Failure of intestinal anastomosis is a major complication following abdominal surgery. Biological materials have been introduced as reinforcement of abdominal wall hernia in contaminated setting. An innovative application of biological patch is its use as reinforcement of gastrointestinal anastomosis. The aim of study was to verify whether the bovine pericardium patch improves the healing of anastomosis, when in vivo wrapping the suture line of pig intestinal anastomosis, avoiding leakage in the event of deliberately incomplete suture. Forty-three pigs were randomly divided: Group 1 (control, n = 14): hand-sewn ileo-ileal and colo-colic anastomosis; Group 2 (n = 14): standard anastomosis wrapped by pericardium bovine patch; Group 3 (n = 1) and 4 (n = 14): one suture was deliberately incomplete and also wrapped by patch in the last one. Intraoperative evaluation, histological, biochemical, tensiometric and electrophysiological studies of intestinal specimens were performed at 48 h, 7 and 90 days after. In groups 2 and 4, no leak, stenosis, abscess, peritonitis, mesh displacement or shrinkage were found and adhesion rate decreased compared to control. Biochemical studies showed mitochondrial function improvement in colic wrapped anastomosis. Tensiometric evaluations suggested that the patch preserves the colic contractility similar to the controls. Electrophysiological results demonstrated that the patch also improves the mucosal function restoring almost normal transport properties. Use of pericardium bovine patch as reinforcement of intestinal anastomosis is safe and effective, significantly improving the healing process. Data of prevention of acute peritonitis and leakage in cases of iatrogenic perforation of anastomoses, covered with patch, is unpublished.
Citation: Testini M, Gurrado A, Portincasa P, Scacco S, Marzullo A, Piccinni G, et al. (2014) Bovine Pericardium Patch Wrapping Intestinal Anastomosis Improves Healing Process and Prevents Leakage in a Pig Model. PLoS ONE 9(1): e86627. https://doi.org/10.1371/journal.pone.0086627
Editor: Giovanni Li Volti, University of Catania, Italy
Received: November 4, 2013; Accepted: December 16, 2013; Published: January 29, 2014
Copyright: © 2014 Testini et al. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Funding: The authors have no support or funding to report.
Competing interests: The authors have declared that no competing interests exist.
Failure of gastrointestinal anastomosis results in dehiscence, leaks and fistulas, and is considered a major complication following abdominal surgery. Despite improved surgical technique, the reported incidence of gastrointestinal anastomosis leakage ranges from 2% to 10% –, and is associated with both increased morbidity (20–30%) and mortality (7–12%) –.
Hypoalbuminemia, chronic obstructive pulmonary disease, colon cancer and IBD have been identified as significant risk factors for anastomotic leakage , , , . However, wide resection margins, absence of tension at the level of the suture site and resection along anatomic blood supply may decrease the risk , . By contrast, hypovolemia, blood transfusions, besides surgical skill, prolonged operative time and difficult operative procedures, have been associated with anastomotic leakage development , .
Biological materials have been introduced in general surgery as reinforcement of abdominal wall hernia in contaminated setting, when the use of alloplastic meshes is contraindicated . Hand to hand with the success of bovine pericardium for valves and patch grafts in cardiac surgery , , this biomaterial has been considered suitable in place of dura mater in anterior abdominal wall defects in pediatric surgery ,  and then for the treatment of the patients affected by incisional hernia particularly in the contaminated or urgent context. An innovative application in this respect is the use of the biomaterials as reinforcement of the gastrointestinal anastomotic suture line –. In particularly, the available experimental data showed that the mechanical anastomoses buttressed with bovine pericardium ,  or small intestinal submucosa had greater bursting strength as compared to non-buttressed anastomoses ,  and a wound healing improvement ,  has been demonstrated. Moreover, a colic perforation treated by positioning a resorbable bilayer collagen band of bovine origin in a pig model showed results very encouraging .
The aim of the study was to verify whether the bovine pericardium patch improves the healing of anastomosis, when affixed in vivo on the hand-sewn suture line of ileo-ileal and colo-colic anastomosis of the pigs. A further end-point was to see whether the patch is able to avoid the anastomotic leakage in the case of deliberately incomplete suture. For this, we used a pig model undergoing intraoperative and histological evaluation, and biochemical, tensiometric and electrophysiological measurements of intestinal specimens because these animals recapitulate several key features of human anatomy and physiology of the grastrointestinal tract.
