PCR-ribotyping has been adopted in many laboratories as the method of choice for C. difficile typing and surveillance. However, issues with the conventional agarose gel-based technique, including inter-laboratory variation and interpretation of banding patterns have impeded progress. The method has recently been adapted to incorporate high-resolution capillary gel-based electrophoresis (CE-ribotyping), so improving discrimination, accuracy and reproducibility. However, reports to date have all represented single-centre studies and inter-laboratory variability has not been formally measured or assessed. Here, we achieved in a multi-centre setting a high level of reproducibility, accuracy and portability associated with a consensus CE-ribotyping protocol. Local databases were built at four participating laboratories using a distributed set of 70 known PCR-ribotypes. A panel of 50 isolates and 60 electronic profiles (blinded and randomized) were distributed to each testing centre for PCR-ribotype identification based on local databases generated using the standard set of 70 PCR-ribotypes, and the performance of the consensus protocol assessed. A maximum standard deviation of only ±3.8bp was recorded in individual fragment sizes, and PCR-ribotypes from 98.2% of anonymised strains were successfully discriminated across four ribotyping centres spanning Europe and North America (98.8% after analysing discrepancies). Consensus CE-ribotyping increases comparability of typing data between centres and thereby facilitates the rapid and accurate transfer of standardized typing data to support future national and international C. difficile surveillance programs.
Citation: Fawley WN, Knetsch CW, MacCannell DR, Harmanus C, Du T, Mulvey MR, et al. (2015) Development and Validation of an Internationally-Standardized, High-Resolution Capillary Gel-Based Electrophoresis PCR-Ribotyping Protocol for Clostridium difficile. PLoS ONE 10(2): e0118150. https://doi.org/10.1371/journal.pone.0118150
Academic Editor: Alex Friedrich, University Medical Center Groningen, NETHERLANDS
Received: August 8, 2014; Accepted: January 5, 2015; Published: February 13, 2015
This is an open access article, free of all copyright, and may be freely reproduced, distributed, transmitted, modified, built upon, or otherwise used by anyone for any lawful purpose. The work is made available under the Creative Commons CC0 public domain dedication
Data Availability: All relevant data are within the paper. Data on isolates in the panel have been made available online in a National Center for Biotechnology Information BioProject database (NCBI) (http://www.ncbi.nlm.nih.gov/bioproject/248340).
Funding: This work was performed within the ECDC network ‘Supporting capacity building for surveillance of Clostridium difficile infections at European level’, and included a collaboration with the European Study group of C. difficile (ESGCD), a subgroup of the European Society for Clinical Microbiology and Infectious Diseases. The funders had no role in study design, data collection and analysis, decision to publish, or preparation of the manuscript.
Competing interests: The authors have declared that no competing interests exist.
Clostridium difficile infection (CDI) is a major nosocomial disease, placing a considerable burden on healthcare resources. A recent European report estimated the associated per-case cost of CDI at £4577–£8843 . C. difficile has commonly been associated with hospital outbreaks, and most notably the epidemic spread of NAP1/BI/PCR ribotype 027 [2, 3]. Studies of C. difficile epidemiology from diverse geographical locations have described different locally or nationally prevalent strains [4, 5, 6], some of which were associated with increased severity and/or reduced antimicrobial susceptibility [2, 5]. DNA typing systems have been crucial in tracking the emergence and spread of C. difficile strains. Notably, individual surveillance programs have seen significant changes in the prevalence and epidemiology of NAP1/BI/PCR ribotype 027 [7, 8]. This highlights the important role that DNA typing systems play in epidemiological surveillance and in targeting informing infection control interventions.
