Arterial spin labeling (ASL) magnetic resonance imaging is increasingly used to quantify task-related brain activation. This study assessed functional ASL (fASL) using pseudo-continuous ASL (pCASL) product sequences from two vendors. By scanning healthy participants twice with each sequence while they performed a motor task, this study assessed functional ASL for 1) its sensitivity to detect task-related cerebral blood flow (CBF) changes, and 2) its reproducibility of resting CBF and absolute CBF changes (delta CBF) in the motor cortex. Whole-brain voxel-wise analyses showed that sensitivity for motor activation was sufficient with each sequence, and comparable between sequences. Reproducibility was assessed with within-subject coefficients of variation (wsCV) and intraclass correlation coefficients (ICC). Reproducibility of resting CBF was reasonably good within (wsCV: 14.1–15.7%; ICC: 0.69–0.77) and between sequences (wsCV: 15.1%; ICC: 0.69). Reproducibility of delta CBF was relatively low, both within (wsCV: 182–297%; ICC: 0.04–0.32) and between sequences (wsCV: 185%; ICC: 0.45), while inter-session variation was low. This may be due to delta CBF’s small mean effect (0.77–1.32 mL/100g gray matter/min). In conclusion, fASL seems sufficiently sensitive to detect task-related changes on a group level, with acceptable inter-sequence differences. Resting CBF may provide a consistent baseline to compare task-related activation to, but absolute regional CBF changes are more variable, and should be interpreted cautiously when acquired with two pCASL product sequences.
Citation: Steketee RME, Mutsaerts HJMM, Bron EE, van Osch MJP, Majoie CBLM, van der Lugt A, et al. (2015) Quantitative Functional Arterial Spin Labeling (fASL) MRI – Sensitivity and Reproducibility of Regional CBF Changes Using Pseudo-Continuous ASL Product Sequences. PLoS ONE 10(7): e0132929. https://doi.org/10.1371/journal.pone.0132929
Editor: Martin Walter, Leibniz Institute for Neurobiology, GERMANY
Received: December 14, 2014; Accepted: June 21, 2015; Published: July 14, 2015
Copyright: © 2015 Steketee et al. This is an open access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited
Data Availability: For ethical reasons the raw data collected cannot be made publicly available. The study was approved by the authors’ Medical Ethics Committee under the condition that access to the data is granted only to 1) members of the research team, 2) the Medical Ethics Committee members that approved this study, and 3) authorized personnel of the Health Care Inspectorate. Hence, participants did not consent to publicly archiving their data. However, requests for anonymized data can be sent to Dr. M. Smits at email@example.com.
Funding: This project was financially supported by the Dutch non-governmental organization ‘Stichting Coolsingel’ (Rotterdam, NL). The funder had no role in the study design; data collection, analysis or interpretation; writing of the report; or decision to submit the article for publication (grant no. 179, received by MS; http://www.stichtingcoolsingel.nl/).
Competing interests: The authors have declared that no competing interests exist.
Arterial spin labeling (ASL) perfusion magnetic resonance imaging (MRI) is being increasingly used for imaging of task-related brain activation. Such functional ASL (fASL) has been used to study the neural correlates of a multitude of cognitive domains, including attention , memory , language , visual  and sensorimotor processing , and is increasingly considered as an alternative to blood oxygen level-dependent (BOLD) functional MRI (fMRI), which has been predominantly used as a marker for neural activation during the last two decades.
ASL has several advantages over BOLD imaging with respect to acquisition and interpretation. First, ASL has better sensitivity in low frequency paradigms. The BOLD signal has been shown to be confounded by slow ‘drift’ effects in baseline signal, which are reduced in ASL imaging as a result of the pairwise subtraction of labeled and unlabeled images . Second, despite the intrinsically low signal-to-noise ratio (SNR) of ASL, spatial localization of neuronal activity seems more accurate when measured with ASL than with BOLD. The BOLD signal is affected by macrovascular venous effects  whereas ASL is more sensitive to the microvasculature . The interpretation of the BOLD signal is more complex as it reflects a combination of cerebral blood flow (CBF), cerebral blood volume (CBV) and cerebral metabolic rate of oxygen consumption (CMRO2) [8,9], whereas ASL provides a measure of CBF that is relatively less sensitive to other hemodynamic parameters. Furthermore, ASL provides an in principle quantitative measure of CBF, whereas the BOLD signal is relative. These advantages favor the application of fASL over fMRI BOLD for task-related brain imaging.
The quantitative aspect of ASL in particular could facilitate the comparison and exchange of CBF values across multiple sites and enable multicenter studies, for instance, to pool data. However, before fASL can be used as such, its variability needs to determined, not only within sessions and scanners, but also between product sequences of different vendors, as each vendor provides its own particular ASL implementation. Reproducibility of ASL in general is affected by intrinsic properties, such as low SNR and relative sensitivity to hemodynamics such as arterial transit time (ATT) . In addition, although within-sequence reproducibility is sufficient for the commonly available labeling schemes [11–22], pseudo-continuous arterial spin labeling (pCASL) has been shown to be best reproducible within session, scanner, and vendor, being more stable and less variable than continuous ASL (CASL) and pulsed ASL (PASL) [23,24].
Nevertheless, not every user is aware of the potential impact of these factors and may assume that any ASL implementation will provide the same information, as can be expected from a quantitative technique. This may seem particularly appealing for the quantification of brain activation in functional imaging studies. The extent to which different vendor implementations affect these data is not known. We will therefore compare two pCASL product sequences as implemented by two different vendors, while limiting adjustment of sequence parameters to within the constraints imposed by the vendor-specific implementation.
