Diffusion tensor imaging (DTI) has recently started to be adopted into clinical investigations of spinal cord (SC) diseases. However, DTI applications to the lower SC are limited due to a number of technical challenges, related mainly to the even smaller size of the SC structure at this level, its position relative to the receiver coil elements and the effects of motion during data acquisition. Developing methods to overcome these problems would offer new means to gain further insights into microstructural changes of neurological conditions involving the lower SC, and in turn could help explain symptoms such as bladder and sexual dysfunction. In this work, the feasibility of obtaining grey and white matter (GM/WM) DTI indices such as axial/radial/mean diffusivity (AD/RD/MD) and fractional anisotropy (FA) within the lumbosacral enlargement (LSE) was investigated using a reduced field-of-view (rFOV) single-shot echo-planar imaging (ss-EPI) acquisition in 14 healthy participants using a clinical 3T MR system. The scan-rescan reproducibility of the measurements was assessed by calculating the percentage coefficient of variation (%COV). Mean FA was higher in WM compared to GM (0.58 and 0.4 in WM and GM respectively), AD and MD were higher in WM compared to GM (1.66 μm2ms-1 and 0.94 μm2ms-1 in WM and 1.2 μm2ms-1 and 0.82 μm2ms-1 in GM for AD and MD respectively) and RD was lower in WM compared to GM (0.58 μm2ms-1 and 0.63 μm2ms-1 respectively). The scan-rescan %COV was lower than 10% in all cases with the highest values observed for FA and the lowest for MD. This pilot study demonstrates that it is possible to obtain reliable tissue-specific estimation of DTI indices within the LSE using a rFOV ss-EPI acquisition. The DTI acquisition and analysis protocol presented here is clinically feasible and may be used in future investigations of neurological conditions implicating the lower SC.
Citation: Yiannakas MC, Grussu F, Louka P, Prados F, Samson RS, Battiston M, et al. (2016) Reduced Field-of-View Diffusion-Weighted Imaging of the Lumbosacral Enlargement: A Pilot In Vivo Study of the Healthy Spinal Cord at 3T. PLoS ONE 11(10): e0164890. doi:10.1371/journal.pone.0164890
Editor: Gernot Reishofer, Medical University of Graz, AUSTRIA
Received: June 28, 2016; Accepted: October 3, 2016; Published: October 14, 2016
Copyright: © 2016 Yiannakas et al. This is an open access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Data Availability: All relevant data are within the paper and its Supporting Information files.
Funding: This work and the NMR Research Unit are supported by the U.K Multiple Sclerosis Society (http://www.mssociety.org.uk/). This work was supported by the National Institute for Health Research University College London Hospitals Biomedical Research Centre. FP is funded by NIHR BRC UCLH/UCL High Impact Initiative. SO receives funding from the EPSRC (EP/H046410/1, EP/J020990/1, EP/K005278), the MRC (MR/J01107X/1), the NIHR Biomedical Research Unit (Dementia) at UCL and the NIHR BRC UCLH/UCL.
Competing interests: The authors have declared that no competing interests exist.
Diffusion-weighted imaging (DWI) can be used to study the diffusion of water molecules in neural tissue . Among several DWI methods, diffusion tensor imaging (DTI)  maps the three-dimensional displacement profile of water molecules due to diffusion, and provides measures that reflect tissue microstructure beyond conventional qualitative magnetic resonance imaging (MRI) assessment. DTI has been used successfully over the years to study neurological conditions affecting the brain [3–5], however the use of DTI to study the spinal cord (SC) has been hampered by a number of technical challenges related mainly to the small cross-sectional size of the structure, susceptibility artifacts introduced by the surrounding vertebral bones and the effect of physiological and involuntary motion during the imaging studies [6, 7].
Technical concerns associated with SC diffusion imaging are related but not limited to the signal-to-noise ratio (SNR), radiofrequency (RF) hardware design, magnetic field inhomogeneities around the SC, cerebrospinal fluid (CSF) pulsation, partial volume effects at the boundary between grey matter (GM), white matter (WM) and CSF and the choice of the acquisition scheme (e.g. type of pulse sequence and readout, number of diffusion directions). Typically, DTI is performed using single-shot echo-planar imaging (ss-EPI) ; ss-EPI reduces the sensitivity to subject motion, but causes phase errors to accumulate due to long readout durations. As a consequence, image distortions become evident in the presence of magnetic susceptibility inhomogeneity. This is of particular concern in SC imaging, where tissue susceptibility differences exist in close proximity to the SC, especially due to the vertebral bones. In order to reduce the readout duration and associated off-resonance related image distortions, previous studies included the use of parallel imaging [9, 10], or alternative readout techniques such as interleaved EPI , turbo spin echo  and line scan imaging [13, 14]. More recently, a number of reduced field-of-view (rFOV) approaches have been proposed for SC imaging [15–21], which offer a significant reduction in the anteroposterior dimension of the imaging volume, thus reducing the readout duration considerably.