Materials and Methods
Animals and Ethics Statement
After approval by the Italian Ministry of Health (protocol number: 02/2010) and in strict accordance with the recommendations in the Guide for the Care and Use of Laboratory Animals of the National Institutes of Health, between September 2010 and April 2012, forty-three domestic pigs (Landrace; female; mean age 5.3±2.2 months; weight 38.7±9.2 Kg) were sourced from commercial piggery affiliated with the Division of Veterinary Clinics and Animal Productions of University Medical School “A. Moro” of Bari (Italy), and were included in this study and underwent two surgical procedures. Vendor health reports indicated that the animals were free of known viral, bacterial and parasitic pathogens. The pigs were weighed before surgery and at regular interval during the experimental period. After a preoperative fasting time of 24 hours, all surgery was performed under an aseptic setting and general anesthesia, and all efforts were made to minimize suffering. At the end of the second surgical procedure, pigs were euthanized with a bolus of thiopental followed by a bolus of KCl. All pigs received ampicilline (25 mg/kg every 12 hours) and tramadol (5 mg/kg daily) for seven days after surgery. Pigs were fed, leaving water ad libitum, and were monitored daily in order to detect any alteration of the clinical conditions (food intake and weight loss; urine and feces production; rectal temperature and behavior changes).
Study groups and Surgical Procedure I
The peritoneal cavity was entered via a midline incision and a segment of colon 30 cm from the anal verge and of ileum 30 cm from the cecum verge were resected. The animals were randomly assigned to four groups: 1) Group 1 (control, n = 14): the hand-sewn ileo-ileal and colo-colic anastomosis were performed using single layer of interrupted suture (PDS 3/0, Ethicon, Germany) according to Gambee; the distance between the single sutures were 3–4 mm; 2) Group 2 (n = 14): the standard anastomosis were wrapped by a 50×20 mm single layer pericardium bovine patch (Tutomesh®, Tutogen Medical GmbH, Germany); in order to allow some swelling during healing, 360° anastomotic sealing was achieved by a 5 mm over-lapping of the two ends of the patch at the mesenterial side of the anastomosis without fixing the ends to each other (Fig. 1A); 3) Group 3 (n = 1): one suture of the anastomosis was deliberately performed incomplete; 4) Group 4 (n = 14): one suture of the anastomosis was deliberately performed incomplete and wrapped by the pericardium bovine patch. The abdominal cavity was closed in layers with absorbable sutures in all pigs. Procedures were carried out by the same surgeon (M.T.) and the randomization was performed by using numbered and sealed envelopes that were opened at the beginning of the operation.
Ileo-ileal anastomosis wrapped by a 50×20 mm one layer pericardium bovine patch (A); at day 7 (B) the patch is enveloped by an heavy lympho-isthyocitic infiltrate without essudation on the serosal surface (Group 2) and after 48 h (C) the necrotic tissue is well contained into the intestinal wall by the patch (Group 4), with minimal leukocyte infiltration of the serosal surface (Haematoxylin-Eosin 200X original magnification).
Surgical Procedure II and Macroscopic examination
In Group 1, 2 and 4, a relaparotomy was performed 48 h (4 pigs for each group), 7 and 90 days after (5 pigs for each group); the animals of the Group 3 were reoperated 48 h thereafter only. The abdominal cavity was inspected for any kind of free fluid collection and anastomotic leakage, stenosis, abscess, peritonitis or intra-abdominal adhesions. Quantity and quality of adhesions were examined in a scoring system (0–3 points). In order to perform the microscopic, biochemical, tensiometric and electrophysiological studies, an intestinal tract was dissected four centimeters upstream and downstream the surgical site.
Preparation of histological specimens and Microscopic examination
All the samples were fixed with neutral buffered formalin for 24–48 hours and paraffin- embedded; consecutive sections of 3 µm thickness were longitudinally cut to include the anastomosis and the adjacent intestinal tracts, in order to observe the whole section from the mucosa to the serosa, and were stained with haematoxylin-eosin and periodic acid-Schiff stains.
Evaluation was carried out by the same pathologist blinded for the experimental protocol. For each section the following histological parameters were evaluated: grade of inflammation, parietal fibrosis, integrity of the mucosal layer, serosal status, and the grade of granulocyte infiltration of the membrane in cases with the biological mesh. Phlogosis, fibrosis and mesh infiltration were employed for scoring system (0–4 points).