Baseline or resting CBF values have been found to be well reproducible within sessions, within scanners, and between scanners of the same vendor on a whole-brain level, whereas on a regional level reproducibility was lower [23,24]. We previously assessed the reproducibility of whole-brain resting CBF within and between pCASL product sequences at 3T scanners of two different vendors . Mean global CBF did not differ between product sequences, but voxel-by-voxel assessment revealed regional differences. Regional variability presents a challenge for fASL, where local effects are of particular interest. In addition to the variability in regional CBF changes, the variability in the detection of such CBF changes needs to be assessed. Sufficient and similar sensitivity to detect local task-induced CBF changes is a prerequisite for multicenter fASL implementations, and essential to good reproducibility.
As of yet, variability of quantitative fASL and variation of sensitivity for task-induced CBF changes between product sequences of different vendors have not been studied. Not only is this information essential for exchanging and comparing fASL data, but results generated by one product sequence can only be generalized to another if variability between them is known. The aim of the present study was to assess quantitative fASL by 1) assessing sensitivity to detect regional CBF changes in a voxel-wise whole-brain analysis, and 2) by investigating regional reproducibility of both resting CBF and task-induced CBF changes in the primary motor cortex, within and between pCASL product sequences from two major vendors. We investigated this by means of paced finger tapping, a simple behavioral paradigm that is known to elicit robust and consistent regional activation in the primary motor cortex in a multitude of activation studies using BOLD as well as fASL (e.g [5–7]). We employed this paradigm in healthy volunteers using product sequences from two different vendors.
Twenty-two healthy volunteers, aged 18–40 years, were recruited as part of a larger study on ASL reproducibility . Participants were recruited through advertisement at the University of Amsterdam. Only participants with no history of neurological or psychiatric disease were included. Participants that used medication other than contraceptives, or had contraindications for MRI were excluded.
Participants were asked to limit their consumption of alcohol, nicotine and caffeine to a maximum of three units 12–24 hours prior to scanning, and to refrain from consuming alcohol, nicotine and caffeine 12 hours prior to scanning. The study was approved by the local medical research ethics committees of both sites: the Erasmus MC–University Medical Center Rotterdam and the Academic Medical Center, Amsterdam and was conducted according to the Declaration of Helsinki. All participants gave written informed consent and received financial compensation for participation.
Imaging was performed on a 3T Intera (Philips Healthcare, Best, the Netherlands) and a 3T Discovery MR750 (GE Healthcare, WI, USA) scanner, using an 8 channel receive head coil. Participants were scanned twice on both scanners, i.e. four sessions in total, in no specific order (Fig 1A). Scanning sessions were separated by at least one week, but no more than four weeks.
Schematic overview of A) experimental design and analyses of B) whole-brain voxel-wise activation sensitivity and of C) reproducibility of regional resting CBF and regional task-induced CBF changes.
A high resolution 3D T1-weighted (T1w) scan for anatomical reference was acquired during one of the two sessions on each scanner. Perfusion data were acquired using pCASL product sequences that were provided by the vendor of each scanner. As we aimed to assess the reproducibility of the currently implemented (i.e. product) sequences, we chose to employ vendor-supplied sequences rather than to reprogram the sequences to make them match completely. Hence, parameters were only adjusted within the given limits of the clinical scanning sequences. Imaging data of the two sites were acquired by two different researchers. Strict agreements were made about the complete process of instructing and positioning participants in order to minimize differences between researchers and subsequently between sequences. Details of both pCASL sequences are listed in Table 1. Note that on the GE scanner a segmented 3D readout was employed, whereas on the Philips scanner single-shot multi-slice 2D imaging was combined with averaging to obtain a temporal resolution similar to the GE-sequence. Another difference between sequences was that on the GE scanner perfusion data are averaged during acquisition, whereas for Philips data is averaged after acquisition, after pair-wise subtraction of label and control images during post-processing. As the GE sequence only provides three fixed post-labeling delays (1 025, 1 525 and 2 025 ms for respectively children, adults, and older adults or adults with cerebrovascular disease), a delay of 1 525 ms, considered most suitable for the current population, was selected for all ASL imaging.
Motor activation paradigm
Eight ASL scans were acquired, during which participants performed a blocked motor activation task (Fig 1). Block length was equal to acquisition time of one scan, i.e. 2 minutes. Participants were instructed to tap the fingers of both hands to the thumbs in random order (finger tapping, FT) during the odd scans and to keep their hands still (rest) during even scans. FT was auditorily paced at a frequency of 1 Hz.
The imaging data were processed according to the methods described in full in Bron et al., 2014 .
The unified tissue segmentation method  of SPM8 (Statistical Parametric Mapping 8, Wellcome Trust Centre for Neuroimaging, University College London, UK) was used to obtain gray matter (GM), white matter (WM) and cerebrospinal fluid (CSF) probability maps from the T1w image.
For Philips data, label and control pCASL images were pair-wise subtracted (without motion correction) and averaged to obtain perfusion weighted images. For GE data, the perfusion-weighted images, as provided by the scanner, were used. For each participant, the perfusion-weighted image and the GM probability map were rigidly registered (Elastix registration software ). The results of the registration were visually inspected and showed good correspondence.
To quantify the perfusion-weighted maps of both pCASL sequences as cerebral blood flow (CBF) maps, a single-compartment model was used :
Parameters used in this model and their values are summarized in Table 2. Differences in effective post-labeling delay for different slices resulting from the 2D multi-slice readout were accounted for in the Philips data (Table 2).
Whole-brain voxel-wise preprocessing and activation sensitivity analysis
Registered T1w images and CBF maps were transformed to a common template space based on the T1w images of all participants ; CBF maps were smoothed using an isotropic 8 mm full width at half maximum (FWHM) kernel.
