29 Sep 2010: Snellman G, Melhus H, Gedeborg R, Byberg L, Berglund L, et al. (2010) Correction: Determining Vitamin D Status: A Comparison between Commercially Available Assays. PLOS ONE 5(9): 10.1371/annotation/23307aa4-726e-4f11-86c0-8a292be33517. https://doi.org/10.1371/annotation/23307aa4-726e-4f11-86c0-8a292be33517 View correction
Vitamin D is not only important for bone health but can also affect the development of several non-bone diseases. The definition of vitamin D insufficiency by serum levels of 25-hydroxyvitamin D depends on the clinical outcome but might also be a consequence of analytical methods used for the definition. Although numerous 25-hydroxyvitamin D assays are available, their comparability is uncertain. We therefore aim to investigate the precision, accuracy and clinical consequences of differences in performance between three common commercially available assays.
Serum 25-hydroxyvitamin D levels from 204 twins from the Swedish Twin Registry were determined with high-pressure liquid chromatography-atmospheric pressure chemical ionization-mass spectrometry (HPLC-APCI-MS), a radioimmunoassay (RIA) and a chemiluminescent immunoassay (CLIA). High inter-assay disagreement was found. Mean 25-hydroxyvitamin D levels were highest for the HPLC-APCI-MS technique (85 nmol/L, 95% CI 81–89), intermediate for RIA (70 nmol/L, 95% CI 66–74) and lowest with CLIA (60 nmol/L, 95% CI 56–64). Using a 50-nmol/L cut-off, 8% of the subjects were insufficient using HPLC-APCI-MS, 22% with RIA and 43% by CLIA. Because of the heritable component of 25-hydroxyvitamin D status, the accuracy of each method could indirectly be assessed by comparison of within-twin pair correlations. The strongest correlation was found for HPLC-APCI-MS (r = 0.7), intermediate for RIA (r = 0.5) and lowest for CLIA (r = 0.4). Regression analyses between the methods revealed a non-uniform variance (p<0.0001) depending on level of 25-hydroxyvitamin D.
Citation: Snellman G, Melhus H, Gedeborg R, Byberg L, Berglund L, Wernroth L, et al. (2010) Determining Vitamin D Status: A Comparison between Commercially Available Assays. PLoS ONE 5(7): e11555. https://doi.org/10.1371/journal.pone.0011555
Editor: Joel Joseph Gagnier, University of Michigan, Canada
Received: January 27, 2010; Accepted: May 30, 2010; Published: July 13, 2010
Copyright: © 2010 Snellman et al. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Funding: This work was supported by grants from the Swedish Research Council. The funders had no role in study design, data collection and analysis, decision to publish, or preparation of the manuscript.
Competing interests: The authors have declared that no competing interests exist.
Vitamin D deficiency is not only associated with osteoporosis and osteomalacia ,  but can also contribute to decreased muscle strength , cancers , cardiovascular disease , type 1 diabetes mellitus  and overall mortality . Defined risk groups are elderly, dark-skinned and obese, as well as inhabitants in northern latitudes where UV B radiation is undetectable during winter , , , . These findings have increased the need for determining vitamin D status in a reliable way.
Vitamin D exists in two forms, namely ergocalciferol (vitamin D2), and cholecalciferol (vitamin D3). Vitamin D status is assessed by measuring serum levels of 25-hydroxyvitamin D. Fatty fish and dairy products are the main dietary sources of vitamin D3 , . The most important source of circulating 25-hydroxyvitamin D3 is, with sufficient solar exposure, the endogenous dermal production of pre-vitamin D3 after exposure to UV B radiation . The less dominant serum 25-hydroxyvitamin D2 is mainly derived from plant foods and in some countries from supplements.
There are many commercially available 25-hydroxyvitamin D assays used for determination of vitamin D status. These include high-pressure liquid chromatography (HPLC) and mass spectrometry (MS) , radioimmunoassays (RIA), enzyme immunoassays (EIA), competitive protein binding assays (CPBA), automated chemiluminescence protein-binding assays (CLPBA) and chemiluminescent immunoassays (CLIA). All these assays are used in both clinical and research settings but it is not widely appreciated that 25-hydroxyvitamin D assays may yield discrepant results. Inter-assay and laboratory disagreement could contribute to uncertainty when comparing results from studies investigating the prevalence or clinical consequence of vitamin D insufficiency. Indeed, several studies including reports from DEQAS, an organization who aim to ensure the analytical reliability of 25-hydroxyvitamin D assays , have indicated high variability between different assays as well as inter-laboratory disagreement, but these studies have been limited by few participants, a non-population based setting, only partially overlapping analyses of the samples included, and consensus opinion regarding accuracy rather than an unbiased comparator , , , , , , .
