Tunneled central venous catheters (TCVCs) provide prolonged intravenous access for pediatric patients with severe primary immunodeficiency disease (PID) undergoing hematopoietic stem cell transplantation (HSCT). However, little is known about the epidemiology and clinical significance of TCVC-related morbidity in this particular patient group. We conducted the retrospective analysis of patients with severe PID who received percutaneous landmark-guided TCVC implantation prior to HSCT. We analyzed 92 consecutive TCVC implantations in 69 patients (median [interquartile range] age 3.0 [0–11] years) with severe combined immune deficiency (n = 39, 42.4%), chronic granulomatous disease (n = 17, 18.4%), and other rare PID syndromes (n = 36, 39.2%). The median length of TCVC observation was 144.1 (85.5–194.6) days with a total of 14,040 catheter days at risk (cdr). The overall rate of adverse events during catheter insertion was 17.4% (n = 16) and 25.0% during catheter dwell period (n = 23, catheter risk [CR] per 1000 cdr = 1.64). The most common complication was TCVC-related infection with an overall prevalence of 9.8% (n = 9, CR = 0.64), followed by late dislocation (n = 6, 6.5%, CR = 0.43), early dislocation (n = 4, 4.3%) and catheter dysfunction (n = 4, 4.3%, CR = 0.28). TCVCs are safe in children with severe PID undergoing HSCT with relatively low rates of TCVC-related infection.
Citation: Martynov I, Klima-Frysch J, Kluwe W, Engel C, Schoenberger J (2020) Safety of tunneled central venous catheters in pediatric hematopoietic stem cell recipients with severe primary immunodeficiency diseases. PLoS ONE 15(5): e0233016. https://doi.org/10.1371/journal.pone.0233016
Editor: Robert Jeenchen Chen, China Medical University Hospital, TAIWAN
Received: January 22, 2020; Accepted: April 26, 2020; Published: May 15, 2020
Copyright: © 2020 Martynov et al. This is an open access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Data Availability: All relevant data are within the manuscript and its Supporting Information files
Funding: The article processing charge was funded by the Baden-Wuerttemberg Ministry of Science, Research and Art and the University of Freiburg in the funding programme Open Access Publishing. The funders had no role in study design, data collection and analysis, decision to publish, or preparation of the manuscript.
Competing interests: The authors have no conflicts of interest relevant to this article to disclose
Primary immunodeficiency diseases (PID) comprise a wide spectrum of disorders, including more than 350 genetically defined inborn errors of adaptive and innate immunity . Although the clinical presentation of PID is highly variable, many disorders are characterized by recurrent, serious infections, autoimmune dysregulation, and aberrant inflammation if not treated appropriately . The management of PID depends on the type and severity of the underlying defect. Severe forms of PID such as severe combined immune deficiency (SCID) or chronic granulomatous disease (CGD) require cellular therapy including hematopoietic stem cell transplantation (HSCT) . The preparation for HSCT and the HSCT procedure itself involves the use of long-term, multilumen, tunneled central venous catheters (TCVCs) for administration of chemotherapy agents during conditioning, infusion of hematopoietic cells, and supportive care management including immunosuppressant agents, antibiotics, analgesics, blood components and parenteral nutrition [4, 5]. Nevertheless, to the profoundly vulnerable immunodepressed HSCT recipients, the TCVCs represents an additional source of morbidity, including procedural adverse events and infective and non-infective complications during catheter dwell period [6–9]. These TCVC-related complications may necessitate additional surgical interventions or culminate in premature catheter removal, both of which coincide with the risks of surgery and general anesthesia while also interrupting or prolonging therapy . Further, it has been shown that infections at the time of HSCT are clearly associated with poorer survival if conditioning is needed . Although the TCVC-related morbidity is highly recognized as a significant problem in pediatric HSCT recipients with severe PID, studies regarding adverse events and complications of TCVCs in this particular patient group are lacking. Therefore, we conducted an observational cohort study which aims to determine the incidence and types of procedural adverse events and dwell period catheter complications in pediatric patients with severe PID undergoing HSCT.
