The consequences of phrenic nerve paralysis vary from a considerable reduction in respiratory function to an apparently normal state. Acoustic analysis of lung sound intensity (LSI) could be an indirect non-invasive measurement of respiratory muscle function, comparing activity on the two sides of the thoracic cage. Lung sounds and airflow were recorded in ten males with unilateral phrenic paralysis and ten healthy subjects (5 men/5 women), during progressive increasing airflow maneuvers. Subjects were in sitting position and two acoustic sensors were placed on their back, on the left and right sides. LSI was determined from 1.2 to 2.4 L/s between 70 and 2000 Hz. LSI was significantly greater on the normal (19.3±4.0 dB) than the affected (5.7±3.5 dB) side in all patients (p = 0.0002), differences ranging from 9.9 to 21.3 dB (13.5±3.5 dB). In the healthy subjects, the LSI was similar on both left (15.1±6.3 dB) and right (17.4±5.7 dB) sides (p = 0.2730), differences ranging from 0.4 to 4.6 dB (2.3±1.6 dB). There was a positive linear relationship between the LSI and the airflow, with clear differences between the slope of patients (about 5 dB/L/s) and healthy subjects (about 10 dB/L/s). Furthermore, the LSI from the affected side of patients was close to the background noise level, at low airflows. As the airflow increases, the LSI from the affected side did also increase, but never reached the levels seen in healthy subjects. Moreover, the difference in LSI between healthy and paralyzed sides was higher in patients with lower FEV1 (%). The acoustic analysis of LSI is a relevant non-invasive technique to assess respiratory function. This method could reinforce the reliability of the diagnosis of unilateral phrenic paralysis, as well as the monitoring of these patients.
Citation: Fiz JA, Jané R, Lozano M, Gómez R, Ruiz J (2014) Detecting Unilateral Phrenic Paralysis by Acoustic Respiratory Analysis. PLoS ONE 9(4): e93595. https://doi.org/10.1371/journal.pone.0093595
Editor: Thomas Penzel, Charité - Universitätsmedizin Berlin, Germany
Received: August 8, 2013; Accepted: March 7, 2014; Published: April 9, 2014
Copyright: © 2014 Fiz et al. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Funding: This work was made possible thanks to a collaboration agreement, between IBEC and IGTP, to create a joint research Unit, and it was supported in part by the Spanish Ministry of Economy and Competitiveness under grant TEC2010-21703-C03-01. No additional external funding received for this study. The funders had no role in study design, data collection and analysis, decision to publish, or preparation of the manuscript.
Competing interests: The authors have declared that no competing interests exist.
There are several causes of diaphragmatic dysfunction that can affect one or both muscles. The decrease in or cessation of motor activity can be caused by compression or section of the phrenic nerve in certain segments of the spinal cord . The consequences of diaphragm dysfunction vary from the most serious cases of bilateral lesions that can require mechanical ventilation, to the mildest unilateral lesions that may to some extent impair breathing and in consequence exercise capacity , .
Diaphragm dysfunction due to phrenic paralysis has been studied with various techniques including x-ray, fluoroscopy, ultrasonography, and external or internal stimuli of the diaphragm. These techniques provide information regarding the position and mobility of the diaphragm muscle –, but do not predict the degree of respiratory dysfunction .
On the contrary, breathing function can be measured by routine spirometry , . Recently, Sokolowska et al. measured variations in breathing patterns in animals with bilateral phrenic paralysis, confirming that the measurement of breathing parameters could be an appropriate method to monitor this diaphragm dysfunction . However, in cases of unilateral paralysis, spirometric function may be normal.
An alternative useful method to monitor breathing function is the measurement of pulmonary sounds –. In fact, it is known that airflow is correlated with lung sound intensity (LSI) , including in pulmonary conditions with restrictive ventilatory function .