25-hydroxyvitamin D levels are partially genetically determined , , . Higher twin resemblance in serum 25-hydroxyvitamin D values indicates enhanced performance of the assay. This fact, thus, enables that the within-pair correlation to be used as an unbiased proxy measure of accuracy. Determination of accuracy has not been possible in previous validation studies , , , , , . In these studies, it has also not been evaluated if the differences in assay results have been uniform or have been dependent on serum 25-hydroxyvitamin D level. Cross-calibration between assays is dependent on the pattern of assay differences. We therefore aimed in a twin study to compare the differences in performance between three common commercially available methods for 25-hydroxyvitamin D analysis with different methodological principles: a combined HPLC-MS method, a RIA and a CLIA assay.
Materials and Methods
The study protocol was approved by the Ethics Committee of Uppsala University and all participants gave written informed consent to participate in the study and to donate blood samples.
Subjects were recruited from The Swedish Twin Registry. All intact like-sexed twin pairs, born 1965 or earlier and living in the county of Uppsala were invited to participate. Uppsala County is located in central Sweden at northern latitude 60°. Totally, 172 twin pairs were found eligible and invited to participate. Of these, 102 twin pairs, i.e. 204 subjects, accepted to participate in the study. No subjects were excluded. Zygosity information in the Swedish twin registry has a high validity . The study included 59 female and 43 male Caucasian twin pairs with an age range between 39 and 85 years. Because of low UV B radiation, vitamin D cannot be synthesized in the skin between late autumn through April at high latitudes . We therefore defined winter as November throughout April and summer as May throughout October. The serum samples were collected during the winter season for 28 twin pairs and during the summer season for 74 pairs. When possible, both members of each pair were examined within the same week to take the individual seasonal variation in vitamin D levels into account. Sixty-one pairs were examined the same day. The median within pair difference in days between the examinations among the remainder of the pairs was 6 days (inter quartile range 2 to 9 days), with a maximum of 17 days. Seven twins reported use of vitamin D supplements.
The study protocol was approved by the local ethical committee of Uppsala and all participants gave written informed consent to participate in the study and to donate blood samples.
Venous blood samples were collected after a 12-h overnight fasting, protected from light, centrifuged and stored at −80°C until analysis. All samples were analyzed in three laboratories using three techniques (described in detail below).
High-pressure liquid chromatography (HPLC) - atmospheric pressure chemical ionisation (APCI) - mass spectrometry (MS).
Determination of 25-hydroxyvitamin D2 and D3 in plasma with HPLC-APCI-MS was done at Vitas, Oslo, Norway. Deuterium labelled 25-hydroxyvitamin D2 and D3 were used for internal standards. One hundred and fifty µL of human plasma were diluted with 450 µL 2-propanol containing BHT (butylhydroxytoluene) as an antioxidant. After thorough mixing (15 min) and centrifugation (10 min, 4000 g at 10°C), an aliquot of 35 µL was injected from the supernatant into the HPLC system. HPLC was performed with a HP 1100 liquid chromatograph (Agilent Technologies, Palo Alto CA, USA) interfaced by atmospheric pressure chemical ionization (APCI) to a HP mass spectrometric detector (MS) operated in single-ion monitoring mode (SIM). 25-hydroxyvitamin D2 and D3 were separated on a 4.6 mm ×50 mm reversed phase column with 1.8 µM particles. The column temperature was 80°C. A two-point calibration curve was made from analysis of albumin solution enriched with known vitamin D concentration. Recovery is 95%; the method is linear from 5–400 nmol/L and the limit of detection is 1–4 nmol/L. The Coefficients of Variation (CV) for inter-assay analyses are 7.6% at 25-hydroxyvitamin D of 47.8 nmol/L and 6.9% at 25-hydroxyvitamin D of 83.0 nM. The assay is accredited by the Vitamin D External Quality Assessment Scheme (DEQAS) .