Patients and methods
In this retrospective, single-center observational study, all consecutive HSCT recipients between 0 and 20 years of age with severe PID who underwent implantation of TCVC at the University Medical Center of Freiburg between 01/01/2008 and 31/12/2019 were included. Data on patient and device characteristics, procedural adverse events, and complications during catheter dwell period resulting in revisional surgery or premature removal were retrieved from the Patient Data Management System (PDMS). At the time of data extraction, one of the patients was seeking treatment. All extracted data were fully anonymized before access and subsequently analysis.
Preoperative evaluation and surgical technique of catheter placement
Preoperative work-up consisted of laboratory studies including coagulation tests. We did not routinely perform ultrasound imaging of the internal jugular vein and carotid artery prior to TCVC implantation. Provided the patient did not receive antibiotic therapy due to the immunodeficiency at the time of the operation, a single-shot prophylactic perioperative antibiotics (e.g. cefuroxim) was administered. In-line filters or anti-infective/microbial lock prophylaxis were not used. Real-time ultrasound guidance for puncture of the vein was not routinely utilized. The implantation was usually performed by surgical residents supervised by board certified pediatric surgeons. Anatomical landmarks and the adapted Seldinger technique were used for catheter cannulation . All catheters were tunneled subcutaneously with the tissue ingrowth cuff positioned at the distal end of the tunnel. Positioning of the catheter tip was confirmed by fluoroscopy.
Procedural adverse events were defined as undesirable conditions during catheter placement, such as complicated guide wire insertion, accidental arterial puncture, multiple unsuccessful venous cannulations, or need for conversion to venae section. Complications during catheter dwell period requiring catheter tube repositioning, catheter removal or exchange were analyzed. These complications included TCVC-related infection, catheter dysfunction/thrombosis, and early and late catheter dislocation presenting as malposition of the catheter tip. Tip malposition was defined as the position of the tube tip far beyond the right atrial entrance, e.g. within atrium itself (deep malposition), in the superior vena cava (high malposition) or in the ipsi- or contralateral internal jugular or subclavian vein (knocked over malposition). Early dislocation was defined as a catheter tip malposition within the first 7 postoperative days. Late catheter dislocations were classified as events occurring thereafter. Accidental catheter dislocation was also considered to be a dwell period complication. Catheter-related infections were defined as the presence of bacteremia (catheter-related bloodstream infections) and/or local skin inflammation (exit-site infections and/or tunnel infections) originating from the catheter. Catheter dysfunction was defined as an inability to administer the medication or inability to aspirate irrespective of cause (e.g. thrombosis, leakage with extravasation, avulsion of tube fragments).
Dwell period complications which were treated conservatively or did not require catheter tube repositioning, removal or exchange were excluded from analysis.
Demographic, disease-related, and device-related variables were described using frequencies and percentages for categorical variables. The total number of catheter days at risk (cdr) was calculated as the total number of days from insertion to elective catheter removal or revisional surgery or premature catheter removal. The complication rate per 1000 cdr (CR) was calculated as 1000 times the number of complications, divided by the total number of cdr. Normality of data was evaluated by the Kolmogorov–Smirnov and Shapiro Wilk tests. Results for continuous non parametric data are expressed as median (interquartile range [IQR]). For analyses of continuous variables the Mann-Whitney-U-test was used. All statistical tests were two-tailed and the tests were considered significant with p < 0.05.
Patient and device characteristics
In total, this study included 92 consecutive catheter positioning procedures in 69 patients with severe PID considered for HSCT. The median age at the time of catheter implantation was 3.0 (0–11) years. The male/female ratio was 2.3 (48 boys and 21 girls). Of the 92 devices inserted, 51 (55.4%) were Groshong and 41 (44.6%) Hickman/Broviac catheters. Of these, TCVC was inserted only once in 49 (71.0%) of the patients, twice in 18 (26.1%) children, three and four times in one patient (1.45%), respectively. The internal jugular vein (n = 89, 96.7%) and the right side (n = 78, 84.8%) were most commonly used. The median time from TCVC implantation to day one of HSCT was 18.5 (13.4–28.6) days. The median length of TCVC observation was 144.1 (85.5–194.6) days and ranged between 5.2 and 588 days for a total of 14,040 cdr. Detailed patients and device characteristics are summarized in Table 1.