Most epidemiological studies of C. difficile are local or national in scope, and as such have used different typing techniques. Pulsed-field gel electrophoresis (PFGE) , restriction enzyme analysis (REA) , PCR-Ribotyping , toxinotyping , arbitrary-primed PCR (AP-PCR) , random amplification of polymorphic DNA (RAPD)  and REP-PCR  have all been used routinely for C. difficile typing; thus, it is not uncommon for isolates to be referred to by multiple typing designations (eg: NAP1/BI/027; PFGE/REA/PCR-ribotyping). This can be problematic when trying to compare studies or conduct large scale epidemiological analyses. A multi-centre, international study assessed seven typing techniques, and drew attention to the lack of a consensus technique with proven inter-laboratory reproducibility . PCR ribotyping has gained acceptance in Europe as the typing method of choice, with a common nomenclature adopted across most countries (although some local systems have emerged) and different primers can be employed [15, 16, 17]. Agarose gel-based methods offer relatively high sensitivity, specificity and reproducibility relative to other PCR-based techniques, but inter-laboratory variation in banding patterns and interpretation mean that data are difficult to compare between typing centres .
The advent of high-resolution, capillary gel electrophoresis-based fragment analysis addressed many of these issues. Reduced PCR cycling and electrophoresis times combined with the ability to increase batch sizes (using multiple fluorescent detection dyes) represents a relatively rapid, high-throughput and low cost means of epidemiological C. difficile investigation [18, 19, 20]. However, CE-ribotyping has yet to gain widespread acceptance, and is hampered by a lack of multi-centre evaluations demonstrating its reliability and the portability of data. Indra et al. proposed and established a web-based repository of electronic C. difficile PCR ribotyping data (WebRibo) . Unfortunately, the lack of protocol standardization and consequent limited capacity to accurately categorize submitted patterns using internationally recognised PCR-ribotype nomenclature has to date diminished the effectiveness of this approach . We describe here the first multi-centre development, validation and evaluation of a standardised protocol for high-resolution PCR-ribotyping using capillary gel electrophoresis.
Materials and Methods
A panel of well characterised C. difficile isolates representing 70 distinct PCR-ribotypes was used in this study . The panel comprised PCR-ribotypes known to be associated with human CDI in Europe. This collection was assembled using type strains previously shared between two established PCR-ribotyping laboratories (Clostridium difficile Network for England and Northern Ireland (CDRN), and National Reference Laboratory for Clostridium difficile at University Medical Centre, Leiden). All PCR-ribotypes were originally assigned in association with the Anaerobic Reference Laboratory at Cardiff (ARL) using agarose gel-based PCR-ribotyping technique (Table 1). Data on isolates in the panel have been made available on-line in a National Center for Biotechnology Information BioProject database (NCBI) (http://www.ncbi.nlm.nih.gov/bioproject/248340). In addition, a subset (European Centre for Disease Prevention and Control (ECDC)-Brazier collection) is available to all reference laboratories in Europe who participate in the European C. difficile infection study network (ECDIS-NET) .
Development of consensus capillary gel electrophoresis-based PCR-ribotyping
Protocols from each of the four participating laboratories were initially compared. Reagents, PCR conditions, sequencer model, polymer, array length and separation parameters were recorded. A consensus protocol was developed by comparing and contrasting the four internal protocols and was tested internally against the panel of reference strains to evaluate performance. Following refinements, a consensus protocol was distributed to the four participating national laboratories for feedback, testing and validation. Individual internal protocols and the final consensus protocol are summarised in Table 2.
DNA was extracted from C. difficile cultures using Chelex-100 resin. Briefly, several colonies of test organism, taken from fresh 24 h cultures on a non-selective solid medium, were fully resuspended in 100μL 5% w/v Chelex-100 resin (Bio-Rad, Hertfordshire, UK) in molecular grade H2O. The bacterial suspension was heated at 100°C for 10 min and the resultant lysate was centrifuged for 2 min at 13,000 rpm. The supernatant was collected and DNA concentration was adjusted to 100ng/μL and used immediately for PCR.