DTI with rFOV has been employed successfully to study the upper SC with sufficiently high resolution to distinguish between GM and WM and to allow tissue- or tract-specific estimation of indices such as mean/axial/radial diffusivity (MD/AD/RD) and fractional anisotropy (FA) in the neurologically intact cord [22, 23] and in neurological conditions such as multiple sclerosis (MS) and neuromyelitis optica [24, 25]. However, similar applications in the lower SC are limited, with only a few reports to date [26, 27]. Reliable DTI acquisition and analysis protocols to study the lower SC could provide new insights into the pathophysiology of symptoms such as bladder and sexual dysfunction, which are often associated with neurological conditions such as MS [28, 29], SC injury (SCI) [30, 31] and multiple system atrophy (MSA) [32, 33].
In this pilot study, the feasibility of obtaining tissue-specific (i.e. GM and WM) DTI indices within the lower SC was investigated in a number of healthy volunteers using DTI with rFOV [15, 26] on a clinical 3T MRI system. The DTI acquisition and analysis protocol presented here addresses key methodological considerations such as: i) the positional variation of the lower SC through the use of the lumbosacral enlargement (LSE) as previously suggested ; ii) the scan-rescan reproducibility of the DTI indices; iii) the influence of the diffusion encoding protocol on the quality and reproducibility of the DTI indices.
Materials and Methods
Fourteen healthy subjects were recruited (6 male and 8 female, mean age 27.3 years; range 21–46 years). The work was approved by the National Hospital for Neurology and Neurosurgery and the Institute of Neurology Joint Research and NRES committee London Bloomsbury (London REC2 Ethics Committee). Written informed consent was obtained from all study participants.
A 3T Philips Achieva system with maximum gradient strength 65 mT m-1, radiofrequency (RF) dual-transmit technology (Philips Healthcare, Best, Netherlands) and the manufacturer’s product 15-channel SENSE receive-only RF spine coil were used. A conventional T2-weighted image of the lumbar spine in the sagittal plane was first obtained using a turbo spin-echo (TSE) sequence and was used to facilitate prescription of subsequent scans perpendicular to the cord. The imaging parameters for the T2-weighted TSE were: TR = 3575 ms, TE = 100 ms, flip angle α = 90°, FOV = 300 × 180 mm2, voxel size = 0.8 × 0.8 × 3 mm3, NEX = 2, slices = 15, acquisition time 3:48 min.
Additionally, a 3D slab-selective fast field-echo (3D-FFE) sequence with fat suppression was acquired in the axial-oblique plane (i.e. perpendicular to the longitudinal axis of the cord). The 3D-FFE scan was performed in order to identify the exact location of the LSE, necessary to obtain quantitative measurements of DTI indices. The 3D-FFE slices were prescribed between the T11—L1 level to ensure coverage of the LSE in all cases . The following parameters were used: TR = 22 ms, TE = 4.4 ms, flip angle α = 10°, FOV = 180 × 180 mm2, voxel size = 0.5 × 0.5 × 5 mm3, NEX = 8, slices = 10, acquisition time 9:51 min.
For estimating the DTI indices within the LSE, a ss-EPI with rFOV was used [15, 26] with cardiac gating and identical slice geometry to the 3D-FFE. Diffusion-weighting was applied along 60 diffusion directions with b = 1000 s/mm2, interleaved with 7 b = 0 measurements; Additional parameters for the DTI acquisition were: TR ~ 4000 ms (4 heart beats), TE = 40 ms, flip angle α = 90°, slices = 10; FOV = 64 x 48 mm2, voxel size = 1 × 1 × 5 mm3, trigger delay for cardiac gating of 150ms. The total acquisition time was ~ 15 min, depending on heart rate.
Motion during the imaging session was minimised with the use of Velcro straps to restrain the torso and with the placement of a large foam wedge beneath the knees, as this increases the level of contact between the lower back and the flat surface of the coil.
Using the 3D-FFE, the three slices (i.e. 15 mm section) covering the widest section of the lumbar cord (i.e. the LSE) were identified, by outlining the cord using the active surface model (ASM) segmentation method available with JIM 6.0 (http://www.xinapse.com), as previously described .
The diffusion-weighted images were corrected for motion using slice-wise linear registration implemented in FSL (http://www.fmrib.ox.ac.uk/fsl/), with registration transformations estimated among non-DW images, interspersed throughout the diffusion acquisition . Specifically, two degrees of freedom were estimated per slice (i.e. in-plane translation) between each interspersed non-DW image and a mean non-DW image (i.e. the registration target) obtained after a first iteration of the motion correction. Subsequently, each DW image was registered to the target using the transformation of the closest, preceding non-DW image .
DTI model fitting.