Our hypothesis in the present study was that in patients with unilateral phrenic paralysis, the LSI on inspiration would be lower on the affected side than the healthy side. If this hypothesis were to be confirmed, measurements of LSI comparing the two sides could be useful to diagnose conditions associated with restricted thoracic mobility , as well as to monitor the response to specific physiotherapy treatments targeting the respiratory muscle.
Materials and Methods
The study was conducted in the Respiratory Function Laboratory at HUGTIP, since February 2011 to December 2013, and approved by the Human Research and Ethics Committee of the hospital. All participants gave written informed consent, following the World Medical Association's Declaration of Helsinki on Ethical Principles for Medical Research Involving Human Subjects.
Patients with unilateral phrenic paralysis , who were previously diagnosed in the Department of Internal Medicine at Germans Trias i Pujol University Hospital (HUGTIP), were considered eligible for this study. All patients underwent chest radiography and computed tomography scanning of the chest, which reveal elevated hemidiaphragm on the affected side. Moreover, according to their medical history, most of the patients had previous thoracic or surgical trauma as the major cause of diaphragmatic paralysis. Only patient ID 2 had an unknown etiology. However, all patients related some level of functional dyspnea.
On the other hand, controls were selected from healthy subjects who had never been diagnosed of phrenic paralysis and had normal baseline spirometric values. According to these inclusion criteria, ten men with unilateral phrenic paralysis in a stable condition and ten controls (five men/five women) were included in the study for pulmonary function test and the acoustic respiratory analysis.
Pulmonary function and lung sound testing
At baseline, lung function was measured by spirometry (Hyp'Air Compact, Medisoft). Table 1 shows baseline spirometric results from each subject. Measurements were obtained in accordance with established guidelines , and results compared to reference values .
After this previous test, each subject was coached to progressively increase the airflow from shallow breathing to the deepest breaths they were able to, reaching 1.2 to 2.4 L/s . Lastly, at the end of the respiratory test, subjects were asked to hold their breath for a few seconds in order to estimate background noise intensity (BNI). One recording of a total 120 seconds was obtained from each subject in a sitting position. Respiratory flow and sounds were acquired simultaneously during the test.
Lung sounds and respiratory airflow measurements
Respiratory sounds were recorded using two contact microphones (TSD108, Biopac Systems, Inc.) with a frequency response of 35–3500 Hz. Microphones were positioned on the surface of the back, at each side of the spinal cord and 3 cm below the bottom tip of the shoulder blades. They were attached to the skin with double-sided adhesive discs, in a noninvasive way. In addition, respiratory airflow was recorded with a pneumotachograph (TSD107B, Biopac Systems, Inc.). Subjects wore a nose clip and breathed through the mouthpiece of the instrument.
Airflow and sound signals were amplified and filtered by hardware, before analog-to-digital conversion and acquisition. On the one hand, high- and low-pass filters with cut-off frequencies of 10 and 5000 Hz, respectively, were applied to respiratory sound signals, and they were amplified by a factor of 200. On the other hand, low-pass filter with a cut-off frequency of 10 Hz was applied to the airflow signal, and this was amplified by a factor of 1000. Then, both sound and flow signals were recorded at a sample rate of 12500 Hz using a 16-bit analog-to-digital converter (MP150, Biopac Systems, Inc.). Since this study is only focused on normal pulmonary sounds, whose bandwidth of interest is below 2000 Hz, respiratory sound signals were digitally filtered using a combination of 8th order Butterworth high- and low-pass filters with cut-off frequencies of 70 and 2000 Hz, respectively.
Lung sound analysis
Respiratory sound signals were automatically segmented by extracting respiratory phases from the airflow signal. Respiratory cycles in which the flow reached at least 0.35 L/s were considered valid cycles. In order to avoid false detections caused by background noise, two thresholds of 0.2 and 4 seconds were established for minimum and maximum durations of breathing phases, respectively, according to time duration of normal respiratory cycles. In addition, a threshold of 0.5 seconds was fixed for the maximum time interval between the end of inspiration and the beginning of the corresponding expiration. All cycles not meeting these criteria were rejected. The final dataset for each subject was formed by audio-visual selection of pairs of sound signals, one from each side, from the same inspiratory cycles, avoiding artifacts such as those from swallowing or rubbing.