Amplification of 16S-23S intergenic spacer region
PCR primers 5’-GTGCGGCTGGATCACCTCCT-3’ (16S) and 5’-CCCTGCACCCTT-AATAACTTGACC-3’) (23S) were used for amplification . The 16S primer was labelled at the 5’ end with either 6-carboxyfluorescein (6-FAM) or 5-tetrachlorofluorescein (TET). Individual PCR reactions comprised 2 μL prepared bacterial DNA, 0.2 uM each primer and 12.5 uL HotStarTaq mastermix (Qiagen, Manchester, UK). Reactions were made up to 25 μL with molecular grade water. Conventional thermal cycling was as follows: initial polymerase activation at 95°C for 15 min, followed by 24cycles at 95°C for 1 min, 57°C for 1 min and 72°C for 1 min, with a final completion stage at 72°C for 30 min.
Capillary gel electrophoresis
PCR products were analysed on either an ABI 3100 or 3130xl genetic analyser using a 16 capillary 36 cm array with either POP-4 or POP-7 separation matrix (all Life Technologies, Paisley, UK). Genetic analysers were calibrated for the G5 dyeset. GeneScan 1200 LIZ standard was used as internal sizing reference (Life Technologies, Paisley, UK). Fragment analysis samples contained 1 μL amplified DNA, 0.5 μL 1200 LIZ standard and 8.5 μL Hi-Di formamide (Life Technologies, Paisley, UK). Samples were injected at 5 kV for 5 sec and resolved using a separation voltage of 6.5 kV for 103 min. Major peaks in fluorescent signal were imported into BioNumerics v.5.1 software (Applied Maths, Sint-Martens-Latem) to complete the method validation. Fragments were initially sized using either PeakScanner v.1.0 or GeneMapper v.4.0 software (both Life Technologies, Paisley, UK), or were imported into BioNumerics directly. All signals with a height <10% that of the highest peak in the individual profile were excluded (as these were considered background rather than evidence of a major DNA fragment). For peaks <1.5bp different in size, the lower intensity peak was also excluded .
Consensus method validation
Multi-centre validation of the proposed consensus method is outlined in Fig. 1. Briefly, validation was performed in three stages:
STAGE 1: 70 well characterised ribotypes disseminated from Netherlands to each laboratory for ribotyping (data (i) held locally for future comparsion/ribotype assignment, (ii) sent to UK laboratory); STAGE 2: 50 anonymised isolates disseminated from Netherlands to each laboratory for ribotype identification (assignments sent to UK laboratory for analysis); STAGE 3: 60 anonymised data files disseminated from UK to each laboratory for ribotype identification (assignments sent to UK for analysis).
VALIDATION STAGE 1: To examine inter-laboratory variation and reproducibility, and establish local reference databases. The panel of 70 known distinct PCR-ribotypes was distributed centrally (from the Netherlands laboratory) to all other participating national surveillance laboratories (training set). CE-ribotyping was performed at each centre, using the proposed consensus method, and data were held locally in BioNumerics. These training data were used for all subsequent PCR-ribotype identifications throughout the validation process, and also returned to the UK laboratory for analysis and use in validation stage 3.
VALIDATION STAGE 2: To examine accuracy of PCR-ribotype identification from locally generated data. A panel of 50 isolates were anonymised and distributed centrally from the Netherlands laboratory (isolate challenge set). CE-ribotyping was performed at each centre using the proposed consensus method. A PCR-ribotype was assigned to each isolate after DNA profile comparison with the data generated using the training set of isolates in validation stage 1 (BioNumerics). PCR-ribotype assignments from each centre were returned to the UK laboratory for analysis. PCR-ribotype identities for the 50 isolates were then circulated by the Netherlands laboratory.
VALIDATION STAGE 3: To examine portability of data and accuracy of PCR-ribotype identification from centrally generated electronic data. A dataset of sized fragments for 60 PCR-ribotypes was extracted from data returned to the UK laboratory in validation stage 1 (15 from each participating centre). Data files were anonymised and distributed by email to all other participating laboratories (data challenge set). A PCR-ribotype was assigned to each data file after DNA profile comparison with the data generated using the training set of isolates in validation stage 1 (BioNumerics). PCR-ribotype assignments from each centre were returned to the UK laboratory for analysis.