Immediately after surgery, the tissue of representative segments of ileo-ileal and colo-colic anastomosis (Group 1, 2) was disrupted by a glass-Teflon homogenizer and mitochondria were isolated according to Bookelman et al. . The mitochondrial respiration was measured polarographically with a Clark-type oxygen electrode in a water-jacketed chamber, magnetically stirred at 37°C. Respiration rates were expressed as nanomoles of molecular oxygen consumption per minute per milligram of mitochondrial protein , after addition of substrates and inhibitors of oxidative phosphorylation. To measure the rotenone sensitive reduced nicotinamide adenine dinucleotide (NADH)-ubiquinone oxidoreductase activity, mitochondria were exposed to ultrasound energy for 15 s at 0°C . Cytochrome c oxidase activity was determined on mitochondria , following the oxidation of ferrocytochrome c 10 μM. Citrate synthase activity  was used as mitochondrial matrix enzymatic marker. For detection of reactive oxygen species (ROS) in ileo-ileal and colo-colic anastomosis whole tissue homogenate, fluorimetric analysis was used  with a Jasco FP6200 spectrofluorimeter. All activities were expressed as nanomoles of substrates per minute per milligram of mitochondrial protein.
Tensiometric studies of freshly excised colic and ileal smooth muscle specimens (Group 1, 2) were conducted at 48 hours, 7 and 90 days after surgery. Specimens were mounted in an organ bath filled with modified Krebs' solution (20 ml at 37°C gassed with O2/CO2 (95%/5%), pH 7.4). Isometric tension was measured with a strain gauge transducer (cat. 7003 Basile, Milan, Italy) connected to data acquisition system (PowerLab and Chart 4.1.2 ADInstruments, Castle Hill, Australia). After 45 minutes equilibration, an initial load of 1.0 g tension was applied to the tissue. Contractile responses were measured to acetylcholine (ACh, 10−7–10−4 M) and KCl (80 mM).
Transepithelial potentials (VT), resistance (RT) and short circuit current (ISC) were measured as markers of colic and ileal mucosa transport efficiency at baseline, early stage (days 2 and 7, tissues pool), and late stage (day 90) after surgery in Groups 1, 2. Freshly excised mucosa was placed in a cooled modified Krebs' bicarbonate/phosphate buffer solution (at 37°C gassed with O2/CO2 (95%/5%), pH 7.4) and mounted in a Ussing chambers (Mussler Scientific Inst., Aachen, Germany) with an exposed area of 1 cm2. Two pairs of Ag/AgCl electrodes were used to monitor VT (mV) and RT (Ω.cm2). ISC (μA/cm2) was measured with the VT clamped to 0 at 5 min intervals. At fixed intervals of 1 min a transepithelial bipolar current pulse (I) of 1 µA amplitude and 200 msec duration was applied to tissue and RT calculated from the change in open-circuit voltage (ΔVT) according to Ohm's law (RT = ΔVT/I). Experiments were conducted simultaneously on multiple specimens, each from an individual pig. Electrical parameters were measured by the software Clamp (v. 2.14, Aachen, Germany) and recorded for 60 min after an initial equilibration time of about 30 min.
Data are given as means ± SEMs. Infra-group comparisons were made using median test with the two sided p-value computed using Fisher's exact test. For biochemical studies, one-way or two-way analysis of variance, as appropriate, followed by paired Student t-test or Newman-Keuls multiple comparison tests were used. For electrophysiological studies, data of the measurements during 60 min were compared between control and treated tissues were assessed using the Student t-test for unpaired data between the data average of early and late follow-up days vs day 0 and for anastomosis with patch vs anastomosis without patch. Data were analyzed by Stata 12 software (StataCorp LP, College Station, Tex, USA). P values of <0.05 were considered statistically significant.
Tables 1 and 2 summarize the intraoperative findings at relaparotomy. In the control group, one leakage was revealed in a colo-colic anastomosis 48 h after surgery. No stenosis, abscess or peritonitis was found in this group and adhesions were progressively observed in the subgroups. In Group 2, the intraoperative evaluation of the abdominal cavity lacked anastomotic leaks, intraabdominal abscess, stenosis or peritonitis. Moreover, adhesion rates (quantity and quality; Table 2) were significantly less than the control group (p = 0.006 and p = 0.018, respectively). As expected, in Group 3, peritonitis was evident and adhesions were extremely solid. By contrast, in Group 4, no leaks, stenosis, abscess and peritonitis were seen in any of the pigs and adhesion rates progressively increased, but less than Group 1 again (p = 0.016 and = 0.006, respectively). No shrinkage and displacement of prosthesis was seen either in Group 2 or 4.