25-hydroxyvitamin D2 and D3 in serum were measured at a research laboratory in Uppsala using Gamma-B 25-hydroxyvitamin D RIA (IDS, Boldon, UK). The CV for inter-assay analyses is 7.9%. Sensitivity, defined as the concentration corresponding to the mean minus 2 standard deviations of 10 replicates of the zero calibrator, is <3 nmol/L. To ascertain analytic quality all standards, controls and samples were analyzed in duplicate and all duplicates with a coefficient of variation >10% were reanalyzed. The control samples provided by the manufacturer were within the recommended range.
Chemiluminescent immunoassay (CLIA).
25-hydroxyvitamin D2 and D3 in serum were measured as a standard procedure at the department of Clinical Chemistry at Uppsala University Hospital. The LIAISON® 25-hydroxyvitamin D Assay (DiaSorin) uses chemiluminescent immunoassay technology. Specific antibody to vitamin D is used for coating magnetic particles (solid phase) and vitamin D is linked to an isoluminol derivative. During the incubation, 25-hydroxyvitamin D is dissociated from its binding protein and competes with labelled vitamin D for binding sites on the antibody. After the incubation, the unbound material is removed with a wash cycle. Subsequently, the starter reagents are added and a flash chemiluminescent reaction is initiated. The light signal is measured by a photomultiplier as relative light units and is inversely proportional to the concentration of 25-hydroxyvitamin D present in samples. CV for inter-assay analyses is 18.4% at a 25-hydroxyvitamin D level of 39.5 nmol/L and 11.7% at 121.25 nmol/L. The quality of the method is evaluated using the Vitamin D External Quality Assessment Scheme (DEQAS)  that is, based on blinded samples with varying concentrations of 25(OH)D, sent out as a within- and between-method comparison to over 500 participating laboratories, the assay results agree within ±30% of All Laboratory Trimmed Mean (ALTM) .
The 25-hydroxyvitamin D values were normally distributed with Shapiro-Wilk's w value greater than 0.95. Mean serum 25-hydroxyvitamin D values with 95% confidence intervals (CI) for each assay results were calculated including stratification by season. The cumulative proportion of twins at each serum level by method was plotted in order to compare the proportion of twins with values below each 25-hydroxyvitamin D level, specifically the 50 nmol/L insufficiency level proposed by expert opinion , . To estimate the accuracy of each method, resemblance in results within twin pairs was calculated by intraclass correlation coefficients (ICCs). The sample coefficient of variation (SCV) was calculated by dividing the standard deviation for each method by its mean value. Taking into account the twinship dependence, bootstrap-estimated 95% confidence intervals of ICCs and SCVs, and p values for assay differences, were obtained by re-sampling the total sample size 10,000 times. Additionally, results of assays were compared using Bland-Altman plots , . The difference in performance by level of serum 25-hydroxyvitamin D was formally tested by linear regression analysis of the difference in absolute values between two methods regressed against the mean of the method results of the two analyses, also taking into account the twin-ship dependence and zygosity. A p-value <0.05 was considered significant.
Our results reveal low inter-assay agreement. Mean 25-hydroxyvitamin D values and basic characteristics of the twins are presented in Table 1. HPLC-APCI-MS measured a mean 25-hydroxyvitamin D level of 85 nmol/L, RIA 70 nmol/L and CLIA 60 nmol/L, p for difference between assays <0.0001 (Figure 1). 25-hydroxyvitamin D2 was detectable in only 20% (n = 40) of our subjects using the HPLC-APCI-MS assay. The mean level among these 40 twins was 8 nmol/L, and they contributed to only 1.5 nmol/L of the mean 25-hydroxyvitamin D level among all 204 twins. Demonstrated in Figure 2, the serum 25-hydroxyvitamin D levels were significantly higher with all methods during the summer compared to the winter season. The greatest inter-seasonal difference, 23 nmol/L (95% CI 13–33), is presented by the HPLC-APCI-MS assay.
The error bars indicate 95% confidence intervals.
The error bars indicate 95% confidence intervals.
There were considerable differences between the methods in the proportion of participants classified as vitamin D insufficient. Using a 50-nmol/L cut-off, only 8% of our subjects were classified as vitamin D insufficient with the HPLC-APCI-MS method, 22% using RIA and 43% with the CLIA method (Figure 3).
HPLC-APCI-MS 8%, RIA 22%, CLIA 43%.