Voxel-wise differences within and between sequences in relation to finger tapping (Fig 1B) were assessed using SPM8. Averaged CBF maps per block of finger tapping (FT) and rest were convolved with the hemodynamic response function and modeled on an individual level using a General Linear Model (GLM), yielding parameter estimates for the main effects of FT and rest and the contrast [FT > rest] which were subsequently used in group analyses (Fig 1B). Task-induced changes were assessed per session per sequence by pair-wise comparison of main effects of FT and rest for each participant during each session. As participants were scanned in a random order, we chose to assess inter-sequence differences by means of a repeated measures ANOVA on the contrast [FT > rest] for GE (session) 1 compared to Philips 2 and GE 2 to Philips 1 (n = 44). By comparing sessions this way, we can assume that temporal physiological variation affected intra-sequence and inter-sequence reproducibility to a similar extent. All voxel-wise results were thresholded at p<0.001 without correction for multiple comparisons, to be maximally sensitive to intra- and inter-sequence differences in the detection of CBF changes.
ROI preprocessing and regional reproducibility analysis
ROI labeling and selection.
Individual CBF maps were transformed to individual T1w image space for region of interest (ROI) analysis. ROIs for each participant were defined using a multi-atlas approach by registering thirty labeled T1w images, each containing 83 ROIs [29,30], to the participants’ T1w images, using a rigid, affine, and non-rigid model consecutively. For the current study, we focused on the bilateral primary motor cortex, i.e. the superior aspect of the precentral gyri, containing the hand-motor area . The hand motor area was identified in all thirty T1w atlas images and followed down to the level of the cingulum, which was used as the inferior cut off of the precentral gyrus. Analysis of CBF in the primary motor cortex was performed in GM only.
CBF post-processing in the primary motor cortex.
For every pCASL scan, mean GM CBF values were obtained from the left and right superior precentral gyrus (primary motor cortex). CBF values were averaged per session over the four FT blocks and over the four rest blocks, and then over the primary motor cortex bilaterally, such that for every session we obtained one mean GM CBF value in the bilateral primary motor cortex for the FT condition (CBFFT) and one for the rest condition (CBFrest, Fig 1C).
Task-induced regional CBF changes within sessions.
To assess task-induced CBF changes in the primary motor cortex, CBFFT and CBFrest within sessions were compared with paired t-tests (p < .05). Absolute CBF changes as a result of finger tapping are referred to as delta CBF: [CBFFT−CBFrest] (Fig 1C).
Reproducibility of regional resting CBF and regional task-induced CBF changes.
Intra- and inter-sequence reproducibility were assessed by the following measures (Fig 1C):
- Within-subject coefficients of variation (wsCV) were calculated as the ratio of the standard deviation of the CBF difference (SDdiff) between sessions to the mean CBF value of those sessions: wsCV = 100% (SDdiff/mean value). The SDdiff, rather than the SD of the mean, was used to reflect the extent of variability in differences in relation to the mean.
- Mean CBF values over sessions, mean CBF differences between sessions, SDdiff, and wsCVs and their 95% confidence intervals (CIs) are reported for CBFrest, CBFFT, and delta CBF.
Intra-sequence measures were calculated between the two sessions per sequence. Inter-sequence measures were calculated by comparing GE (session) 1 to Philips 2 and GE 2 to Philips 1 (n = 44).
- Intraclass correlation coefficients (ICC) and 95% CIs were calculated for CBFrest, CBFFT, and delta CBF. A two way-random model and absolute agreement were employed to allow for generalization of the results and to take into account systematic variability between sequences, respectively. ICCs were defined as function of ANOVA mean squares using the following formula :
in which BMS refers to the between-targets mean square (i.e. variance between participants), JMS refers to the between-judges mean square (i.e. variance between intra- or inter-sequence sessions) and EMS to the residual mean square (i.e. residual sources of variance), in a two-way ANOVA with n = 22 (intra-sequence) or 44 (inter-sequence) targets and k = 2 judges.
Inter-sequence ICCs were calculated between the two sessions per sequence by comparing GE (session) 1 to Philips 2 and GE 2 to Philips 1 (n = 44).
- Bland-Altman plots and 95% limits of agreement (mean difference ± 1.96 SDdiff) were created for CBFrest and delta CBF to visualize agreement within and between sequences.
Statistical analyses were carried out in IBM SPSS Statistics, version 20.0 (New York, USA).
Nine male and 13 female volunteers with a mean age of 22.1 ± 2.1 years (range: 19–27 years) participated in the study. It should be noted that one participant had CBF values that were 2–3 standard deviations higher than the group mean, but this participant was retained in the analysis as data were normally distributed (Kolmogorov-Smirnov tests did not detect significant deviations from normality in any session, p>.05). The two sessions scanned using the GE sequence were separated by 2.8 ± 1.0 weeks and those using the Philips sequence by 2.6 ± 0.9 weeks (not significant (n.s.)). Inter-sequence sessions were separated by 3.1 ± 1.1 weeks (GE 1 –Philips 2) and 2.6 ± 1.9 weeks (GE 2 –Philips 1), n.s. Both sessions scanned with the GE sequence took place at an earlier time of day than those scanned with the Philips sequence: 3:26pm ± 4h00min and 3:55pm ± 3h34min versus 8:16pm ± 2h06min and 7h47pm ± 2h38min respectively, p < .05.