DTI was fitted with standard routines in CAMINO (http://cmic.cs.ucl.ac.uk/camino/) and voxel-wise maps of AD, RD, MD and FA were generated. These indices measure: the amount of diffusion along/across the dominant diffusion direction (AD/RD); the mean amount of diffusion (MD); and the anisotropy of the diffusion tensor (a measure of spatial variability of the amount of diffusion) (FA).
Image segmentation was performed with the aim of identifying GM and WM voxels within the three slices of the LSE, in diffusion space. Using the mean b = 0 volume, the three slices corresponding to the LSE, previously identified from the 3D-FFE, were segmented using ASM to obtain the whole cord outline. GM was manually outlined on an image obtained by averaging DW images with good contrast between GM and WM, as previously demonstrated [23, 24]. Binary masks for the GM, WM and whole cord were subsequently created and eroded prior to their application to the DTI maps; WM was defined as the difference between the whole cord and GM masks.
Six out of fourteen volunteers had a repeated scan on a different occasion, with a minimum of 7 days and a maximum of 14 days in between the first and the second visits. For these subjects, the reproducibility of the DTI indices was evaluated within GM, WM and the whole cord by calculating the percentage coefficient of variation (%COV).
Influence of the diffusion encoding protocol on the quality and reproducibility of the DTI measurements.
MR imaging of the lower SC mostly involves RF coil designs with receiving elements beneath a flat surface. This means that inter-subject variability in positioning the volume of interest (i.e. LSE) with respect to the RF coil surface can influence image quality and therefore the reliability of the DTI measurements, for a predefined diffusion protocol. For instance, high inter-subject variations of SNR due to variable distance between the receiving elements and the volume of interest can cause subject-dependent bias of FA (at moderate b, FA has been reported to increase as SNR decreases ). This could ultimately lead to artificial increase of the between-subject variability of this index. On the other hand, for a fixed distance between the receiving elements and the volume of interest, variations of the diffusion protocol in terms of the number of diffusion encoding directions used can affect the estimation of the diffusion properties .
In this work, the quality and reproducibility of the DTI indices were evaluated by studying the effect of altering the number of diffusion encoding directions within the same protocol in all subjects. For this purpose, the DTI model was retrospectively fitted to gradually reduced sets of measurements optimally spread over the sphere (60 to 10, with steps of 10) , extracted with CAMINO from the original 60-direction scheme.
The quality of the fitted DTI indices was evaluated by visual inspection and by calculating the mean values within each tissue-type (GM, WM, whole cord) in order to quantify the contrast (C) and the contrast-to-noise ratio (CNR) between GM and WM as previously demonstrated , using the relations: and where, SIWM, SIGM, SDWM and SDGM respectively indicate the mean value of each one of the DTI indices within the WM and GM masks, and the corresponding standard deviation values. C and CNR were calculated in all subjects but without inclusion of repeated scans, which were performed as part of the reproducibility assessment.
Reproducibility (%COV) as a function of the number of diffusion encoding directions was also evaluated in order to establish an adequate number of diffusion-weighted measurements needed to obtain reliable tissue-specific estimation of DTI indices for the population under investigation. DTI indices were assessed in 6 study participants who had repeated scanning, and for whom the cord/coil distance was within the lower and upper bounds previously identified in a larger population (S1 Appendix).
To obtain the %COV, linear mixed null models were fitted for each measure and for each diffusion encoding scheme, with subject random effect; the %COV was then calculated using the estimated within subject standard deviations and means.
To examine the variation in means of the DTI indices within each tissue-type across the number of diffusion encoding schemes, linear mixed models were fitted over the various diffusion encoding schemes with diffusion encoding scheme as fixed effect; a joint wald test was then used to test the null hypothesis of no variation across the schemes. There was no evidence of departure from model normality assumptions. These models were fitted in Stata 14.1 (Stata Corporation, College Station, Texas, USA).
Possible differences in DTI indices between GM and WM were assessed using paired t-tests, after checking the normality of the paired differences. Statistical significance was accepted at p<0.05.
All the study participants tolerated the imaging session well without the need to discard any data because of motion related artifacts. The number of slices obtained in the present study was sufficiently high to ensure coverage of the LSE in all cases, although it was possible that, in some cases (e.g. small-framed individuals), the same number of slices may have covered a section beyond the minimum required, in which case the extra slices were simply discarded. An example of the high-resolution structural images acquired through the T11-L1 level in a single healthy subject and the corresponding quantitative DTI parameter maps are shown in Fig 1.
The rightmost column corresponds to the T11 level, with contiguous columns moving inferiorly towards L1 (leftmost).
GM, WM and whole cord values (mean ± SD) of AD, RD, MD and FA in 14 healthy subjects are shown in Table 1. From Table 1 it can be seen that FA is higher in WM compared to GM (mean across 14 participants of 0.58 and 0.40 in WM and GM respectively, p<0.001), AD is higher in WM compared to GM (1.66 μm2ms-1 and 1.20 μm2ms-1 respectively, p<0.001), MD is higher in WM compared to GM (0.94 μm2ms-1 and 0.82 μm2ms-1 respectively, p<0.001) and RD is lower in WM compared to GM (0.58 μm2ms-1 and 0.63 μm2ms-1 respectively, p = 0.007). Separate box plots displaying the distribution of the DTI indices in all study participants are shown in Fig 2.