Each inspiratory sound cycle was firstly classified according to the maximum airflow reached. For that purpose, the airflow scale was divided into intervals of 0.2 L/s, from 1.2 L/s upwards. Furthermore, only inspiratory sound segments corresponding to the top airflow interval, whose duration is at least 20% of cycle length, of each inspiratory cycle were used for assessing the LSI.
The LSI was calculated as the mean power, in the frequency band from 70 to 2000 Hz, obtained from the power spectral density (PSD) of each inspiratory sound segment, according to the following expression:where fm is the sample rate, and NFFT is the number of points for the fast Fourier transform (FFT). Just as in some previous studies, which were focused on the intensity of respiratory sounds , ,  the PSD was calculated using Welch's periodogram, with a Hanning window of 1000 data samples (80 ms), a 50% overlap between adjacent segments, and 1024 points for the FFT. The same method was applied to apnea segments from both left and right sides, in order to calculate the mean background noise intensity (). The resultant LSI values from all inspiratory sound segments were expressed in dB with respect to this . Having calculated the LSI, each subject was characterized by the relationship between the LSI and the airflow on both left and right sides. In addition, the LSI was averaged over the airflow range 1.2–2.4 L/s, in order to obtain a mean LSI for each side. Normality in the mean LSIs of both sides, from patients and healthy subjects, as well as in their differences was tested with a Lilliefors test. Since we did not know the parameters of the hypothesized distributions, and those parameters must be estimated from the data sample, the Lilliefors test was preferable. On the other hand, the statistical differences were tested between: 1) the mean LSIs of both sides, and 2) the differences in the mean LSIs of both sides from patients and healthy subjects. Since normality could not be assumed in all cases, and the sample size (n = 20) was small, a non-parametric test, such as the Wilcoxon rank sum test, was used to check for statistical differences.
Lung sound intensity in unilateral phrenic paralysis
Acoustic and spirometric parameters were analyzed in patients and healthy subjects. As shown in Table 1, eight patients had left side paralysis (ID 2–7, 9, and 10) and two patients had right side paralysis (ID 1 and 8). Regardless of the side affected, all patients had lower FVC (57±9%) and FEV1 (57±10%) values than healthy subjects, in whom the percentages were 94±11% and 93±7%, respectively.
With regard to lung sounds, the signal amplitude was much lower on the paralyzed side than the healthy side, in patients with unilateral phrenic nerve paralysis, as shown in the example from Figure 1. It contains the lung sound and the airflow signals from a patient with left phrenic paralysis (ID 4). Accordingly, the magnitude of the PSDs from both sides, and the consequent signal powers, are quite different, as shown in Figure 2. It exhibits the PSDs from two inspiratory sound segments, one from each side, of an inspiratory cycle from patient ID 4. As shown, the PSD of the right sound segment (healthy side) is a long way from the PSD of the right background noise segment, in all the frequency range. On the other hand, the PSD of the left sound segment (affected side) is slightly above the left background noise. As a result, the LSI calculated from the PSD of the healthy side is much larger than the affected side.
Airflow signal (black) and the corresponding lung sound signals, in arbitrary units, for both right (blue) and left (red) sides, in a patient with left side phrenic paralysis (ID 4). Sound amplitudes from the left side were lower than those from the healthy right side.
Airflow (L/s), lung sound signals (arbitrary units), and the corresponding power spectral densities (dBW), for both sides of an inspiratory cycle from a patient with left side phrenic paralysis (ID 4). Solid and dotted lines in the PSDs correspond to the central sound segments and the background noise segments from both sides, respectively.