Procedural adverse events and dwell period complications
Overall, 16 (17.4%) adverse events during catheter placement were observed. Among them, complicated guide wire insertion after successful venous cannulation was the most common, with an overall rate of 7.6% (n = 7). Conversion to venous cut-down was necessary in 5 (5.4%) cases. No catheter insertion procedures were complicated by pneumothorax or hemothorax. No TCVC implantation related mortality occurred. The overall complication rate during catheter dwell period was 25.0% (n = 23, CR of 1.64 per 1000 cdr) with median time to surgical revision (TTR) or premature TCVC removal of 62.9 (13.3–172.1) days. Table 2 summarizes both procedural adverse events and Table 3 dwell period complications. The most frequent complication was TCVC-related infection with an overall prevalence of 9.8% (n = 9, CR = 0.64) occurring in 7 patients. Of these patients, one experienced 3 TCVC-related infections and the remaining six experienced one TCVC-related infection. Overall, 66.6% (n = 6) of the pathogens causing TCVC-related infections were Gram-positive, while in 11.1% (n = 1) the blood cultures remained sterile in the setting of applied broad-spectrum antibiotics. Fungal infection was observed in two (22.2%) patients (Table 4). There were no significant differences in median times to elective catheter explantation in patients who completed therapy compared with patients experiencing TCVC-related infections necessitating catheter revision or premature explantation (150 [114–201, range: 20–588] vs. 139 [35–209, range: 35–612], p = 0.97) (Fig 1). Further catheter dwell period complications were relatively infrequent, including late dislocation with overall frequency of 6.5% (n = 6, CR = 0.43), followed by early dislocation (n = 4, 4.3%) and catheter dysfunction (n = 4, 4.3%, CR = 0.28).
Our study addressed the poorly investigated epidemiology of TCVC-related morbidity in pediatric HSCT recipients with severe PID. This is, to our knowledge, the first study examining adverse events during catheter implantation and dwell period complications in this particular patient group. We found that both adverse events and dwell period complications are common in our patient cohort, with an overall incidence of 17.4% and 25.0% respectively.
The incidence of adverse events observed during catheter implantation is consistent with the 4.5%–22.0% incidence reported in prior studies of pediatric oncology patients [12–14]. Although ultrasound guidance has been shown to reduce the incidence of failures and inadvertent arterial punctures in pediatric central venous catheterization , percutaneous landmark technique was predominantly utilized in our study as this approach is traditional at our institution. In our study, the most frequent procedural adverse event was complicated guide wire insertion after successful venous puncture, which accounted for 7.6%. Although we consistently included this type of adverse events in our analysis, this does not constitute a cannulation failure in the stricter sense. Moreover, it is questionable whether ultrasound guidance would lower the incidence rate, as the insertion of the guide wire is not ultrasound guided. In contrast, inadvertent arterial and multiple unsuccessful venous punctures represent a typical cannulation failure and the incidence of both could be reduced when using ultrasound . Nevertheless, in our study, the incidence of both complications (inadvertent arterial puncture, n = 2, 2.2%; multiple unsuccessful venous punctures, n = 2, 2.2%) was even lower than that reported in the meta-analysis conducted by de Souza et al., with an overall incidence rate of 5.4% for inadvertent arterial puncture and 5% for unsuccessful venous punctures as reported by others [13, 17]. Finally, in our study population, there were 5 (5.4%) patients requiring conversion from percutaneous catheter insertion to open venous cut-down. Although we classified this mode of tunneled central venous catheter implantation as an adverse event (conversion procedure), this technique is also used as the primary procedure in several centers, especially in small children [18, 19]. In our patients, the conversion cut-down was not associated with further complications. Although the implantation setting, catheter cannulation technique and the experiences of the pediatric surgeons were the same as recently reported by our group investigating TCVC-related morbidity in pediatric oncological patients, the 17.4% incidence of adverse events in the present study was higher compared to 12.8% incidence .