VALIDATION STAGE ONE—training set
The consensus CE-ribotyping method was applied to the panel of 70 well characterised ribotypes at all four participating centres. Each laboratory generated between 5 and 11 fragments per ribotype, ranging in size from 223–667 bp. A high level of profile similarity was observed between several ribotypes in the panel (Fig. 2). A high level of reproducibility was observed; the standard deviations (SDs) for fragment sizes reported from all four laboratories on each individual fragment ranged from 0.14–3.8 bp (mean 1.25, median 1.17). The maximum reported difference in individual fragment size ranged from 0.3–8.4 bp (mean 2.6 bp, median 2.5 bp) (Table 3). The magnitude of inter-laboratory variation was clearly associated with increasing fragment size. The maximum reported size differences of the smallest and largest fragments per ribotype profile were significantly different (Mann Whitney U test, (p<0.05); median SD of smallest and largest fragments per ribotype profile: 0.39 and 2.13, respectively). No significant differences were observed in the fragment sizes generated on the 3100 versus the 3130xl automated sequencing instruments.
PCR-ribotypes 027 and 081 differ from one another by only a ~20bp difference at position d. Similarly PCR-ribotypes 015 and 046 differ by only a ~20bp difference at position b. Discriminating features between these very similar profiles are indicated (arrows) and associated fragment sizes are highlighted in bold. Relative fragment size was the only parameter used to discriminate between ribotype profiles. Relative peak heights (relative fluorescent units, y-axis) within profiles lacked reproducibility for some ribotypes and therefore this parameter was not used.
VALIDATION STAGE TWO—isolate challenge set
The CE-ribotyping method was applied to the panel of 50 anonymised isolates distributed from the Netherlands laboratory (isolate challenge set). Formal PCR-ribotype identification was performed by each laboratory using training data generated in validation stage one. A list of PCR-ribotypes represented in this challenge set and PCR-ribotyping performance scores for each centre are shown in Tables 1 and 4.
Laboratories identified the correct PCR-ribotype in 195/200 (97.5%) of cases (or correctly stated that the profile generated was not recognisable, based on the training data generated in validation stage one). Forty-five of 50 (90.0%) PCR-ribotypes were discriminated successfully by all four centres; all PCR-ribotypes were discriminated successfully by a minimum of three centres. Analysis of the discrepancies revealed that identification failures (n = 5) were not associated with any particular testing centre or any specific PCR-ribotype (Table 5). Data consistent with PCR-ribotype 064 were generated in place of PCR-ribotype 029 by centre B indicating likely transcription of the test isolate. The profile for PCR-ribotype 035 was obscured with additional peaks consistent with sample contamination at centre C. PCR-ribotypes 058 and 118 were associated with amplification failure at centre C and likely to represent attempted testing of an organism other than C. difficile. An error in the construction of the challenge set resulted in the circulation of PCR-ribotype 036 in place of PCR-ribotype 198. Centre D successfully identified the isolate as PCR-ribotype 036, but should have technically returned a result of “unidentified” as PCR-ribotype 036 was not present in the training set. All test failures highlighted errors associated with the manipulation of test isolates as opposed to inconsistencies with the prescribed method, or data quality.
VALIDATION STAGE THREE—data challenge set
A dataset of sized fragments for 60 blinded ribotypes was distributed electronically from the UK laboratory. Formal PCR-ribotype identification was performed by each laboratory using training data generated in validation stage one. A list of PCR-ribotypes represented in this challenge set and PCR-ribotyping performance scores for each centre are shown in Tables 1 and 4. Laboratories identified the correct ribotype in 237/240 (98.8%) cases. All four centres correctly identified 57/60 (95%) ribotype profiles. All isolates were correctly identified by a minimum of three centres. Failure to identify the correct ribotype was only associated with Centre B for the remaining three isolates. Analysis of the discrepancies revealed that an incomplete set of PCR-ribotype profiles (generated from the training set) had been used to formally identify PCR-ribotypes at this centre, resulting in three unidentified profiles. Using the same profile data, a second identification attempt (containing a full set of PCR-ribotype reference profiles) was performed at centre B and 60/60 (100%) PCR-ribotype profiles were correctly identified. This process highlighted an error during database identification at a single centre as opposed to inconsistencies with the prescribed method, or data quality. Following analysis of the discrepancies, the number of correctly identified PCR-ribotype profiles in both validation stage 3 and the overall study was increased to 240/240 (100%) and 435/440 (98.9%), respectively.