The results from the reproducibility assessment (%COV) using the original acquisition protocol, with diffusion-weighting applied along 60 directions, are reported in Table 2. The %COV was found to be lower than 10% in all cases, with the highest values observed for FA and the lowest for MD.
The effect of fitting the DTI model with gradually reduced sets of measurements on the quality of the DTI parameter maps is shown in Fig 3. It is apparent from the figure that the quality of the indices degrades as the number of diffusion directions decreases. The distribution of C and CNR values between GM and WM for all DTI indices and in all subjects as a function of the diffusion encoding scheme are shown in Fig 4. The figure shows that, both for C and CNR, there is a trend towards increasing values when a higher number of diffusion encoding direction is used. This effect appears to be evident across all the DTI indices, with more pronounced improvements noted between 10–40 diffusion encoding directions, while more consistent values were obtained between 40–60 directions.
The results from fitting the DTI model with gradually reduced sets of measurements on the %COV of the DTI indices within each tissue-type is demonstrated in Fig 5. This figure shows that, at least in WM and the whole cord, there was a trend towards decreasing %COV values in most of the DTI indices when the model was fitted with 20 or more sets of measurements. In GM, apart from MD and RD, there was no other apparent gain from using a number of diffusion directions higher than 20.
The results of the linear mixed models examining variation in means of the DTI indices within each tissue-type across the number of diffusion directions are shown in Table 3. From the table, two main observations can be made. Firstly, that the MD measures are the least affected in terms of the number of diffusion directions used in the present study. Secondly, in all other DTI indices the variation in mean values becomes less significant as the number of diffusion directions increases, and the percentage size of the largest difference for each measurement also becomes lower.
This study has shown that tissue-specific (i.e. GM and WM) DTI indices within the LSE can be obtained reliably using a commercially available 3T MR system and analysis software. However, in order to achieve the results presented here, several technical considerations had to be addressed such as the choice of the acquisition and readout type, the localisation of the LSE to ensure consistent measurements across all subjects and the reproducibility of measurements.
EPI-based rFOV acquisition variants have recently shown promise for studying the SC [15–21]. The rFOV acquisition and readout scheme used in this work is based on a product sequence [15, 26], which has been shown to be suitable for use at any level of the SC in the axial plane . The sequence offers reduced readout duration, combines fat saturation with outer volume suppression, and enables the reliable acquisition of images of the lower SC, which are free from aliasing and susceptibility artifacts and with sufficiently high resolution to allow tissue-specific assessment. Using this acquisition it was also possible to account for physiological motion through the use of cardiac gating, even though CSF pulsation is known to be considerably lower in the lumbar spine than in the cervical and thoracic spine [38, 39].
Recently, rFOV ss-EPI  and large FOV ss-EPI [40, 41] acquisitions have been used to obtain tissue-specific DTI indices in the neurologically intact lower SC. Whilst such studies are necessary to establish reference values in the healthy population in order to better interpret results in neurological disease, it is essential that measurements are obtained consistently, as this enables more meaningful comparisons. For example, positional variation of the lower SC must be accounted for, as has been shown in a recent study suggesting the use of the LSE as anatomical landmark . In the present study, the suggested high-resolution acquisition was included in the protocol, as well as a single b-value diffusion-weighted scan, to identify the LSE and facilitate subsequent inter-subject and intra-subject comparison of quantitative measurements. This allowed normative tissue-specific DTI indices within the LSE to be obtained consistently between-subjects, but also allowed a more robust assessment of the reproducibility of the measurements from repeated scans i.e. within-subject assessment.
The reproducibility of DTI measurements obtained using rFOV ss-EPI acquisitions in the lower SC has not been reported to date although studies using similar acquisitions have provided results in the upper SC [22, 23]. The results from this study have shown that the %COV values were lower than 10% in all cases, when using the original protocol with diffusion encoding applied in 60 directions, and these results are in line with the previous reports in the upper SC. However, results from studies investigating different levels of the SC may not be directly comparable because of various technical reasons such as the b-values used for diffusion encoding or the number of diffusion directions.
In this study, the effect of fitting the DTI model to gradually reduced sets of measurements was examined retrospectively by assessing any changes in the quality (C and CNR measures) and reproducibility (%COV) of the DTI indices. In terms of quality, it was shown that for CNR in particular there was a trend towards increasing values when a higher number of diffusion encoding directions was used, and this effect was more pronounced between 10–40 diffusion encoding directions, while more consistent values were obtained between 40–60 directions. This suggests that there maybe a higher benefit from using 40 or more diffusion encoding directions. In terms of reproducibility, this exploratory study has shown that, even though a surface RF coil was used, the %COV values for all the DTI indices were below 10% when the DTI model was fitted with 20 or more distributed diffusion directions; this result is in agreement with previous general recommendations [42, 43]. Therefore, this exploratory study confirms the feasibility of obtaining DTI indices reliably in the lower SC using a surface coil and quantifies for the first time the effect of the angular resolution of the diffusion encoding protocol in terms of DTI scan-rescan reproducibility.