Whole-brain voxel-wise activation sensitivity of pCASL sequences
Voxel-wise CBF changes in relation to FT are illustrated with t-statistic maps in Fig 2. Both with GE (Fig 2A) and Philips (Fig 2B) CBF increases were observed in the bilateral primary motor cortex in both sessions. Additional activation was observed in the supplementary motor area and the left cerebellum in GE session 2; and in the thalamus, and supplementary motor area in both Philips sessions. As can be appreciated visually, Philips (Fig 2B) seems to be more sensitive to detect activation than GE (Fig 2A). Upon formal assessment with repeated measures ANOVA (Fig 2C), differences between pCASL sequences were found in the right primary motor cortex, left precuneus, right posterior cingulate, and in the bilateral thalamus.
Activation maps are overlaid on a mean T1w scan. T-maps for the two sessions of A) GE and B) Philips sequences are thresholded at t = 3.52, p < .001 (uncorrected). C) shows the F-map depicting differences in activation between pCASL sequences, thresholded at F(2,63) = 7.7, p < .001 (uncorrected).
Task-induced regional CBF changes within sessions
Absolute CBF values differed systematically between pCASL sequences, with mean CBFrest values being 4.1 mL/100g GM/min lower as measured with GE than with Philips. The increase in CBF in the motor cortex as a result of FT was significant for both GE sessions and the second Philips session (Table 3). The increase in CBF due to FT relative to CBFrest was larger for the two GE sessions (6.13 and 4.27%) than for the two Philips sessions (2.27 and 3.53%).
Reproducibility of regional resting CBF and regional task-induced CBF changes
WsCV, SDdiff, mean CBF values and ICCs and 95% CIs are reported for CBFrest, CBFFT, and delta CBF in Table 4. WsCVs were comparable within and between pCASL sequences for CBFrest and CBFFT.
Reproducibility of CBFrest and CBFFT in terms of ICCs was moderate to good for both sequences, with ICCs of .69 and .66 for GE and .77 and .75 for Philips, respectively (Fig 3). Reproducibility of absolute delta CBF was poor for both sequences with ICCs of .32 (GE) and .04 (Philips), and CIs being 1.5–2.3 times larger than for CBFrest and CBFFT. Between sequences, reproducibility was reasonable for CBFrest (.69) and CBFFT (.61) and fair for absolute delta CBF (ICC: .45). Inter-sequence CIs for delta CBF were approximately 1.3 times larger than for CBFrest and CBFFT.
Error bars denote 95% confidence intervals.
Fig 4-I and 4-II show the agreement within and between sequences for CBFrest and absolute delta CBF respectively. For both CBFrest and delta CBF, differences within sequences were somewhat larger for measurements performed with GE (Fig 4A-I and 4A-II) than with Philips (Fig 4B-I and 4B-II). The low intra-sequence reproducibility of delta CBF with Philips in particular (Fig 4B-II) is illustrated by the variability relative to the mean effect being higher for delta CBF than for CBFrest (Fig 4B-I). Both CBFrest and delta CBF show a comparable spread in differences within and between sequences. Fig 4C-I and 4C-II show the agreement between sequences for CBFrest and delta CBF respectively, and illustrate that although the spread in differences is higher for CBFrest (Fig 4C-I), the variability relative to the mean effect is 1.5 times as large for delta CBF (Fig 4C-II).
Bland Altman plots of agreement between the two sessions per sequence (A, B) and between sequences (C) for resting CBF (I) and delta CBF (II) in the primary motor cortex. The solid line indicates the mean difference between sessions, dotted lines the 95% limits of agreement.
Using pCASL product sequences as supplied by two MR vendors, we found that sensitivity to detect primary motor cortex activation was sufficient and comparable for both sequences. Secondly, we found intra- and intersequence reproducibility of resting CBF in the motor cortex to be reasonably good, as was reproducibility of CBF during finger tapping. More important in the context of fASL however, we found absolute CBF changes to be only moderately reproducible, both within and between sequences, despite a more or less consistent task-induced CBF increase within sessions at the group level.
This study adopted a pragmatic and clinically applicable approach in employing pCASL product sequences as supplied by the vendors. Parameters were adjusted to match each other as much as possible, but within the constraints of the provided sequence. This reflects the de facto situation of commercially supplied sequences in which parameters can only be adjusted to a certain extent. As CBF measured with ASL is claimed to be and promoted as a quantitative measure, one would assume that measurements are independent of sequence parameters. Here we show to what extent standard implementation of the sequences gives rise to substantial differences and thus limited reproducibility for functional ASL.
The increase in CBF in the motor cortex during finger tapping is in line with previous cross-sectional fASL studies employing finger tapping paradigms [5,7,33,34]. Our results replicate these findings within both sequences, except in one of the Philips sessions. Although we found the extent and spatial patterns of activation to differ between sequences, they both detected activation in the primary motor cortex as a result of finger tapping, and formal comparison of activation patterns between sequences demonstrated only minor differences in motor areas. This supports the notion that on the group level, fASL is sufficiently sensitive to detect activation in the primary motor cortex, and that differences in sensitivity between pCASL product sequences are acceptable. The magnitude of CBF changes was, however, surprisingly modest in our study compared to previous studies, which may–at least partially–be due to methodological differences. For instance, our ROI comprised the entire superior precentral gyrus, which may have resulted in only a modest increase since CBF is averaged over this entire ROI. Previous studies quantified CBF in ROIs more spatially specific to activation, such as the ‘precentral hand knob’ or activated volumes [5,35].