Increased awareness of CDI is accompanied by the need to develop a consensus typing and analysis method that will allow the establishment of a standardised nomenclature and greater understanding of C. difficile epidemiology. This should be robust, reliable and generate high resolution, portable data, and function at local, national and international levels. Existing reports involving CE-ribotyping have represented only single-centre studies and inter-laboratory variability has not been formally assessed [18, 20]. We report a high level of reproducibility and accuracy associated with a consensus CE-ribotyping protocol. A maximum SD of only ±3.8 bp was recorded in individual fragment sizes, and 98.2% blinded ribotypes were correctly identified at four international ribotyping centres spanning Europe and North America (rising to 98.8% after analysing the discrepancies). Our results demonstrate that this consensus protocol generates sufficiently low level inter-laboratory variation to support accurate PCR-ribotyping for international C. difficile surveillance.
Indra et al. first combined conventional PCR-ribotyping and capillary based electrophoresis techniques (CE-ribotyping) and demonstrated superiority of this approach over the conventional agarose gel-based method . More recently, Xiao et al. reported that the performance of capillary-electrophoresis exceeded that of a commercial high performance electrophoresis system when applied to PCR-ribotyping. . Several epidemiological studies have been conducted using subtly different CE-ribotyping protocols, making future inter-laboratory comparison of results difficult [18, 20, 24, 25]. We hope that the provision of a consensus CE-ribotyping protocol will encourage laboratories to use the technique in the knowledge that data will be directly comparable across centres.
We have shown that a basic crude DNA extraction method is sufficient to demonstrate good reproducibility of the system. However, many laboratories have more sophisticated higher-throughput DNA extraction platforms, which invariably yield good quality DNA, and these could be expected to be viable alternatives. Several different sets of PCR primers have been reported for use in PCR-ribotyping in recent years [15, 17, 26]. The primers originally described by Stubbs et al. unfortunately had a 4 bp mismatch in the 23S reverse primer  and were subsequently redesigned . It is generally accepted that these primer sets offer equivalent levels of performance and discrimination, but generate different relative fragment lengths. Bidet et al. primers generate fragments ~34 bp shorter than those designed by Stubbs et al., and ~53 bp shorter than those designed by Sadeghifard et al. The loss of accuracy associated with sizing larger DNA fragments when using electrophoresis should ideally be minimised, and the selection of primers that generate smaller fragments would therefore be preferable. In order to avoid potential primer:template mismatches and minimise inter-laboratory variation, the primers designed by Bidet et al., were selected for use in this consensus method . It is important that standardized protocols for multi-centre surveillance can be easily adopted by participating laboratories. PCR consumables can be acquired or changed relatively easily, but access to specific DNA sequencing instrumentation can present difficulties. It is therefore essential that a standardised protocol can operate on more than one instrument type without significant variation in ribotype profiles. Our study has incorporated the use of two different sequencing instruments with little effect on PCR-ribotype recognition or fragment-size variation. These results suggest that laboratories can adopt the consensus protocol for use on different automated sequencers; further validation is required to confirm this issue.