The results of the linear mixed models examining variation in means of the DTI indices within each tissue-type across the number of diffusion directions have shown that the MD measures are the least affected in terms of the number of diffusion directions used. This is likely due to the fact that MD represents the average diffusivity over the unit sphere, and the mathematical operation of averaging effectively acts as a low pass filter i.e. it is robust towards noise. With regard to all other DTI indices, the variation in mean values becomes less significant as the number of diffusion directions increases, and the percentage size of the largest difference for each measurement also becomes lower, suggesting that there maybe a higher benefit in such measures when more directions are used e.g. 40 or more (Table 3). Of course, the use of a higher number of diffusion directions requires longer examination times, and this may not always be feasible in practice. Nevertheless, the results from this study provide a useful index to guide future applications.
In order to better interpret the DTI indices obtained in this study, one could examine previous measurements obtained in the upper SC using similar methods to establish whether DTI indices vary naturally along the length of the cord; one could also examine previous measurements obtained in the lower SC using similar methods to investigate whether or not the results are comparable. In previous research [23, 24], identical acquisition and analysis protocols as in this work were used to investigate the cervical SC and have obtained GM values for FA (0.56–0.57), MD (0.82–0.92 μm2ms-1), RD (0.53–0.54 μm2ms-1), AD (1.41–1.6 μm2ms-1) and WM values for FA (0.76–0.8), MD (0.91–0.97 μm2ms-1), RD (0.36–0.39 μm2ms-1), AD (1.96–2.16 μm2ms-1). From Table 1 it can be seen that, in comparison, values in the LSE-GM were lower for FA (0.4), AD, (1.2 μm2ms-1), higher for RD (0.63 μm2ms-1) and similar for MD (0.82 μm2ms-1); values in the LSE-WM were lower for FA (0.58), AD, (1.66 μm2ms-1), higher for RD (0.58 μm2ms-1) and similar for MD (0.94 μm2ms-1). These results are in agreement with previous studies showing a reduction in FA and AD in the lower SC (both in GM and WM) as compared to the cervical SC, whereas MD remains relatively consistent throughout the entire length of the cord [26, 41].
Considering the aforementioned methodological differences between this and other rFOV DTI studies of the lower SC, it may not be possible to directly compare the DTI indices obtained here. Two studies have used similar acquisition protocols as in the present study to investigate the lower SC and have obtained WM average values for FA (~0.63–0.67), MD (~0.85–0.92 μm2ms-1) [26, 27]. From Table 1 it can be observed that, in comparison, values in the LSE-WM in this work were lower for FA (0.58) and higher for MD (0.94 μm2ms-1). One of the two studies also reported GM values for FA, which were lower than the present study (0.32 and 0.4, respectively) . Possible explanations of these differences could be related to the sample size, the b-values used for diffusion encoding, the number of diffusion directions and the possibility that measurements were obtained from dissimilar regions in the lower SC in each case. Such results emphasise the importance to identify and reduce the sources of errors as much as possible, developing robust imaging protocols to facilitate reliable longitudinal observational studies in neurological disease.
In the present study, variations in the DTI indices at different spinal cord levels were not examined specifically, thus the true magnitude of such potential variations is unknown. However, assuming that the same DTI protocol is employed to study the different levels of the spinal cord each time, hence minimising any influence related to technical factors, any variations in the DTI indices along the length of the spinal cord are likely to be related to microstructural differences within each tissue-type (GM and WM). For example, previous studies in the cervical cord have suggested that there is a higher alignment of white matter fibres at this level, with a proportion of these fibres having a rather large diameter [15, 41, 44]. In turn, this may explain the higher FA and AD (due to the higher alignment of fibres) and the lower RD (as the presence of larger axons reduces the effective diffusion length in both intra and extra-axonal space) values observed in the upper cord as compared to the lower cord, where a progressive decrease in the number of long axons is suspected [26, 41]. Furthermore, previous detailed examination of the cervical cord from C1–C6 demonstrated that higher FA and AD values were mostly obtained at the C2/C3 level and that these values reduced when moving caudally towards the cervical enlargement, possibly due to the higher neural input and output to the upper limbs, which could lead to some disruption of the directional coherence of the nerve fibres on the scale of a voxel [15, 45]. On the other hand, when calculating MD, the influence of the axon/dendrite orientation distribution is factored out, possibly explaining the rather constant MD values along the length of the cord. Differences in DTI indices between the cervical cord and lumbar cord, could be further explained by considering the differences in GM volume fraction at the two levels: GM volume fraction in the cervical cord is 18% whereas in the lumbar cord is 36.3% .