The aforementioned pattern was confirmed by comparison of the acoustic parameters in all patients with unilateral phrenic paralysis (Table 2). Calculation of the from the BNI for left and right sides allowed us to express the mean LSI from each side in dB with respect to the same reference value. In addition, the mean LSI was calculated for both sides from the same set of cycles.
In healthy subjects, the mean LSI was much higher than the on both left (15.1±6.2 dB) and right (17.4±5.7 dB) sides. However, patients had mean LSIs only a few dBs above the on the affected side (5.7±3.5 dB) while their mean LSIs on the healthy side (19.3±4.0 dB) were not significantly different from the values measured in the healthy participants. To show this trend clearly, we calculated the difference between the mean LSI of each side.
Figure 3 shows the mean LSI, for each side, as a function of airflow level in all patients, and all healthy subjects. On the one hand, considerable differences, of more than 13 dB, can be seen between the LSI from the affected and healthy sides. On the other hand, differences in LSI between the sides are less than 3 dBs in healthy subjects. It should be noted that the LSI from the affected sides are close to the (0 dB) at low airflows. As the airflow increases, the LSI from the affected sides does also increase, but never reaches the levels seen in healthy subjects.
Mean inspiratory LSI (dB) as a function of airflow (L/s), from the ten patients and the ten healthy subjects. All values are the mean ± SD.
Furthermore, Figure 3 shows a clear linear relationship between the LSI and the airflow level. This sound-flow relationship has been reported in some previous studies –, and it usually follows a power law. In a logarithmic scale (dB), this relationship can be formulated by a linear equation:where m is the slope of the line, and b is the y-intercept. As shown in Table 3, all LSI-airflow relationships from Figure 3 can be properly expressed by a linear equation. Moreover, there is a clear difference between the slope of healthy subjects (around 5 dB/L/s) and patients (around 10 dB/L/s), independently of the analyzed side.
The mean inspiratory LSI from both sides of patients and healthy subjects has been statistically analyzed, as shown in Figure 4. The null hypothesis that the mean LSIs were normally distributed was accepted as much for both healthy and paralyzed sides in patients (p = 0.4135 and 0.9436, respectively), as for the right side in healthy subjects (p = 0.5790). However, the null hypothesis was rejected for the left side in healthy subjects (p = 0.0104).
Mean inspiratory LSI (dB) in healthy and paralyzed sides (ten patients), and in right and left sides (ten healthy subjects). The mean LSI from patients is significantly higher in healthy side than paralyzed side (p = 0.0002). On the contrary, there are not significant differences between mean LSI from both hemithoraxes in healthy subjects (p = 0.2730).
The Wilcoxon rank sum test showed that the mean LSIs of healthy and paralyzed sides in all patients were statistically different (p = 0.0002). On the contrary, the difference between the mean LSIs of right and left sides in all healthy subjects was not statistically significant (p = 0.2730).
Lung sound intensity differences and FEV1 relationship
Figure 5-A shows the absolute value of the differences between the mean LSIs of both sides (|LSILeft-LSIright|). In this case, the null hypothesis that the differences were normally distributed was accepted as much in patients (p = 0.6078), as in healthy subjects (p = 0.4693).
A: Mean inspiratory LSI difference (dB) between both hemithoraxes, in ten patients with phrenic nerve paralysis and ten healthy subjects. The LSI difference was higher in patients than healthy subjects (p = 0.0002). Solid lines indicate the mean and SD for each group. B: Mean inspiratory LSI difference as a function of FEV1 in patients and healthy subjects.
The Wilcoxon rank sum test showed that the differences between the mean LSIs of both sides were statistically significant in both groups (p = 0.0002). Moreover, it was found that there was a clear cut off around 6–8 dB which distinguished patients from healthy subjects.
Figure 5-B illustrates the relationship between the mean LSI difference and FEV1, showing high differences in the LSI and low FEV1 in patients with phrenic nerve paralysis. Moreover, in patients there is an inverse relationship between the two parameters, namely the lower the FEV1, the higher the mean LSI difference. In contrast, healthy subjects have low mean LSI differences, and there is no any clear relationship between these LSI differences and the corresponding FEV1.