Previous reports using CE-ribotyping have called for validation of the technique using internationally accepted reference strains [18, 20]. The standardised consensus method used here was tested on a well characterised collection of 70 different PCR-ribotypes [22, 23] to evaluate the performance of the protocol against prevalent and clinically relevant PCR-ribotypes. Data on this set of strains are available on-line in a NCBI BioProject database to facilitate working in this field (http://www.ncbi.nlm.nih.gov/bioproject/248340). Several examples of profiles that shared a very high level of similarity were incorporated in this study (eg. 027 and 081; 015 and 046; 106 and 174; 014 and 020) (Fig. 2). CE ribotyping was able to consistently discriminate between all these isolates. It is generally accepted that such discrimination is difficult with conventional agarose gel-based PCR-ribotyping unless gel quality is particularly high. The number of unique PCR-ribotypes has expanded rapidly in recent years and now exceeds 650 (Fawley, Wilcox, unpublished). As this library expands, the number of ribotypes with similar profile is likely to increase. In turn, the interpretation of PCR-ribotype profiles will become more complex. CE-ribotyping offers a solution for future ribotyping strategies by offering accurate fragment sizing and increased discrimination between ribotypes, including those with similar profiles that can be confused using conventional agarose gel-based PCR-ribotyping. CE-ribotyping has already confirmed reproducible differences in DNA profile of isolates previously assigned to the same ribotype using agarose gel electrophoresis, most notably those associated with ribotypes 014, 001 and 027 [18, 20, 27]. Multi-locus sequence typing studies have further shown that individual ribotypes tend to be associated with a single ST-type, but that a small number are associated with multiple ST-types [22, 27, 28].
CE-ribotyping represents a cost-effective technique to further discriminate between ribotypes of very similar DNA profile in order to further understand phylogenetic relationships. The consumables price per test for PCR-ribotyping by CE-method and by conventional agarose gel-based method are comparable (under EUR 10) . Current opinion would indicate that the largest barrier to widespread adoption of CE-ribotyping remains the acquisition of an automated sequencing instrument. However, such instruments are now in widespread use; most medical and academic institutions use automated sequencing technology and many offer access to their instruments (on or off site) on a service provision basis. In this way CE-ribotyping could reasonably be performed in any laboratory and need not be reserved for larger reference centres.
Our results suggest that data generated using the consensus method are fully portable; 98.8% (rising to 100% after analysing the discrepancies) of ribotype profiles were correctly identified when data were exchanged electronically between participating laboratories. Hitherto, surveillance programs have been impeded due to the inability of laboratories to exchange typing data efficiently. Notably, understanding the spread of NAP1/BI/PCR ribotype 027 strains across N. America and Europe was affected by inefficient inter-laboratory exchange of PCR-ribotyping data, involving the laborious exchange of bacterial isolates to establish significant epidemiological links [2, 6]. We have demonstrated that a standardised CE-ribotyping protocol allows easy electronic exchange of typing data, which could streamline communication and so reduce the time for countries/regions to become aware of new and emerging ribotypes.
There have been calls for improvements to PCR-ribotype nomenclature [19, 29]. Stubbs et al. first assembled a library of PCR-ribotypes  but, while this collection has expanded to >650 types, lack of direct access to this library has led some to use local ribotype nomenclatures. These have only local immediate applicability. The UK library was developed using agarose gel-based ribotyping, and further development is hindered by interpretation issues associated with this technique. Standardised and fully portable ribotyping data would be a significant step forwards to consolidating a single library of C. difficile reference strains, which would bring clarity for researchers and those involved in surveillance of this important pathogen.
The four testing centres that participated in this study (in no specific order) were:
The Department of Microbiology, Leeds Teaching Hospitals NHS Trust, Leeds, UK; The Centers for Disease Control and Prevention (CDC), Atlanta, USA; The Department of Medical Microbiology, Centre of Infectious Diseases, Leiden University Medical Centre, Leiden, Netherlands and The Public Health Agency of Canada (PHAC), Winnipeg, Canada.
Conceived and designed the experiments: WNF MHW EJK CWK TD DRM. Performed the experiments: WNF CWK CH TD DRM AP LA. Analyzed the data: WNF DRM. Contributed reagents/materials/analysis tools: EJK MHW MM DRM. Wrote the paper: WNF MHW.