We found reproducibility of resting CBF in the motor cortex to be reasonably good within and good between pCASL sequences. Our intra-sequence results are in line with previous whole-brain resting CBF studies that have shown reasonable to good reproducibility within and between sessions on the same scanner , as well as between scanners of the same vendor [16,24]. We recently compared resting gray matter CBF between two different pCASL sequences and also found it to be well reproducible on the global level . The current study focused on regional resting gray matter CBF and showed that, when compared to global CBF, reproducibility was slightly lower. Previous studies found that smaller regions are subject to higher variability and thus may yield lower reproducibility estimates [11,23]. In addition, regional reproducibility has been found to be lower for CBF measured at an interval of 2–4 weeks than for measurements within one day, implying that temporal physiological differences dominate between-weeks reproducibility [23,36]. Other sources of physiological variation may also be present, such as the one participant with CBF values that were consistently higher than the group mean. This physiological variation could have been accounted for by adding global CBF as a covariate, but as we aimed to demonstrate the variability in absolute regional CBF changes, scaling the signal would have defeated the objective of the current study.
Apart from temporal dynamics in physiology, regional variability could also be affected by the difference in effective post-labeling delay (PLD) between pCASL sequences. In order for labeled blood to reach the relatively superiorly located motor cortex, arterial transit time (ATT) is longer than for inferior regions, because of the larger distance between the labeling plane and the target tissue. This was previously demonstrated by Gonzalez-At et al. , who measured ATT to the visual region (5.1 cm from the labeling plane) to be 514 ms and to the motor region (11.5 cm from the labeling plane) 906 ms. Although in the current study the same initial PLD (1 525 ms) was applied for both sequences, the 2D multi-slice acquisition employed by Philips may have allowed labeled blood more time to reach superior slices, including regions with longer arrival times, than did the single time-point 3D acquisition employed by GE. This difference in effective PLD between sequences may have contributed to decreased reproducibility between their respective measurements. Moreover, the level of reproducibility of different regions seems to vary with PLD , with a PLD of 2 500 ms yielding better reproducibility than a PLD of 1 500 ms when using a 3D single time-point sequence. Therefore, effective PLD differences may have affected our results, as the precentral gyrus is located at the superior aspect of the brain, and therefore exhibits longer transit delays. This may have led to an underestimation of CBF values due to incomplete inflow of label, and to higher variability due to difference in arterial arrival times. Note that this is not resolved by merely prolonging the PLD, because although this may be a benefit for optimal bolus delivery, it also compromises signal due to label decay.
Differences between sequences may not only affect resting CBF measures, but also those of motor activation. Studies that compared fASL data obtained with 2D and 3D sequences found that activated clusters are generally larger when using 3D sequences, while 2D sequences yield larger effect sizes in terms of relative CBF changes [38,39]. Our results on the other hand showed larger activated clusters with the 2D sequence, and larger relative CBF changes with the 3D sequence. The larger effective PLD of Philips may have allowed more labeled blood to reach the primary motor cortex during finger tapping than GE, yielding larger activation clusters. In addition, the 3D sequence is more susceptible to spatial blurring, which obscures the gray matter to white matter contrast , and may attenuate signal from the gray matter. The larger relative signal change measured with GE on the other hand may be explained by decreasing ATT as a result of finger tapping . Although (pseudo-) continuous ASL techniques are not very sensitive to changes (especially decreases) in ATT , it is likely that the known decreased ATT during neuronal activation  also leads to faster extravasation of the label into the tissue compartment, leading to faster decay of the label as the longitudinal relaxation time of tissue is shorter than that of blood. This would lead to an underestimation of CBF during activation. Additionally, because of its shorter effective PLD, particularly in the superior regions, such shorter ATT after finger tapping may have caused the relative signal change as measured by GE to be higher than by Philips.
The reproducibility of delta CBF, i.e. the CBF difference observed between finger tapping and rest, is less straightforward to interpret. Despite the smaller inter-session variation of delta CBF differences (as indicated by smaller standard deviations of differences) when compared to that of resting CBF and finger tapping CBF, wsCVs are extremely high. The relatively small effect of delta CBF is more likely to be susceptible to high variabililty than resting and finger tapping CBF, which may affect its reproducibility to a larger extent than the other CBF measures. This is in fact reflected by the ICCs indicating poor intra-sequence reproducibility. Previous fASL studies reported wsCVs of 10–11% [35,42], and ICCs up to 0.74 , between sessions that were a week apart. However, these studies assessed the reproducibility of relative CBF changes, instead of absolute CBF changes. It has been suggested that relative CBF changes are more accurate and robust than absolute CBF changes, as they may reduce potential effects of basal perfusion variations on measures of neuronal activation . Relative CBF changes may therefore generate higher ICCs than absolute CBF changes. Nevertheless, as mentioned earlier, as absolute quantification is a specific advantage of fASL, it seems more appropriate to investigate the reproducibility of absolute CBF changes.
Slight variation in signal change as a result of finger tapping has been observed, but shown to be similar between sessions that took place on the same day or on different days . Raoult et al.  found similar levels of variation using a finger flexion-extension paradigm. Moreover, they found task-induced CBF to be higher, albeit not significantly, with shorter sequence lengths, and concluded that a motor paradigm with 4 blocks of 30s on/off activation is optimal for clinical practice . Longer sequence durations are considered to induce habituation and thus decreased activation. Our paradigm consisted of 2 minute blocks of on/off activation because of the limited temporal resolution of the GE sequence. These relatively long blocks thus may have attenuated activation, and thus the effect of finger tapping as compared to other studies, which may have reduced reproducibility.
On the other hand, one of the major reasons to use ASL for functional imaging is its suitability for low frequency designs, as it is much less sensitive to drift effects over time than BOLD fMRI . Wang et al.  even demonstrated that fASL shows constant sensitivity across different task frequencies corresponding to blocks lengths ranging from 0.5–5 minutes, with ASL outperforming BOLD contrast at a block length of 4 minutes. Some higher cognitive functions, such as sustained attention , depend on an experimental design with even longer blocks to detect slow, low-frequeny signal changes of interest, for which fASL is particularly well suited. Despite ASL’s appropriateness for such cognitive paradigms, we purposely chose a simple behavioral paradigm known to elicit robust and consistent regional activation, before moving on to more complex processes and paradigms. We find that even this simple motor activation paradigm gives rise to substantial variability, which warrants caution with respect to more complex and less robust designs.