As measures of assay accuracy, intraclass correlation coefficients (ICC) for within twin pair similarity in 25-hydroxyvitamin D levels, and 95% CI are presented in Table 2. HPLC-APCI-MS had a significantly higher ICC relative to both RIA and CLIA, and RIA had had a higher value than CLIA. The precision of the assays was determined by sample coefficient of variation (SCV). HPLC-APCI-MS had a SCV of 32%, RIA 34% and CLIA 44% (Table 3).
According to the Bland-Altman plots, RIA and CLIA had a non-proportional bias relative to HPLC-APCI-MS (Figure 4, panel A–C). Both positive and negative bias became more accentuated with increasing 25-hydroxyvitamin D value, i.e. the inter-assay disagreement increases with an increasing serum level of 25-hydroxyvitamin D. To formally test that there was a non-uniform variability at different serum levels of 25-hydroxyvitamin D between the methods we used linear regression to analyze the relation between absolute differences in serum values between the methods against the mean of the two values (Table 4). Both parameter estimates were positive and highly statistically significantly different from zero, and accordingly, the differences in variation between the methods were higher at increasing levels of serum 25-hydroxyvitamin D.
HPLC-APCI-MS vs. RIA (panel A), HPLC-APCI-MS vs. CLIA (panel B) and RIA vs. CLIA (panel C). Each circle represents one twin.
We observed high variability between the HPLC-APCI-MS, RIA and CLIA assay results, emphasizing that a gold standard for the 25-hydroxyvitamin D assay is needed to establish consensus on the required level for sufficient vitamin D status. Highest accuracy was found with the HPLC-APCI-MS and lowest with the CLIA method.
Previous studies support our findings, reporting variability between different assays as well as inter-laboratory differences using the same assay , , , , . Lips et al  analyzed samples from a selected population of vitamin D supplement users (n = 104) with three different methods. The mean 25-hydroxyvitamin D level was 80% higher when using a competitive protein-binding assay as compared with HPLC while intermediate values were found with a RIA assay. The accuracy of the methods was not possible to evaluate, however. Binkley et al reported 18% and 90% insufficiency proportions in two similar populations (n = 20 and 42, respectively) using two RIA assays. IDS and DiaSorin-RIA as well as a Nichols Advantage automated protein binding assay detected less than 50% of the changes in 25-hydroxyvitamin D2 detected by HPLC according to Glendenning et al .
HPLC can discriminate 25-hydroxyvitamin D2 and D3 metabolites, whereas our RIA and CLIA method measure total 25-hydroxyvitamin D levels, i.e., the sum of 25-hydroxyvitamin D2 and D3 metabolites. In some countries, including the USA, vitamin D2 has been the only form of vitamin D available for prescription, even though both vitamin D2 and vitamin D3 are used as non-prescribed supplements, while in Europe vitamin D3 is dominating . HPLC could therefore have an advantage when evaluating the effect of supplementation with D2. Nevertheless, this is of minor importance in our setting since only 20% of our participants have measurable 25-hydroxyvitamin D2 values, contributing on average to only 2% higher total 25-hydroxyvitamin D levels in the cohort. Similar low values were found by Högström et al  in young, healthy Swedish men. Thus, in European countries it may not be of major clinical importance that some assays underestimate D2 (although not relevant to our study) or cannot separate between D2 and D3.
There is no well defined and international accepted definition of optimal serum 25-hydroxyvitamin D value for bone and nutritional health , , but low plasma 25-hydroxyvitamin D and secondary hyperparathyroidism are the biochemical hallmarks for insufficient vitamin D status . A common definition of vitamin D insufficiency is 25-hydroxyvitamin D <50 nmol/L although many authors suggest that clinicians should aim at higher levels , , . Our great inter-assay differences in insufficiency proportion, as well as in accuracy and in precision indicate that it is questionable to rely on immunoassays when determining whether a patient is insufficient. Some assays may be too imprecise for both clinical and research use. Assay-specific decision limits to define appropriate thresholds for insufficiency have been suggested as a solution  but that will be cumbersome for the clinician. Furthermore, our results suggest that a simple reliable calibration between assays cannot be accomplished.
Assay disagreement and inter-laboratory variation naturally have important clinical implications. It may hamper comparison of studies from diverse populations and countries, exemplified by Lips . This may be the reason why there is inconsistent evidence regarding the degree of association between vitamin D status, bone mineral density and the risk of low energy fractures .
Because the HPLC-APCI-MS method provides the most prominent difference between summer and winter levels of 25-hydroxyvitamin D, the lowest SCV value and the best twin resemblance in serum levels, our conclusion is that HPLC-APCI-MS is a more accurate and reliable method than both RIA and CLIA.