Data regarding the incidence of TCVC-related complications during dwell period in pediatric HSCT recipients with PID are lacking. There are only few studies predominantly from the adult literature available investigating TCVC-related morbidity on patients with oncological diseases undergoing HSCT [21–23]. Amongst our patient population, we observed the overall rate of 25.0% (n = 23, CR = 1.64) complications during catheter dwell period. Where comparable, this complications rate is in concordance with the data from a systematic review conducted by Ullman et al. including children with oncological diseases without HSCT and showing that 25% of central vein access devices failed before completion of therapy (CR = 1.97 per 1000 catheter days) .
In the current study, the most frequent dwell period complications were TCVC-related infections with the overall incidence rate of 9.8% (n = 9, CR = 0.64) leading to premature catheter removal after median time of 139 catheter in situ days. Even if not directly comparable with the presented data, according to the literature regarding pediatric oncological patients without HSCT, the catheter-associated infections represent the most common complications ranging from 14% to 58% with corresponding CR of 0.57–2.8 episodes per 1000 CVC days at risk [8, 24–27]. The relatively low rates of TCVC-related infection observed in our patients with severe PID undergoing HSCT were highly unexpected due to the fact that these children are per se at very high risk for infections . However, it should be noted that, in this study, we investigated only TCVC-related infections leading to catheter removal, neglecting conservatively treated TCVC-related infection and non-TCVC-related infections. Further non-infective catheter-related complications were very low ranging from 4.3% to 6.5%. These low rates of non-infective complications are in concordance with other studies [29, 30].
Our study has several limitations, including its retrospective nature, single-center setup, and relatively small sample size. The implantation technique and management of TCVC during HSCT are based on local protocols, which may differ from those applied in other centers. Thus, our data on TCVC-related morbidity should be interpreted with caution, and prospective studies are needed. However, while we acknowledge these limitations, our study has also strengths, including patient cohort with very rare diseases.
This is the first study to date characterizing TCVC-related morbidity in pediatric HSCT recipients with severe PID. We demonstrated that the implantation of TCVC is associated with 17.4% of adverse events and that dwell period complications occur in 25.0% of patients. Unexpected, the incidence of TCVC-related infections was relatively low. The findings of our study provide valuable information on the epidemiology of tunneled central venous catheters indicating their safety in this particular patient group.
- 1. Picard C, Bobby Gaspar H, Al-Herz W, Bousfiha A, Casanova JL, Chatila T, et al. International Union of Immunological Societies: 2017 Primary Immunodeficiency Diseases Committee Report on Inborn Errors of Immunity. Journal of clinical immunology. 2018;38(1):96–128. Epub 2017/12/12. pmid:29226302.
- 2. McCusker C, Upton J, Warrington R. Primary immunodeficiency. Allergy, asthma, and clinical immunology: official journal of the Canadian Society of Allergy and Clinical Immunology. 2018;14(Suppl 2):61. Epub 2018/10/03. pmid:30275850.
- 3. Modell V, Gee B, Lewis DB, Orange JS, Roifman CM, Routes JM, et al. Global study of primary immunodeficiency diseases (PI)—diagnosis, treatment, and economic impact: an updated report from the Jeffrey Modell Foundation. Immunologic research. 2011;51(1):61–70. Epub 2011/09/22. pmid:21935653.
- 4. Wahlstrom JT, Dvorak CC, Cowan MJ. Hematopoietic Stem Cell Transplantation for Severe Combined Immunodeficiency. Current pediatrics reports. 2015;3(1):1–10. Epub 2015/03/31. pmid:25821657.