To our knowledge, no studies exist on the reproducibility of task-induced CBF changes using pCASL product sequences from two different vendors. In the current study, we found reproducibility of task-induced CBF to be comparable within and between sequences, both in terms of wsCV and ICC. Nevertheless, the findings indicate that absolute CBF changes in the motor cortex still vary considerably, and this variation needs to be taken into account when comparing regional quantitative CBF changes, particularly between sequences. Therefore, absolute fASL data should not be simply pooled between product sequences.
This study has some limitations. First, we quantified CBF using a model that simplifies the actual in vivo situation, which would have required measurement of many variables that are difficult to obtain on an individual basis. This is illustrated by the discrepancy between the results of the voxel-wise group analysis and region-wise analysis: whereas the qualitative voxel-wise analysis showed acceptable inter-sequence differences, results from region-wise quantitative analysis are far less similar. Nevertheless, this lends support to our conclusion that current standard implementation of ASL and recommended analysis of ASL are not–yet–suited for quantitative functional ASL experiments. Second, we did not collect information on motor behavior. Although variations in frequency were avoided by externally pacing the finger tapping, we may have missed individual variations in tapping, which may have added to the variability. Next, as time of acquisition differed between sequences, with GE data collected earlier on the day, diurnal fluctuations in CBF may have added to variability between sequences. This potentially affected reproducibility of resting CBF more than that of delta CBF, as the latter is based on a subtractive measure. Still, inter-sequence reproducibility of resting CBF was found to be reasonably good. Furthermore, subsequent analysis of the Philips data was performed in a similar manner to maximize comparability with the GE data, i.e. by averaging over rest and activation periods, whereas one would normally choose to exploit the higher temporal resolution in a more formal manner within the design matrix. Finally, due to practical constraints we studied the product sequences of only two out of the three major vendors on the market. Although assessment of variability between the three vendors would have been more comprehensive, the current study was conducted as a proof-of principle, and demonstrated as such that substantial variability is already evident when product sequences of two vendors are compared. Future work should be directed at optimizing ASL sequences for functional imaging, and at assessing sensitivity and reproducibility of fASL in single-subject designs, as longitudinal studies and clinical application of fASL will eventually need to be aimed at repeated measurements within individuals.
In conclusion, in a voxel-wise whole-brain analysis, fASL shows sufficient sensitivity to detect regional CBF changes on a group level, both within and between pCASL product sequences of two different vendors. The between sequence reproducibility of fASL is comparable with within sequence reproducibility, although inter-sequence differences in readout should be taken into account. Although reproducibility of regional resting CBF is affected by differences in sequence implementation, particularly in the readout, resting CBF in the motor cortex may provide a reasonably consistent baseline to compare task-induced CBF to. The relatively low reproducibility of task-induced CBF changes in the primary motor cortex, however, should be taken into consideration when comparing fASL data between sessions and particularly between pCASL product sequences as implemented by different vendors. Its interpretation should be performed with caution in repeated measurements and multicenter designs, as current vendor-specific implementations do not allow for simple pooling of functional ASL data.
The authors are grateful to C.P. Mendez Orellana, MSc and S. van Bekkum, BSc, for assistance with data collection. The authors wish to thank the COST-AID (European Cooperation in Science and Technology–Arterial spin labeling Initiative in Dementia) Action BM1103 for facilitating meetings for researchers to discuss various application of ASL.
Conceived and designed the experiments: RMES HJMMM AJN MS. Performed the experiments: RMES HJMMM MS. Analyzed the data: RMES HJMMM EEB. Contributed reagents/materials/analysis tools: RMES HJMMM EEB. Wrote the paper: RMES HJMMM EEB MJPvO CBLMM AvdL AJN MS.
- 1. Demeter E, Hernandez-Garcia L, Sarter M, Lustig C. Challenges to attention: a continuous arterial spin labeling (ASL) study of the effects of distraction on sustained attention. Neuroimage. 2011;54: 1518–1529. pmid:20851189
- 2. Fernandez-Seara MA, Wang J, Wang Z, Korczykowski M, Guenther M, Feinberg DA, et al. Imaging mesial temporal lobe activation during scene encoding: comparison of fMRI using BOLD and arterial spin labeling. Hum Brain Mapp. 2007;28: 1391–1400. pmid:17525983
- 3. Kemeny S, Ye FQ, Birn R, Braun AR. Comparison of continuous overt speech fMRI using BOLD and arterial spin labeling. Hum Brain Mapp. 2005;24: 173–183. pmid:15486986
- 4. Cavusoglu M, Bartels A, Yesilyurt B, Uludag K. Retinotopic maps and hemodynamic delays in the human visual cortex measured using arterial spin labeling. Neuroimage. 2012;59: 4044–4054. pmid:22040735
- 5. Wang J, Aguirre GK, Kimberg DY, Roc AC, Li L, Detre JA. Arterial spin labeling perfusion fMRI with very low task frequency. Magn Reson Med. 2003;49: 796–802. pmid:12704760
- 6. Gaxiola-Valdez I, Goodyear BG. Origins of intersubject variability of blood oxygenation level dependent and arterial spin labeling fMRI: implications for quantification of brain activity. Magn Reson Imaging. 2012;30: 1394–1400. pmid:22795932
- 7. Pimentel MA, Vilela P, Sousa I, Figueiredo P. Localization of the hand motor area by arterial spin labeling and blood oxygen level-dependent functional magnetic resonance imaging. Hum Brain Mapp. 2013;34: 96–108. pmid:22121040
- 8. Ogawa S, Menon RS, Tank DW, Kim SG, Merkle H, Ellermann JM, et al. Functional brain mapping by blood oxygenation level-dependent contrast magnetic resonance imaging. A comparison of signal characteristics with a biophysical model. Biophys J. 1993;64: 803–812. pmid:8386018
- 9. Obata T, Liu TT, Miller KL, Luh WM, Wong EC, Frank LR, et al. Discrepancies between BOLD and flow dynamics in primary and supplementary motor areas: application of the balloon model to the interpretation of BOLD transients. Neuroimage. 2004;21: 144–153. pmid:14741651
- 10. Alsop DC, Detre JA, Golay X, Günther M, Hendrikse J, Hernandez-Garcia L, et al. Recommended implementation of arterial spin-labeled perfusion MRI for clinical applications: A consensus of the ISMRM perfusion study group and the European consortium for ASL in dementia. Magn Res Med. 2015;73: 102–116.