The advantages of our study are the comparable large sample size, the population-based design, that our participants comprised of twins and that all the samples are analyzed with all three assays. One limitation is that other available methods for 25-hydroxyvitamin D measurement were not evaluated. Moreover, with analyses performed at three laboratories, such a design could limit the ability to separate assay-specific from laboratory-specific bias. Differences between laboratories may be caused by unfamiliarity with the analytical method, but all laboratories in our study were experienced with the method used at that site and all methods are used for clinical decision making. Moreover, Vitas is an approved R&D institution in the SkatteFUNN scheme held by the Norwegian Research Council and is the national reference laboratory for vitamin D analysis. The department of Clinical Chemistry at the Uppsala University Hospital is accredited by the Swedish Board for Accreditation and Conformity Assessment (SWEDAC). SWEDAC must in its turn fulfil certain requirements that are set out in ISO/IEC 17011. Compliance is confirmed by the international assessments that are performed within the framework of the European Accreditation cooperation (EA).
We conclude that a single threshold value to define an optimal 25-hydroxyvitamin D level is presently impossible to determine because of differences in assay results.
Conceived and designed the experiments: GS HM RG LB LB LW KM. Performed the experiments: LB LB LW KM. Analyzed the data: GS HM RG LB LB LW KM. Contributed reagents/materials/analysis tools: HM KM. Wrote the paper: GS HM RG LB KM.
- 1. Bischoff-Ferrari HA, Dietrich T, Orav EJ, Dawson-Hughes B (2004) Positive association between 25-hydroxy vitamin D levels and bone mineral density: a population-based study of younger and older adults. Am J Med 116: 634–639.
- 2. Lips P (2001) Vitamin D deficiency and secondary hyperparathyroidism in the elderly: consequences for bone loss and fractures and therapeutic implications. Endocr Rev 22: 477–501.
- 3. Bischoff-Ferrari HA, Dietrich T, Orav EJ, Hu FB, Zhang Y, et al. (2004) Higher 25-hydroxyvitamin D concentrations are associated with better lower-extremity function in both active and inactive persons aged > or = 60 y. Am J Clin Nutr 80: 752–758.
- 4. Holick MF (2004) Vitamin D: importance in the prevention of cancers, type 1 diabetes, heart disease, and osteoporosis. Am J Clin Nutr 79: 362–371.
- 5. Forman JP, Bischoff-Ferrari HA, Willett WC, Stampfer MJ, Curhan GC (2005) Vitamin D intake and risk of incident hypertension: results from three large prospective cohort studies. Hypertension 46: 676–682.
- 6. Hypponen E, Laara E, Reunanen A, Jarvelin MR, Virtanen SM (2001) Intake of vitamin D and risk of type 1 diabetes: a birth-cohort study. Lancet 358: 1500–1503.
- 7. Melamed ML, Michos ED, Post W, Astor B (2008) 25-hydroxyvitamin D levels and the risk of mortality in the general population. Arch Intern Med 168: 1629–1637.
- 8. Holick MF (1995) Environmental factors that influence the cutaneous production of vitamin D. Am J Clin Nutr 61: 638S–645S.
- 9. Wortsman J, Matsuoka LY, Chen TC, Lu Z, Holick MF (2000) Decreased bioavailability of vitamin D in obesity. Am J Clin Nutr 72: 690–693.
- 10. MacLaughlin J, Holick MF (1985) Aging decreases the capacity of human skin to produce vitamin D3. J Clin Invest 76: 1536–1538.
- 11. Clemens TL, Adams JS, Henderson SL, Holick MF (1982) Increased skin pigment reduces the capacity of skin to synthesise vitamin D3. Lancet 1: 74–76.
- 12. Burgaz A, Akesson A, Oster A, Michaelsson K, Wolk A (2007) Associations of diet, supplement use, and ultraviolet B radiation exposure with vitamin D status in Swedish women during winter. Am J Clin Nutr 86: 1399–1404.
- 13. Macdonald HM, Mavroeidi A, Barr RJ, Black AJ, Fraser WD, et al. (2008) Vitamin D status in postmenopausal women living at higher latitudes in the UK in relation to bone health, overweight, sunlight exposure and dietary vitamin D. Bone 42: 996–1003.