- 5. Lazarus HM, Trehan S, Miller R, Fox RM, Creger RJ, Raaf JH. Multi-purpose silastic dual-lumen central venous catheters for both collection and transplantation of hematopoietic progenitor cells. Bone marrow transplantation. 2000;25(7):779–85. Epub 2000/04/04. pmid:10745265.
- 6. Ruggiero A, Barone G, Margani G, Nanni L, Pittiruti M, Riccardi R. Groshong catheter-related complications in children with cancer. Pediatric blood & cancer. 2010;54(7):947–51. Epub 2010/02/18. pmid:20162685.
- 7. Ullman AJ, Marsh N, Mihala G, Cooke M, Rickard CM. Complications of Central Venous Access Devices: A Systematic Review. Pediatrics. 2015;136(5):e1331–44. Epub 2015/10/16. pmid:26459655.
- 8. Cecinati V, Brescia L, Tagliaferri L, Giordano P, Esposito S. Catheter-related infections in pediatric patients with cancer. European Journal of Clinical Microbiology & Infectious Diseases. 2012;31(11):2869–77. pmid:22661169
- 9. Johnson EM, Saltzman DA, Suh G, Dahms RA, Leonard AS. Complications and risks of central venous catheter placement in children. Surgery. 1998;124(5):911–6. Epub 1998/11/21. pmid:9823406.
- 10. Pai S-Y, Logan BR, Griffith LM, Buckley RH, Parrott RE, Dvorak CC, et al. Transplantation Outcomes for Severe Combined Immunodeficiency, 2000–2009. New England Journal of Medicine. 2014;371(5):434–46. pmid:25075835.
- 11. Seldinger SI. Catheter replacement of the needle in percutaneous arteriography; a new technique. Acta radiologica. 1953;39(5):368–76. Epub 1953/05/01. pmid:13057644.
- 12. Grebenik CR, Boyce A, Sinclair ME, Evans RD, Mason DG, Martin B. NICE guidelines for central venous catheterization in children. Is the evidence base sufficient? British journal of anaesthesia. 2004;92(6):827–30. Epub 2004/05/04. pmid:15121722.
- 13. Bruzoni M, Slater BJ, Wall J, St Peter SD, Dutta S. A prospective randomized trial of ultrasound- vs landmark-guided central venous access in the pediatric population. Journal of the American College of Surgeons. 2013;216(5):939–43. Epub 2013/03/13. pmid:23478546.
- 14. van den Bosch CH, van der Bruggen JT, Frakking FNJ, Terwisscha van Scheltinga CEJ, van de Ven CP, van Grotel M, et al. Incidence, severity and outcome of central line related complications in pediatric oncology patients; A single center study. Journal of pediatric surgery. 2019;54(9):1894–900. Epub 2018/11/13. pmid:30415957.
- 15. de Souza TH, Brandao MB, Nadal JAH, Nogueira RJN. Ultrasound Guidance for Pediatric Central Venous Catheterization: A Meta-analysis. Pediatrics. 2018;142(5). Epub 2018/10/27. pmid:30361397.
- 16. Brass P, Hellmich M, Kolodziej L, Schick G, Smith AF. Ultrasound guidance versus anatomical landmarks for internal jugular vein catheterization. The Cochrane database of systematic reviews. 2015;1:Cd006962. Epub 2015/01/13. pmid:25575244.
- 17. Zanolla GR, Baldisserotto M, Piva J. How useful is ultrasound guidance for internal jugular venous access in children? Journal of pediatric surgery. 2018;53(4):789–93. pmid:28843837
- 18. Alshafei A, Tareen F, Maphango N, White D, O’Connor B, Sriparan T. Open tunneled central line insertion in children—External or internal jugular vein? Journal of pediatric surgery. 2018;53(11):2318–21. Epub 2018/07/29. pmid:30054059.