Discussion and Conclusions
Our study shows that patients with unilateral phrenic nerve paralysis have a lower inspiratory sound intensity on the affected side than the healthy one. We did not analyze expiratory sounds due to the lower values of expiratory intensity with respect to inspiratory values at isoflows . This study illustrates the potential of lung acoustic analysis for the diagnosis and management of these patients.
Respiratory sounds are an alternative method to measure both pulmonary  and diaphragmatic function. Some previous studies reported decreased breath sound on the affected side in patients with unilateral phrenic paralysis , , , but they were assessed by traditional auscultation. However, there are no references about quantitative analysis of respiratory sounds for the diagnosis of these patients. In what is related to laterality of respiratory sounds, they have been used to distinguish between bilateral and unilateral lung ventilation in intubated patients . Nevertheless, many studies have analyzed the differences between the LSI of both sides in healthy subjects , , , , thus reporting slight differences of a few dB. In any case, sound analysis can detect differences in airflow entering the two sides of the thoracic cage in diseases that affect respiratory ventilation, and our study demonstrates this for the case of unilateral phrenic nerve paralysis. Consistently, we found a clear cut-off in the mean differences of LSI between the two sides in healthy subjects and patients.
In addition to unilateral phrenic paralysis, it has been recognized by other authors that lung sound analysis is also a very useful technique to study many others pulmonary diseases , , , .
When the diaphragm is paralyzed, it does not have an influence on expansion of the homolateral lung and breathing is maintained by accessory muscles such as those of the chest wall. The movement of the paralyzed hemidiaphragm is determined by the balance between the change in pleural pressure and the shortening of the healthy hemidiaphragm. This is manifested by a cranial displacement of the ipsilateral hemidiaphragm and a small caudal displacement of the contralateral hemidiaphragm . Such a retraction is ineffective for respiration and has been related to patient dyspnea .
It has been suggested that the airflow to dependent areas of the lung is directed by the diaphragm and non-dependent areas by the intercostal muscles . The gas flow to the dependent areas of the paralytic side would therefore be lower than that to the healthy side.
In this study, there was considerably less airflow entering the dependent areas of the pathological side, measured in an indirect way by the quantification of the LSI. Specifically, the LSI of the affected side was close to the level of the background noise for a low airflow rate, as seen in Figure 3, while the signal from the healthy side remained a long way from the background noise at all measured flow rates. Although the BNIs from both sides are slightly different, it is not relevant for the results of this study, since the is used as the unique reference value in order to express the LSI in dB.
In addition, pulmonary perfusion is redistributed from the base toward the apex in these patients . The result of this pathological situation is that the work of breathing (measured in terms of oxygen uptake) is increased, which suggests that intercostal muscle breathing is less efficient than diaphragm breathing . Spirometric changes have been widely commented on in the literature. In our study, spirometry values of patients with phrenic paralysis were low with respect to normal reference values, as has been found previously in other studies , .
With respect to traditional techniques to diagnose the unilateral phrenic paralysis, they include: x-ray imaging, fluoroscopy, ultrasonography, and phrenic nerve stimulation , . Usually, unilateral phrenic paralysis is diagnosed by a combination of these techniques, since none of them is totally concluding by itself. Of these techniques, x-ray imaging is the simplest and it has some obvious limitations: it uses ionizing radiation, and it does not allow us to assess the diaphragm or the pulmonary function of patients. Moreover, in unilateral diaphragmatic paralysis, the sensitivity of plain chest radiograph is as high as 90%, whereas its specificity is, however, low (44%) .
Fluoroscopy and the external or internal stimuli of the diaphragm allow evaluating the diaphragm mobility . However, fluoroscopy also makes use of x-rays to obtain dynamic images of the diaphragm, and both fluoroscopy and stimuli of the diaphragm are invasive techniques. Moreover, none of these methods provides information about the pulmonary function.