- 11. Wang Y, Saykin AJ, Pfeuffer J, Lin C, Mosier KM, Shen L, et al. Regional reproducibility of pulsed arterial spin labeling perfusion imaging at 3T. Neuroimage. 2011;54: 1188–1195. pmid:20800097
- 12. Parkes LM, Rashid W, Chard DT, Tofts PS. Normal cerebral perfusion measurements using arterial spin labeling: reproducibility, stability, and age and gender effects. Magn Reson Med. 2004;51: 736–743. pmid:15065246
- 13. Hermes M, Hagemann D, Britz P, Lieser S, Rock J, Naumann E, et al. Reproducibility of continuous arterial spin labeling perfusion MRI after 7 weeks. MAGMA. 2007;20: 103–115. pmid:17429703
- 14. Pfefferbaum A, Chanraud S, Pitel AL, Shankaranarayanan A, Alsop DC, Rohlfing T, et al. Volumetric cerebral perfusion imaging in healthy adults: regional distribution, laterality, and repeatability of pulsed continuous arterial spin labeling (PCASL). Psychiatry Res. 2010;182: 266–73. pmid:20488671
- 15. Xu G, Rowley HA, Wu G, Alsop DC, Shankaranarayanan A, Dowling M, et al. Reliability and precision of pseudo-continuous arterial spin labeling perfusion MRI on 3.0 T and comparison with 15O-water PET in elderly subjects at risk for Alzheimer's disease. NMR Biomed. 2010;23: 286–293. pmid:19953503
- 16. Wu B, Lou X, Wu X, Ma L. Intra- and interscanner reliability and reproducibility of 3D whole-brain pseudo-continuous arterial spin-labeling MR perfusion at 3T. J Magn Reson Imaging. 2014;39: 402–409. pmid:23723043
- 17. Yen YF, Field AS, Martin EM, Ari N, Burdette JH, Moody DM, et al. Test-retest reproducibility of quantitative CBF measurements using FAIR perfusion MRI and acetazolamide challenge. Magn Reson Med. 2002;47: 921–928. pmid:11979571
- 18. Jahng GH, Song E, Zhu XP, Matson GB, Weiner MW, Schuff N. Human brain: reliability and reproducibility of pulsed arterial spin-labeling perfusion MR imaging. Radiology. 2005;234: 909–916. pmid:15734942
- 19. Petersen ET, Mouridsen K, Golay X, all named co-authors of the QUASAR test-retest study. The QUASAR reproducibility study, Part II: Results from a multi-center Arterial Spin Labeling test-retest study. Neuroimage. 2010;49: 104–113. pmid:19660557
- 20. Floyd TF, Ratcliffe SJ, Wang J, Resch B, Detre JA. Precision of the CASL-perfusion MRI technique for the measurement of cerebral blood flow in whole brain and vascular territories. J Magn Reson Imaging. 2003;18: 649–655. pmid:14635149
- 21. Gevers S, Majoie CB, van den Tweel XW, Lavini C, Nederveen AJ. Acquisition time and reproducibility of continuous arterial spin-labeling perfusion imaging at 3T. AJNR Am J Neuroradiol. 2009;30: 968–971. pmid:19193760
- 22. Wu WC, Jiang SF, Yang SC, Lien SH. Pseudocontinuous arterial spin labeling perfusion magnetic resonance imaging—a normative study of reproducibility in the human brain. Neuroimage. 2011;56: 1244–1250. pmid:21385620
- 23. Chen Y, Wang DJ, Detre JA. Test-retest reliability of arterial spin labeling with common labeling strategies. J Magn Reson Imaging. 2011;33: 940–949. pmid:21448961
- 24. Gevers S, van Osch MJ, Bokkers RP, Kies DA, Teeuwisse WM, Majoie CB, et al. Intra- and multicenter reproducibility of pulsed, continuous and pseudo-continuous arterial spin labeling methods for measuring cerebral perfusion. J Cereb Blood Flow Metab. 2011;31: 1706–1715. pmid:21304555
- 25. Mutsaerts HJ, Steketee RME, Heijtel DF, Kuijer JP, van Osch MJ, Majoie CB, et al. Inter-vendor reproducibility of pseudo-continuous arterial spin labeling at 3 tesla. PLoS One. 2014;9: e104108. pmid:25090654
- 26. Bron EE, Steketee RME, Houston GC, Oliver RA, Achterberg HC, Loog M, et al. Diagnostic classification of arterial spin labeling and structural MRI in presenile early stage dementia. Hum Brain Mapp. 2014;35: 4916–4931. pmid:24700485
- 27. Ashburner J, Friston KJ. Unified segmentation. Neuroimage. 2005;26: 839–851. pmid:15955494
- 28. Klein S, Staring M, Murphy K, Viergever MA, Pluim JP. Elastix: a Toolbox for Intensity-Based Medical Image Registration. IEEE Trans Med Imaging. 2010;29: 196–205. pmid:19923044
- 29. Hammers A, Allom R, Koepp MJ, Free SL, Myers R, Lemieux L, et al. Three-dimensional maximum probability atlas of the human brain, with particular reference to the temporal lobe. Hum Brain Mapp. 2003;19: 224–247. pmid:12874777