In summary, the results of this work demonstrate that it is possible to obtain reliable tissue-specific DTI measurements within the LSE using a clinical MR system at 3T in vivo. Future work will aim to translate the acquisition and analysis method presented here to clinical studies of neurological conditions affecting the lower SC.
S1 Appendix. Variability among individuals in terms of the distance between the lumbosacral enlargement and the surface of the coil.
S1 Dataset. Raw data.
S1 Fig. The distance between the lumbosacral enlargement and the surface of the coil was determined manually using a line distance tool and verified on both the sagittal (left) and axial planes (right).
S2 Fig. Distance between the lumbosacral enlargement and the surface of the RF coil measured in 14 study participants.
The plot demonstrates the distribution of the measurements obtained in this study as compared with the median, lower and upper quartile values previously determined from a total of 60 randomly selected scans.
This work and the NMR Research Unit are funded by the U.K Multiple Sclerosis Society (http://www.mssociety.org.uk/). This work was supported by the National Institute for Health Research University College London Hospitals Biomedical Research Centre (NIHR BRC UCLH/UCL). FP and SO are funded by NIHR BRC UCLH/UCL High Impact Initiative (BW.mn.BRC10269). SO also receives funding from the EPSRC (EP/H046410/1, EP/J020990/1, EP/K005278), the MRC (MR/J01107X/1) and the NIHR Biomedical Research Unit (Dementia) at UCL.
- Conceptualization: MCY FG PL FP RSS MB DRA SO DHM CAMGW.
- Data curation: SO DHM CAMGW.
- Formal analysis: MCY FG PL DRA.
- Funding acquisition: SO DHM CAMGW.
- Investigation: MCY FG PL FP RSS MB.
- Methodology: MCY FG PL FP RSS MB DRA CAMGW.
- Project administration: SO DHM CAMGW.
- Resources: SO DHM CAMGW.
- Software: MCY FG PL FP RSS MB DRA SO DHM CAMGW.
- Supervision: SO DHM CAMGW.
- Validation: SO DHM CAMGW.
- Visualization: MCY FG.
- Writing – original draft: MCY FG PL FP RSS MB DRA SO DHM CAMGW.
- Writing – review & editing: MCY FG PL FP RSS MB DRA SO DHM CAMGW.
- 1. Le Bihan D. Looking into the functional architecture of brain with diffusion MRI. Nat Rev Neurosci. 2003;4(6):469–80. doi: 10.1038/nrn1119. pmid:12778119
- 2. Basser PJ, Mattiello J, LeBihan D. MR diffusion tensor spectroscopy and imaging. Biophys J 1994;66:259–267. doi: 10.1016/S0006-3495(94)80775-1. pmid:8130344
- 3. Ciccarelli O, Werring DJ, Wheeler-Kingshott CA, Barker GJ, Parker GJ, Thompson AJ, et al. Investigation of MS normal-appearing brain using diffusion tensor MRI with clinical correlations. Neurology 2001;56(7):926–33. pmid:11294931
- 4. Mahoney CJ, Simpson IJ, Nicholas JM, Fletcher PD, Downey LE, Golden HL, et al. Longitudinal diffusion tensor imaging in frontotemporal dementia. Ann Neurol. 2015;77(1):33–46. doi: 10.1002/ana.24296. pmid:25363208
- 5. Gregory S, Cole JH, Farmer RE, Rees EM, Roos RA, Sprengelmeyer R, et al. Longitudinal diffusion tensor imaging shows progressive changes in white matter in Huntington’s disease. J Huntington’s Dis. 2015;4(4):333–46.
- 6. Wheeler-Kingshott CA, Stroman PW, Schwab JM, Bacon M, Bosma R, Brooks J, et al. (2014) The current state-of-the-art of spinal cord imaging: Applications. Neuroimage 84:1082–1093. doi: 10.1016/j.neuroimage.2013.07.014. pmid:23859923
- 7. Stroman PW, Wheeler-Kingshott C, Bacon M, Schwab JM, Bosma R, Brooks J, et al. (2014) The current state-of-the-art of spinal cord imaging: Methods. Neuroimage 84:1070–1081. doi: 10.1016/j.neuroimage.2013.04.124. pmid:23685159
- 8. Turner R, Bihan DL. Single shot diffusion imaging at 2.0 Tesla. J Magn Reson 1990;86:445–452.