Ultrasonography is an alternative non-invasive technique to assess the diaphragmatic function , , since it works on ultrasounds. Nevertheless, just as the aforementioned techniques, ultrasounds do not provide any data about the pulmonary function of patients. Moreover, ultrasonography is operator dependent and requires significant expertise .
Recently, a new non-invasive method has been proposed to measure the movements of the thoracic wall . This new method makes use of a motion analysis system, which is called optoelectronic plethysmography. It was used to estimate the total rib cage volume, as well as its changes in both healthy and paralyzed sides.
In this study, the potential of acoustic respiratory analysis for detecting unilateral phrenic paralysis has been clearly shown. Despite a relatively small population has been analyzed, the results from 20 subjects (10 patients and 10 healthy subjects) reinforce the reliability of the proposed method. On the other hand, in the database, there is a slight difference in the male-female ratio between patients and healthy subjects, but gender is not a relevant factor in the analysis of normal lung sound intensity . However, further studies will be needed to clinically validate this technique as a new complementary tool for phrenic paralysis diagnosis.
In conclusion, measurement of LSI can provide quantitative information about the extent of impairment of respiratory function in patients with unilateral phrenic nerve paralysis. In these patients, LSI is an indirect measure of the airflow that enters the lungs, this being lower on the affected side due to inefficient diaphragmatic muscle function. This technique represent a step forward in the diagnostic procedure of unilateral phrenic nerve paralysis, since it has some advantages with respect to current techniques: non-invasiveness, objectivity, simplicity, easiness and cost. The acoustic respiratory analysis, in conjunction with spirometry, could reinforce the reliability of the diagnosis of unilateral phrenic paralysis.
Regarding the future use of the method, its major application is the non-invasive assessment of respiratory function, providing objective information of the affected side. Therefore, the method offers the capability for long-term monitoring of recovery in respiratory function in patients who undergo physical therapy . These patients are regularly monitored in order to check whether the physical therapy is improving their pulmonary function in the affected side or not. In this context, the advantages of the proposed technique gain relevance since several and repeated tests are required for the long-term monitoring of these patients.
All authors would like to thank all the team in the Pulmonary Function Testing Laboratory at Germans Trias i Pujol University Hospital, for their collaboration in the patient recruitment as well as the performing of the spirometry tests.
Conceived and designed the experiments: JAF RJ ML JR. Performed the experiments: JAF ML RG. Analyzed the data: JAF RJ ML. Contributed reagents/materials/analysis tools: JAF RJ ML. Wrote the paper: JAF RJ ML. Interpreted results of experiments: JAF RJ ML.
- 1. DeVita MA, Robinson LR, Rehder J, Hattler B, Cohen C (1993) Incidence and natural history of phrenic neuropathy occurring during open heart surgery. Chest 103(3): 850–856.
- 2. Brochard L, Harf A, Lorino H, Lemaire F (1989) Inspiratory pressure support prevents diaphragmatic fatigue during weaning from mechanical ventilation. Am Rev Respir Dis 139(2): 513–521.
- 3. Freilich S, Janssen JC, Polkey MI (2011) An unusual case of diaphragm paralysis. Thorax 66(2): 133.
- 4. Yi LC, Nascimento OA, Jardim JR (2011) Reliability of an analysis method for measuring diaphragm excursion by means of direct visualization with videofluoroscopy. Arch Bronconeumol 47(6): 310–314.
- 5. Kim SH, Na S, Choi JS, Na SH, Shin S, et al. (2010) An evaluation of diaphragmatic movement by M-code sonography as a predictor of pulmonary dysfunction after upper abdominal surgery. Anesth Analq 110(5): 1349–1354.
- 6. Scillia P, Cappello M, De Troyer A (2004) Determinants of diaphragm motion in unilateral diaphragmatic paralysis. J Appl Physiol 96(1): 96–100.