- 1. Wiegand PN, Nathwani D, Wilcox MH, Stephens J, Shelbaya A, et al. Clinical and economic burden of Clostridium difficile infection in Europe: a systematic review of healthcare-facility-acquired infection. J Hosp Infect. 2012;77: 1–14.
- 2. Pepin J, Valiquette L, Alary ME, Villemure P, Pelletier A, et al. Clostridium difficile-associated diarrhea in a region of Quebec from 1991 to 2003: a changing pattern of disease severity. CMAJ. 2004;171: 466–472. pmid:15337727
- 3. Khanna S, Pardi DS, Aronson SL, Kammer PP, Orenstein R, et al. The Epidemiology of Community-acquired Clostridium difficile infection: a population-based study. Am J Gastroenterol. 2012;107: 89–95. pmid:22108454
- 4. Bauer MP, Notermans DW, van Benthem BH, Brazier JS, Wilcox MH, et al. ECDIS Study Group. Clostridium difficile infection in Europe: a hospital-based survey. Lancet. 2011;377: 63–73. pmid:21084111
- 5. Tenover FC, Tickler IA, Persing DH. Antimicrobial-resistant strains of Clostridium difficile from North America. Antimicrob Agents Chemother. 2012;56: 2929–2932. pmid:22411613
- 6. Freeman J, Bauer MP, Baines SD, Corver J, Fawley WN, Goorhuis B, et al. The changing epidemiology of Clostridium difficile infections. Clin Microbiol Rev. 2010;23: 529–49. pmid:20610822
- 7. Wilcox MH, Shetty N, Fawley WN, Shemko M, Coen P, Birtles A, et al. Changing epidemiology of Clostridium difficile infection following the introduction of a national ribotyping-based surveillance scheme in England. Clin Infect Dis. 2012;55: 1056–1063. pmid:22784871
- 8. Hensgens MP, Goorhuis A, Notermans DW, van Bentham BH, Kuijper EJ. Decrease of hypervirulent Clostridium difficile PCR ribotype 027 in the Netherlands. Euro Surveill. 2009;14: 19402. pmid:19941791
- 9. Tenover FC, Akerlund T, Gerding DN, Goering RV, Bostrom T, et al. Comparison of strain typing results for Clostridium difficile isolates from North America. J Clin Microbiol. 2011;49: 1831–1837. pmid:21389155
- 10. Rupnik M, Avesani V, Janc M, von Eichel-Streiber C, Delmee M. A novel toxinotyping scheme and correlation of toxinotypes with serogroups of Clostridium difficile isolates. J Clin Microbiol. 1998;36: 2240–2247. pmid:9665999
- 11. Fawley WN, Parnell P, Verity P, Freeman J, Wilcox MH. Molecular epidemiology of endemic Clostridium difficile infection and the significance of subtypes of the United Kingdom epidemic strain (PCR ribotype 1). J Clin Microbiol. 2005;43: 2685–2696. pmid:15956384
- 12. Chachaty E, Saulnier P, Martin A, Mario N, Andremont A. Comparison of ribotyping, pulsed-field gel electrophoresis and random amplified polymorphic DNA for typing Clostridium difficile strains. FEMS Microbiol Lett. 1994;122: 61–68. pmid:7958778
- 13. Eckert C, Van Broeck J, Spigaglia P, Burghoffer B, Delmee M, et al. Comparison of a commercially available repetitive-element PCR system (DiversiLab) with PCR ribotyping for typing of Clostridium difficile strains. J Clin Microbiol. 2011;49: 3352–3354. pmid:21775548
- 14. Killgore G, Thompson A, Johnson S, Brazier J, Kuijper E, et al. Comparison of seven techniques for typing international epidemic strains of Clostridium difficile: restriction endonuclease analysis pulsed-field gel electrophoresis, PCR—ribotyping, multilocus sequence typing, multilocus variable-number tandem-repeat analysis, amplified fragment length polymorphism and surface layer protein A gene sequence typing. J Clin Microbiol. 2008;46: 431–437. pmid:18039796