- 9. Cercignani M, Horsfield MA, Agosta F, Filippi M. Sensitivity-encoded diffusion tensor MR imaging of the cervical cord. AJNR Am J Neuroradiol. 2003;24(6):1254–1256. pmid:12812966
- 10. Tsuchiya K, Fujikawa A, Suzuki Y. Diffusion tractography of the cervical spinal cord by using parallel imaging. AJNR Am J Neuroradiol. 2005;26:398–400. pmid:15709143
- 11. Bammer R, Augustin M, Prokesch RW, Stollberger R, Fazekas F. Diffusion-weighted imaging of the spinal cord: interleaved echo-planar imaging is superior to fast spin-echo. J Magn Reson Imaging 2002;15(4):364–373. pmid:11948825
- 12. Le Roux PH, Darqui A, Carlier PG, Clark CA. Feasibility study of non Carr Purcell Meiboom Gill single shot fast spin echo in spinal cord diffusion imaging. MAGMA 2002;14(3):243–247. pmid:12098567
- 13. Gudbjartsson H, Maier SE, Mulkern RV, Mórocz IA, Patz S, Jolesz FA. Line scan diffusion imaging. Magn Reson Med. 1996;36(4):509–519. pmid:8892201
- 14. Bammer R, Herneth AM, Maier SE, Butts K, Prokesch RW, Do HM, et al. Line scan diffusion imaging of the spine. AJNR Am J Neuroradiology 2003;24(1):5–12.
- 15. Wheeler-Kingshott CA, Hickman SJ, Parker GJ, Ciccarelli O, Symms MR, Miller DH, et al. Investigating cervical spinal cord structure using axial diffusion tensor imaging. Neuroimage 2002;16(1):93–102. doi: 10.1006/nimg.2001.1022. pmid:11969321
- 16. Rieseberg S, Frahm J, Finsterbusch J. Two-dimensional spatially-selective RF excitation pulses in echo-planar imaging. Magn Reson Med 2002;47(6):1186–1193. doi: 10.1002/mrm.10157. pmid:12111965
- 17. Jeong EK, Kim SE, Guo J, Kholmovski EG, Parker DL. High-resolution DTI with 2D interleaved multislice reduced FOV single-shot diffusion-weighted EPI (2D ss-rFOV-DWEPI). Magn Reson Med 2005;54(6):1575–1579. doi: 10.1002/mrm.20711. pmid:16254946
- 18. Wilm BJ, Svensson J, Henning A, Pruessmann KP, Boesiger P, Kollias SS. Reduced field-of-view MRI using outer volume suppression for spinal cord diffusion imaging. Magn Reson Med. 2007;57(3):625–30. doi: 10.1002/mrm.21167. pmid:17326167
- 19. Saritas EU, Cunningham CH, Lee JH, Han ET, Nishimura DG. DWI of the spinal cord with reduced FOV single-shot EPI. Magn Reson Med. 2008;60(2):468–73. doi: 10.1002/mrm.21640. pmid:18666126
- 20. Gaggl W, Jesmanowicz A, Prost RW. High resolution reduced field-of-view diffusion tensor imaging using spatially selective RF pulses. Magn Reson Med. 2014;72(6):1668–79. doi: 10.1002/mrm.25092. pmid:24399609
- 21. Samson RS, Lévy S, Schneider T, Smith AK, Smith SA, Cohen-Adad J, et al. ZOOM or non-ZOOM? Assessing spinal cord diffusion tensor imaging protocols for multi-centre studies. PloS One 2016;11(5):e0155557. doi: 10.1371/journal.pone.0155557. pmid:27171194
- 22. Xu J, Shimony JS, Klawiter EC, Snyder AZ, Trinkaus K, Naismith RT, et al. Improved in vivo diffusion tensor imaging of human cervical spinal cord. Neuroimage 2013;67:64–76. doi: 10.1016/j.neuroimage.2012.11.014. pmid:23178538
- 23. Grussu F, Schneider T, Zhang H, Alexander DC, Wheeler-Kingshott CA. Neurite orientation dispersion and density imaging of the healthy cervical cord in vivo. Neuroimage 2015;111: 590–601. doi: 10.1016/j.neuroimage.2015.01.045. pmid:25652391
- 24. Kearney H, Schneider T, Yiannakas MC, Altmann DR, Wheeler-Kingshott CA, Ciccarelli O, et al. Spinal cord grey matter abnormalities are associated with secondary progression and physical disability in multiple sclerosis. J Neurol Neurosurg Psychiatry 2015;86(6):608–14. doi: 10.1136/jnnp-2014-308241. pmid:25097217
- 25. Naismith RT, Xu J, Klawiter EC, Lancia S, Tutlam NT, Wagner JM, et. al. Spinal cord tract diffusion tensor imaging reveals disability substrate in demyelinating disease. Neurology 2013;80(24):2201–2209. doi: 10.1212/WNL.0b013e318296e8f1. pmid:23667060
- 26. Wilm BJ, Gamper U, Henning A, Pruessmann KP, Kollias SS, Boesiger P. Diffusion-weighted imaging of the entire spinal cord. NMR Biomed. 2009;22(2):174–81. doi: 10.1002/nbm.1298. pmid:18727164