- 19. Blum LV, Abdel-Rahman U, Klingebiel T, Fiegel H, Gfroerer S, Rolle U. Tunneled central venous catheters in children with malignant and chronic diseases: A comparison of open vs. percutaneous implantation. Journal of pediatric surgery. 2017;52(5):810–2. Epub 2017/02/12. pmid:28188038.
- 20. Martynov I, Raedecke J, Klima-Frysch J, Kluwe W, Schoenberger J. Outcome of landmark-guided percutaneously inserted tunneled central venous catheters in infants and children under 3 years with cancer. Pediatric blood & cancer. 2018;65(10):e27295. Epub 2018/06/27. pmid:29943891.
- 21. Yeral M, Boğa C, Oğuzkurt L, Alışkan HE, Özdoğu H, Demiroğlu YZ. Tunnelled central venous catheter-related problems in the early phase of haematopoietic stem cell transplantation and effects on transplant outcome. Turkish journal of haematology: official journal of Turkish Society of Haematology. 2015;32(1):51–7. Epub 2015/03/26. pmid:25805675.
- 22. Biffi R, Pozzi S, Agazzi A, Pace U, Floridi A, Cenciarelli S, et al. Use of totally implantable central venous access ports for high-dose chemotherapy and peripheral blood stem cell transplantation: results of a monocentre series of 376 patients. Annals of oncology: official journal of the European Society for Medical Oncology. 2004;15(2):296–300. Epub 2004/02/05. pmid:14760125.
- 23. Platzbecker U, Illmer T, Schaich M, Freiberg-Richter J, Helwig A, Plettig R, et al. Double lumen port access in patients receiving allogeneic blood stem cell transplantation. Bone marrow transplantation. 2001;28(11):1067–72. pmid:11781618
- 24. Chesshyre E, Goff Z, Bowen A, Carapetis J. The prevention, diagnosis and management of central venous line infections in children. The Journal of infection. 2015;71 Suppl 1:S59–75. Epub 2015/05/03. pmid:25934326.
- 25. Pinon M, Bezzio S, Tovo PA, Fagioli F, Farinasso L, Calabrese R, et al. A prospective 7-year survey on central venous catheter-related complications at a single pediatric hospital. European journal of pediatrics. 2009;168(12):1505–12. Epub 2009/03/21. pmid:19301034.
- 26. Fratino G, Molinari AC, Parodi S, Longo S, Saracco P, Castagnola E, et al. Central venous catheter-related complications in children with oncological/hematological diseases: an observational study of 418 devices. Annals of oncology: official journal of the European Society for Medical Oncology. 2005;16(4):648–54. Epub 2005/01/29. pmid:15677621.
- 27. Paioni P, Kuhn S, Strassle Y, Seifert B, Berger C. Risk factors for central line-associated bloodstream infections in children with tunneled central venous catheters. American journal of infection control. 2019. Epub 2019/08/10. pmid:31395289.
- 28. Srinivasan A, Wang C, Srivastava DK, Burnette K, Shenep JL, Leung W, et al. Timeline, epidemiology, and risk factors for bacterial, fungal, and viral infections in children and adolescents after allogeneic hematopoietic stem cell transplantation. Biology of blood and marrow transplantation: journal of the American Society for Blood and Marrow Transplantation. 2013;19(1):94–101. Epub 2012/08/28. pmid:22922523.
- 29. Beck O, Muensterer O, Hofmann S, Rossmann H, Poplawski A, Faber J, et al. Central Venous Access Devices (CVAD) in Pediatric Oncology Patients-A Single-Center Retrospective Study Over More Than 9 Years. Frontiers in pediatrics. 2019;7:260. Epub 2019/07/12. pmid:31294007.
- 30. Cesaro S, Cavaliere M, Pegoraro A, Gamba P, Zadra N, Tridello G. A comprehensive approach to the prevention of central venous catheter complications: results of 10-year prospective surveillance in pediatric hematology-oncology patients. Annals of hematology. 2016;95(5):817–25. Epub 2016/03/11. pmid:26961934.