- 7. Watson AC, Hughes PD, Louise Harris M, Hart N, Ware RJ, et al. (2001) Measurement of twitch transdiaphragmatic, esophageal, and endotracheal tube pressure with bilateral anterolateral magnetic phrenic nerve stimulation in patients in the intensive care unit. Crit Care Med 29(7): 1325–1331.
- 8. Chetta A, Rehman AK, Moxham J, Carr DH, Polkey MI (2005) Chest radiography cannot predict diaphragm function. Respir Med 99(1): 39–44.
- 9. Lisboa C, Paré PD, Pertuzé J, Contreras G, Moreno R, et al. (1986) Inspiratory muscle function in unilateral diaphragmatic paralysis. Am Rev Respir Dis 134(3): 488–492.
- 10. Epstein SW, Vanderlinden RG, Man SF, Hyland RH, Lenkei SC, et al. (1979) Lung function in diaphragm pacing. Can Med Assoc J 120(11): 1360–1368.
- 11. Sokolowska B, Józwik A, Pokorski M (2003) A fuzzy-classifier system to distinguish respiratory patterns evolving after diaphragm paralysis in the cat. Jpn J Physiol 53(4): 301–307.
- 12. Pasterkamp H, Patel S, Wodicka GR (1997) Asymmetry of respiratory sounds and thoracic transmission. Med Biol Eng Comput 35(2): 103–106.
- 13. Hult P, Wranne B, Ask P (2000) A bioacoustic method for timing of the different phases of the breathing cycle and monitoring of breathing frequency. Med Eng Phys 22(6): 425–433.
- 14. Fiz JA, Gnitecki J, Kraman SS, Wodicka GR, Pasterkamp H (2008) Effect of body position on lung sounds in healthy young men. Chest 133(3): 729–736.
- 15. Alshaer H, Fernie GR, Bradley TD (2011) Monitoring of breathing phases using a bioacoustic method in healthy awake subjects. J Clin Monit Comput 25(5): 285–294.
- 16. Morice RC, Jimenez CA, Eapen GA, Mehran RJ, Keus L, et al. (2010) Using quantitative breath sound measurements to predict lung function following resection. J Cardiothorac Surg 5: 81.
- 17. Shykoff BE, Ploysongsang Y, Chang HK (1988) Airflow and normal lung sounds. Am Rev Respir Dis 137(4): 872–876.
- 18. Gibson GJ (1989) Diaphragmatic paresis: pathophysiology, clinical features, and investigation. Thorax 44(11): 960–970.
- 19. Miller MR, Hankinson J, Brusasco V, Burgos F, Casaburi R, et al. (2005) Standardisation of spirometry. Eur Respir J 26(2): 319–338.
- 20. Roca J, Burgos F, Sunyer J, Saez M, Chinn S, et al. (1998) References values for forced spirometry. Group of the European Community Respiratory Health Survey. Eur Respir J 11(6): 1354–1362.
- 21. Oud M, Dooijes EH, Van Der Zee JS (2000) Asthmatic airways obstruction assessment based on detailed analysis of respiratory sound spectra. IEEE Trans Biomed Eng 47(11): 1450–5.
- 22. Montazeri A, Giannouli E, Moussavi Z (2012) Assessment of obstructive sleep apnea and its severity during wakefulness. Ann Biomed Eng 40(4): 916–24.
- 23. Hossain I, Moussavi Z (2004) Finding the lung sound-flow relationship in normal and asthmatic subjects. Conf Proc IEEE Eng Med Biol Soc 5: 3852–5.
- 24. Golabbakhsh M, Moussavi Z (2004) Relationship between airflow and frequency-based features of tracheal respiratory sound. Can Conf Electr Comput Eng 2: 751–754.
- 25. Yadollahi A, Montazeri A, Azarbarzin A, Moussavi Z (2013) Respiratory flow-sound relationship during both wakefulness and sleep and its variation in relation to sleep apnea. Ann Biomed Eng 41(3): 537–46.