- 27. Petersen JA, Wilm BJ, von Meyenburg J, Schubert M, Seifert B, Najafi Y, et al. Chronic cervical spinal cord injury: DTI correlates with clinical and electrophysiological measures J Neurotrauma 2012;29(8):1556–66. doi: 10.1089/neu.2011.2027. pmid:22150011
- 28. McCombe PA, Gordon TP, Jackson MW. Bladder dysfunction in multiple sclerosis. Expert Rev Neurother. 2009;9(3):331–40. doi: 10.1586/14737220.127.116.111. pmid:19271942
- 29. Fernandez O. Mechanisms and current treatments of urogenital dysfunction in multiple sclerosis. J. Neurol. 2002;249(1):1–8. pmid:11954855
- 30. Anderson KD, Borisoff JF, Johnson RD, Stiens SA, Elliott SL. The impact of spinal cord injury on sexual function: concerns of the general population. Spinal Cord 2007;45(5):328–37. doi: 10.1038/sj.sc.3101977. pmid:17033620
- 31. Watanabe T, Vaccaro AR, Kumon H, Welch WC, Rivas DA, Chancellor MB. High incidence of occult neurogenic bladder dysfunction in neurologically intact patients with thoracolumbar spinal injuries. J. Urol. 1998;159(3):965–8. pmid:9474194
- 32. Sakakibara R, Hattori T, Tojo M, Yamanishi T, Yasuda K, Hirayama K. Micturitional disturbance in multiple system atrophy. Jpn J Psychiatry Neurol. 1993;47(3):591–8. pmid:8301874
- 33. Sakakibara R, Hattori T, Uchiyama T, Kita K, Asahina M, Suzuki A, et al. Urinary dysfunction and orthostatic hypotension in multiple system atrophy: which is the more common and earlier manifestation? J Neurol Neurosurg Psychiatry 2000;68(1):65–9. doi: 10.1136/jnnp.68.1.65. pmid:10601404
- 34. Yiannakas MC, Kakar P, Hoy LR, Miller DH, Wheeler-Kingshott CA. The Use of the Lumbosacral Enlargement as an Intrinsic Imaging Biomarker: Feasibility of Grey Matter and White Matter Cross-Sectional Area Measurements Using MRI at 3T. PLoS One 2014;9(8):e105544. doi: 10.1371/journal.pone.0105544. pmid:25170763
- 35. Jones DK, Basser PJ. “Squashing peanuts and smashing pimpkins”: how noise distorts diffusion-weighted MR data. Magn Reson Med. 52(5):979–93. doi: 10.1002/mrm.20283. pmid:15508154
- 36. Jones DK. Precision and accuracy in diffusion tensor magnetic resonance imaging. Top Magn Reson Imaging 2010;21(2):87–99. doi: 10.1097/RMR.0b013e31821e56ac. pmid:21613874
- 37. Jansons KM, Alexander DC. Persistent angular structure: new insights from diffusion MRI data. Inverse Problems 2003;19(5):1031–46. http://dx.doi.org/10.1088/0266-5611/19/5/303.
- 38. Du Boulay GH (1966) Pulsatile movements in the CSF pathways. Br J Radiol 39:255–62. doi: 10.1259/0007-1285-39-460-255. pmid:5295581
- 39. Lane B, Kricheff II (1974) Cerebrospinal fluid pulsations at myelography: a videodensitometric study. Radiology 110:579–87. doi: 10.1148/110.3.579. pmid:4811678
- 40. Bosma RL, Stroman PW. Characterisation of DTI indices in the cervical, thoracic, and lumbar spinal cord in healthy humans. Radiol Res Pract. 2012;2012:143705. doi: 10.1155/2012/143705. pmid:22295179
- 41. Ellingson BM, Ulmer JL, Kurpad SN, Schmit BD. Diffusion tensor MR imaging of the neurologically intact human spinal cord. AJNR Am J Neuroradiol. 2008;29(7):1279–84. doi: 10.3174/ajnr.A1064. pmid:18417607
- 42. Papadakis NG, Murrills CD, Hall LD, Huang CL, Adrian Carpender T. Minimal gradient encoding for robust estimation of diffusion anisotropy. Magn Reson Imaging 2000;18(6):671–9. pmid:10930776
- 43. Jones DK. The effect of gradient sampling schemes on measures derived from diffusion tensor MRI: a Monte Carlo study. Magn Reson Med. 2004;51(4):807–15. doi: 10.1002/mrm.20033. pmid:15065255
- 44. Feirabend H, Choufoer H, Ploeger S, Holsheimer J, Van Gool J. Morphometry of human superficial dorsal and dorsolateral column fibres: significance to spinal cord stimulation. Brain 2002;125: 1137–1149. pmid:11960902
- 45. Mamata H, De Girolami U, Hoge WS, Jolesz FA, Maier SE. Collateral nerve fibers in human spinal cord: visualization with magnetic resonance diffusion tensor imaging. Neuroimage 2006;31(1):24–30. doi: 10.1016/j.neuroimage.2005.11.038. pmid:16431139
- 46. Goto N, Otsuka N. Development and anatomy of the spinal cord. Neuropathology 1997;17(1):25–31.