rohde H, Wolters M, Sobottka I, Wegscheider K, et al. (2010) Rapid identification of bacteria from positive blood culture bottles by use of matrix-assisted laser desorption-ionization time of flight mass spectrometry fingerprinting. J Clin Microbiol 48: 1584–1591.
- 23. Prod'hom G, Bizzini A, Durussel C, Bille J, Greub G (2010) Matrix-assisted laser desorption ionization-time of flight mass spectrometry for direct bacterial identification from positive blood culture pellets. J Clin Microbiol 48: 1481–1483.
- 24. Stevenson LG, Drake SK, Murray PR (2010) Rapid identification of bacteria in positive blood culture broths by matrix-assisted laser desorption ionization-time of flight mass spectrometry. J Clin Microbiol 48: 444–447.
- 25. La Scola B, Raoult D (2009) Direct identification of bacteria in positive blood culture bottles by matrix-assisted laser desorption ionisation time-of-flight mass spectrometry. PLoS ONE 4(11): e8041.