- 26. Dernaika TA, Younis WG, Carlile PV (2008) Spontaneous recovery in idiopathic unilateral diaphragmatic paralysis. Respir Care 53(3): 351–4.
- 27. Habib GS (2012) Unilateral diaphragm paralysis following vaccination. J Med Cases 3(3): 164–166.
- 28. Tejman-Yarden S, Lederman D, Eilig I, Zlotnik A, Weksler N, et al. (2006) Acoustic monitoring of double-lumen ventilated lungs for the detection of selective unilateral lung ventilation. Anesth Analg 103(6): 1489–1493.
- 29. Kompis M, Pasterkamp H, Oh Y, Wodicka GR (1997) Distribution of inspiratory and expiratory respiratory sound intensity on the surface of the human thorax. Conf Proc IEEE Eng Med Biol Soc 5: 2047–2050.
- 30. Torres-Jiménez A, Charleston-Villalobos S, González-Camarena R, Chi-Lem G, Aljama-Corrales T (2008) Asymmetry in lung sound intensities detected by respiratory acoustic thoracic imaging (RATHI) and clinical pulmonary auscultation. Conf Proc IEEE Eng Med Biol Soc 2008: 4797–4800.
- 31. Dellinger RP, Parrillo JE, Kushnir A, Rossi M, Kushnir I (2008) Dynamic visualization of lung sounds with a vibration response device: a case series. Respiration 75(1): 60–72.
- 32. Pasterkamp H, Kraman SS, Wodicka GR (1997) Respiratory sounds. Advances beyond the stethoscope. Am J Respir Crit Care Med 156(3 Pt 1): 974–987.
- 33. Gavriely N, Nissan M, Cugell DW, Rubin AH (1994) Respiratory health screening using pulmonary function tests and lung sounds analysis. Eur Respir J 7(1): 35–42.
- 34. Gharagozloo F, McReynolds SD, Snyder L (1995) Thoracoscopic plication of the diaphragm. Surg Endosc 9(11): 1204–1206.
- 35. Roussos CS, Fixley M, Genest J, Cosio M, Kelly S, et al. (1977) Voluntary factors influencing the distribution of inspired gas. Am Rev Respir Dis 116(3): 457–467.
- 36. Amis TC, Ciofetta G, Hughes JM, Loh L (1980) Regional lung function in bilateral diaphragmatic paralysis. Clin Sci (Lond) 59(6): 485–492.
- 37. DiMarco AF, Connors AF Jr, Kowalski KE (2004) Gas exchange during separate diaphragm and intercostal muscle breathing. J Appl Physiol 96(6): 2120–2124.
- 38. Xu WD, Gu YD, Lu JB, Yu C, Zhang CG, et al. (2005) Pulmonary function after complete unilateral phrenic nerve transection. J Neurosurg 103(3): 464–467.
- 39. McCool FD, Tzelepis GE (2012) Dysfunction of the diaphragm. N Engl J Med 366(10): 932–42.
- 40. Balaji S, Kunovsky P, Sullivan I (1990) Ultrasound in the diagnosis of diaphragmatic paralysis after operation for congenital heart disease. Br Heart J 64(1): 20–2.
- 41. Boudarham J, Pradon D, Prigent H, Falaize L, Durand MC, et al. (2013) Optoelectronic plethysmography as an alternative method for the diagnosis of unilateral diaphragmatic weakness. Chest 144(3): 887–95.
- 42. Gross V, Dittmar A, Penzel T, Schüttler F, von Wichert P (2000) The relationship between normal lung sounds, age, and gender. Am J Respir Crit Care Med 162(3): 905–909.
- 43. Gayan-Ramirez G, Gosselin N, Troosters T, Bruyninckx F, Gosselink R, et al. (2008) Functional recovery of diaphragm paralysis: a long-term follow-up study. Respir Med 102(5): 690–698.