- 14. Tsugawa N, Suhara Y, Kamao M, Okano T (2005) Determination of 25-hydroxyvitamin D in human plasma using high-performance liquid chromatography—tandem mass spectrometry. Anal Chem 77: 3001–3007.
- 15. DEQAS (Vitamin D External Quality Assessment Scheme) Availible at: http://www.deqas.org/.
- 16. Binkley N, Krueger D, Cowgill CS, Plum L, Lake E, et al. (2004) Assay variation confounds the diagnosis of hypovitaminosis D: a call for standardization. J Clin Endocrinol Metab 89: 3152–3157.
- 17. Binkley N, Krueger D, Gemar D, Drezner MK (2008) Correlation among 25-hydroxy-vitamin D assays. J Clin Endocrinol Metab 93: 1804–1808.
- 18. Carter GD, Carter R, Jones J, Berry J (2004) How accurate are assays for 25-hydroxyvitamin D? Data from the international vitamin D external quality assessment scheme. Clin Chem 50: 2195–2197.
- 19. Glendenning P, Taranto M, Noble JM, Musk AA, Hammond C, et al. (2006) Current assays overestimate 25-hydroxyvitamin D3 and underestimate 25-hydroxyvitamin D2 compared with HPLC: need for assay-specific decision limits and metabolite-specific assays. Ann Clin Biochem 43: 23–30.
- 20. Lensmeyer GL, Wiebe DA, Binkley N, Drezner MK (2006) HPLC method for 25-hydroxyvitamin D measurement: comparison with contemporary assays. Clin Chem 52: 1120–1126.
- 21. Lips P, Chapuy MC, Dawson-Hughes B, Pols HA, Holick MF (1999) An international comparison of serum 25-hydroxyvitamin D measurements. Osteoporos Int 9: 394–397.
- 22. Carter GD (2009) 25-Hydroxyvitamin D assays: the quest for accuracy. Clin Chem 55: 1300–1302.
- 23. Hunter D, De Lange M, Snieder H, MacGregor AJ, Swaminathan R, et al. (2001) Genetic contribution to bone metabolism, calcium excretion, and vitamin D and parathyroid hormone regulation. J Bone Miner Res 16: 371–378.
- 24. Shea MK, Benjamin EJ, Dupuis J, Massaro JM, Jacques PF, et al. (2009) Genetic and non-genetic correlates of vitamins K and D. Eur J Clin Nutr 63: 458–464.
- 25. Snellman G, Melhus H, Gedeborg R, Olofsson S, Wolk A, et al. (2009) Seasonal genetic influence on serum 25-hydroxyvitamin D levels: a twin study. PLoS One 4: e7747.
- 26. Lichtenstein P, De Faire U, Floderus B, Svartengren M, Svedberg P, et al. (2002) The Swedish Twin Registry: a unique resource for clinical, epidemiological and genetic studies. J Intern Med 252: 184–205.
- 27. Persson T (2000) Measurements of solar radiation in Sweden 1983–1998. SHMI reports Meteorology and climaology 89:
- 28. Dawson-Hughes B, Heaney RP, Holick MF, Lips P, Meunier PJ, et al. (2005) Estimates of optimal vitamin D status. Osteoporos Int 16: 713–716.
- 29. Lips P (2004) Which circulating level of 25-hydroxyvitamin D is appropriate? J Steroid Biochem Mol Biol 89-90: 611–614.
- 30. Bland JM, Altman DG (1986) Statistical methods for assessing agreement between two methods of clinical measurement. Lancet 1: 307–310.
- 31. Bland JM, Altman DG (1999) Measuring agreement in method comparison studies. Stat Methods Med Res 8: 135–160.
- 32. Holick MF (2005) The vitamin D epidemic and its health consequences. J Nutr 135: 2739S–2748S.
- 33. Hogstrom M, Nordstrom A, Nordstrom P (2006) Relationship between vitamin D metabolites and bone mineral density in young males: a cross-sectional and longitudinal study. Calcif Tissue Int 79: 95–101.
- 34. Chapuy MC, Preziosi P, Maamer M, Arnaud S, Galan P, et al. (1997) Prevalence of vitamin D insufficiency in an adult normal population. Osteoporos Int 7: 439–443.
- 35. Cranney A, Horsley T, O'Donnell S, Weiler H, Puil L, et al. (2007) Effectiveness and safety of vitamin D in relation to bone health. Evid Rep Technol Assess (Full Rep) 1–235.