- 30. Gousias IS, Rueckert D, Heckemann RA, Dyet LE, Boardman JP, Edwards AD, et al. Automatic segmentation of brain MRIs of 2-year-olds into 83 regions of interest. Neuroimage. 2008;40: 672–684. pmid:18234511
- 31. Yousry TA, Schmid UD, Alkadhi H, Schmidt D, Peraud A, Buettner A, et al. Localization of the motor hand area to a knob on the precentral gyrus. A new landmark. Brain. 1997;120 (Pt 1): 141–157. pmid:9055804
- 32. Shrout PE, Fleiss JL. Intraclass correlations: uses in assessing rater reliability. Psychol Bull. 1979;86: 420–428. pmid:18839484
- 33. Mildner T, Trampel R, Moller HE, Schafer A, Wiggins CJ, Norris DG. Functional perfusion imaging using continuous arterial spin labeling with separate labeling and imaging coils at 3 T. Magn Reson Med. 2003;49: 791–795. pmid:12704759
- 34. Ye FQ, Yang Y, Duyn J, Mattay VS, Frank JA, Weinberger DR, et al. Quantitation of regional cerebral blood flow increases during motor activation: A multislice, steady-state, arterial spin tagging study. Magn Reson Med. 1999;42: 404–407. pmid:10440966
- 35. Raoult H, Ferre JC, Petr J, Bannier E, Stamm A, Barillot C, et al. Functional arterial spin labeling: Optimal sequence duration for motor activation mapping in clinical practice. J Magn Reson Imaging. 2012;36: 1435–1444. pmid:22926803
- 36. Klomp A, Caan MW, Denys D, Nederveen AJ, Reneman L. Feasibility of ASL-based phMRI with a single dose of oral citalopram for repeated assessment of serotonin function. Neuroimage. 2012;63: 1695–1700. pmid:22842212
- 37. Gonzalez-At JB, Alsop DC, Detre JA. Cerebral perfusion and arterial transit time changes during task activation determined with continuous arterial spin labeling. Magn Reson Med. 2000;43: 739–746. pmid:10800040
- 38. Vidorreta M, Wang Z, Rodriguez I, Pastor MA, Detre JA, Fernandez-Seara MA. Comparison of 2D and 3D single-shot ASL perfusion fMRI sequences. Neuroimage. 2012;66C: 662–671.
- 39. Hu Y, Glover GH. Three-dimensional spiral technique for high-resolution functional MRI. Magn Reson Med. 2007;58: 947–951. pmid:17969117
- 40. Alsop DC, Detre JA. Reduced transit-time sensitivity in noninvasive magnetic resonance imaging of human cerebral blood flow. J Cereb Blood Flow Metab. 1996;16: 1236–1249. pmid:8898697
- 41. van Osch MJ, Hendrikse J, van der Grond J. Sensitivity comparison of multiple vs. single inversion time pulsed arterial spin labeling fMRI. J Magn Reson Imaging. 2007;25: 215–221. pmid:17154371
- 42. Tjandra T, Brooks JC, Figueiredo P, Wise R, Matthews PM, Tracey I. Quantitative assessment of the reproducibility of functional activation measured with BOLD and MR perfusion imaging: implications for clinical trial design. Neuroimage. 2005;27: 393–401. pmid:15921936
- 43. Raoult H, Petr J, Bannier E, Stamm A, Gauvrit JY, Barillot C, et al. Arterial spin labeling for motor activation mapping at 3T with a 32-channel coil: reproducibility and spatial accuracy in comparison with BOLD fMRI. Neuroimage. 2011;58: 157–167. pmid:21689761
- 44. Aguirre GK, Detre JA, Zarahn E, Alsop DC. Experimental design and the relative sensitivity of BOLD and perfusion fMRI. Neuroimage. 2002;15: 488–500. pmid:11848692
- 45. St Lawrence KS, Wang J. Effects of the apparent transverse relaxation time on cerebral blood flow measurements obtained by arterial spin labeling. Magn Reson Med. 2005;53: 425–433. pmid:15678532
- 46. Heijtel DF, Mutsaerts HJ, Bakker E, Schober P, Stevens MF, Petersen ET, et al. Accuracy and precision of pseudo-continuous arterial spin labeling perfusion during baseline and hypercapnia: A head-to-head comparison with 15O H2O positron emission tomography. Neuroimage. 2014;92C: 182–192.
- 47. Lu H, Clingman C, Golay X, van Zijl PC. Determining the longitudinal relaxation time (T1) of blood at 3.0 Tesla. Magn Reson Med. 2004;52: 679–682. pmid:15334591
- 48. Aslan S, Xu F, Wang PL, Uh J, Yezhuvath US, van Osch M, et al. Estimation of labeling efficiency in pseudocontinuous arterial spin labeling. Magn Reson Med. 2010;63: 765–771. pmid:20187183
- 49. Garcia DM, Duhamel G, Alsop DC. Efficiency of inversion pulses for background suppressed arterial spin labeling. Magn Reson Med. 2005;54: 366–372. pmid:16032674