- 15. O’Neill GL, Ogunsola FT, Brazier JS, Duerden BI. Modification of a PCR-ribotyping method for application as a routine typing scheme for Clostridium difficile. Anaerobe. 1996;2: 205–209. pmid:9160624
- 16. Stubbs SL, Brazier JS, O’Neill GL, Duerden BI. PCR targeted to the 16S-23S rRNA gene intergenic spacer region of Clostridium difficile and construction of a library consisting of 116 different PCR ribotypes. J Clin Microbiol. 1999;37: 461–463. pmid:9889244
- 17. Bidet P, Barbut F, Lalande V, Burghoffer B, Petit JC. Development of a new PCR-ribotyping method for Clostridium difficile based on ribosomal RNA gene sequencing. FEMS Microbiol Lett. 1999;175: 261–266. pmid:10386377
- 18. Indra A, Huhulescu S, Schneeweis M, Hasenberger P, Kernbichler S, et al. Characterisation of Clostridium difficile isolates using capillary gel electrophoresis-based PCR ribotyping. J Med Microbiol. 2008;57: 1377–1382. pmid:18927415
- 19. Knetsch CW, Lawley TD, Hensgens MP, Corver J, Wilcox MH, et al. Current application and future perspectives of molecular typing methods to study Clostridium difficile infections. Euro Surveill. 2013;18: 20381. pmid:23369393
- 20. Xiao M, Kong F, Jin P, Wang Q, Xiao K, et al. Comparison of two capillary gel electrophoresis systems for Clostridium difficile ribotyping, using a panel of ribotype 027 isolates and whole-genome sequences as a reference standard. J Clin Microbiol. 2012;50: 2755–2760. pmid:22692737
- 21. Kurka H, Ehrenreich A, Ludwig W, Monot M, Rupnik M, et al. Sequence similarity of Clostridium difficile strains by analysis of conserved genes and genome content is reflected by their ribotype affiliation. Plos One. 2014;9: e86535. pmid:24482682
- 22. Knetsch CW, Terveer EM, Lauber C, Gorbalenya AE, Harmanus C, et al. Comparative analysis of an expanded Clostridium difficile reference strain collection reveals genetic diversity and evolution through six lineages. Infect Genet Evol. 2012;12: 1577–1585. pmid:22705462
- 23. European Clostridium difficile infection study network (ECDIS-NET). Europe: Supporting capacity building for surveillance of Clostridium difficile. Leiden: ECDIS-NET. [Accessed 1 Jul 2012]. Available: http://www.ecdisnet.eu/.
- 24. Walk ST, Micic D, Jain R, Lo ES, Trivedi I, et al. Clostridium difficile ribotype does not predict severe infection. Clin Infect Dis. 2012;55: 1661–1668. pmid:22972866
- 25. Behroozian AA, Chludzinski JP, Lo ES, Ewing SA, Waslawski S, et al. Detection of mixed populations of Clostridium difficile from symptomatic patients using capillary-based polymerase chain reaction ribotyping. Inf Con Hosp Epidemiol. 2013;34: 961–966. pmid:23917911
- 26. Sadeghifard N, Gurtler V, Beer M, Seviour RJ. The mosaic nature of intergenic 16S–23S rRNA spacer regions suggests rRNA operon copy number variation in Clostridium difficile strains. Appl Environ Microbiol. 2006;72: 7311–7323. pmid:16980415
- 27. Valiente E, Dawson LF, Cairns MD, Stabler RA, Wren BW. Emergence of new PCR ribotypes from the hypervirulent Clostridium difficile 027 lineage. J. Med. Microbiol. 2012;61: 49–56. pmid:21903827
- 28. Griffiths D, Fawley W, Kachrimanidou M, Bowden R, Crook DW, et al. Multilocus sequence typing of Clostridium difficile. J Clin Microbiol. 2010;48: 770–778. pmid:20042623
- 29. Huber CA, Foster NF, Riley TV, Paterson DL. Challenges for standardization of Clostridium difficile typing methods. J Clin Microbiol. 2013;51: 2810–2814. pmid:23784128