Table 3 shows the microscopic findings of the anastomosis at the relaparotomy. In control group, a heavy inflammatory infiltrate was observed at early stage, decreasing progressively and being virtually absent on day 90 postoperatively. Conversely, the fibrogenic process was initially evident from the 7th postoperative day and completed on the 90th. The mucosal surface and the serosal layer rapidly recovered. In Groups 2 (Fig. 1B) and 4 (Fig. 1C), there was no statistical difference in comparison with the control group in terms of the phlogosis. Moreover, the fibrotic reaction was statistically less evident in Groups 2 and 4 (both p = 0.033) compared to the control. Indeed, a significant granulation tissue was initially present at the border of the bovine pericardium patch and subsequently in the deeper areas. On the 90th day there were only small, hardly recognizable fragments of the patch surrounded by macrophages and lymphocytes. While the mucosal surface rapidly recovered, the serosa became smooth only 90 days after the operation. In the Group 3, the inflammatory cell infiltrate and the fibrin deposition were massive and diffuse.
We measured the mitochondrial respiratory activity by endogenous substrates (NADH-dependent respiration through complex I, III and IV; Fig. 2A) and from added substrates, like succinate (respiration through complex II, III and IV; Fig. 2B) and ascorbate (final respiratory step through complex IV; Fig. 2C), in ileo-ileal and colo-colic anastomosis samples (Group 1, 2). In Fig. 2D, respiratory control ratios indicates the coupling of mitochondria, oxidative phosphorylation efficiency and adenosine triphosphate (ATP) production. In the same samples, at each time point, the enzyme marker for mitochondrial matrix (citrate synthase; Fig. 3A), and the enzymatic activities of complex I (NADH-ubiquinone oxidoreductase; Fig. 3B) and IV (cytochrome c oxidase; Fig. 3C) were measured. Within 48 hours after anastomosis, mitochondrial respiration (Fig. 2A–C) and enzymatic activities (Fig. 3A–C) showed a marked decline of all functional parameters in ileo-ileal and colo-colic anastomosis, which lasted for one week. After this phase, the recovery showed a difference between the two intestinal parts. Specifically, the biochemical parameters of ileo-ileal anastomosis showed a gradual recovery and in three months were almost fully restored to the initial values, without a significant influence by the patch (Group 2). On the contrary, the parameters of colo-colic anastomosis showed no recovery even after three months, except when the patch was applied, with a partial early recovery after two days, which was full after three months. Uncoupling of mitochondria from two days to one week after anastomosis was observed in both colo-colic and ileo-ileal anastomosis, with the latter being more affected (Fig. 2D). The Tutomesh®, once again, was shown to improve the recovery of mitochondrial coupling in colo-colic anastomosis from one week to three months. Measurement of hydrogen peroxide generation in whole tissue homogenates after anastomosis showed significant differences between ileo-ileal and colo-colic anastomosis (Fig 3D). In particular, in ileo-ileal anastomosis a significant increase of H2O2 production was detected from two days to one week after anastomosis, before returning to basal levels after three months and this did not seem to be significantly influenced by the patch. In classic colo-colic anastomosis, however, the increase of H2O2 generation was significantly higher two days after anastomosis and decreased progressively in the subsequent days reaching normal basal values after three months. Unlike this application of Tutomesh®, in colo-colic anastomosis it appeared to limit the early increase of H2O2 production, two days after anastomosis, which was delayed to a threefold increase at one week and gradually decreased to normal basal values in three months.
Mitochondrial respiratory activities, from endogenous substrates (A), succinate (B) and ascorbate (C), and mitochondrial respiratory control ratios from endogenous respiration (D) in ileo-ileal (gray bars) and colo-colic (white bars) anastomosis. In each coupled bars values from Group 1 (left) and Group 2 (right) are compared.
Mitochondrial activities of citrate synthase (A), NADH-ubiquinone oxidoreductase (B), cytochrome c oxidase (C) and hydrogen peroxide production (D) in ileo-ileal (gray bars) and colo-colic (white bars) anastomosis. In each coupled bars values from Group 1 (left) and Group 2 (right) are compared.
At baseline, ACh induced dose-response contractions in ileum and colon specimens (controls), which were slightly more pronounced in ileum. The response to KCl was comparable in both bowel tracts. Forty-eight hours after surgery (Fig. 4A), the contractile response of the colon to ACh was lower (p<0.05) in Group 1 vs control, while Tutomesh® prevented this effect. No difference was found in ileal specimens (Fig. 4B). Seven days after surgery the contractile response of the colon to ACh increased in Group 1 (p<0.05) vs control, but was unchanged in Group 2 (Fig. 4C). In the ileum, Tutomesh® determined a reduced contractile response to ACh (p<0.05), while Group 1 response was similar to control (Fig. 4D). Ninety days after surgery, the response to ACh or KCl was comparable in different surgical treatments both in the colon and ileum (Fig. 4E,F). The response to KCl was similar in colic and ileal specimens (Fig. 4A–F).
At baseline, the ileum showed ISC −12.1±1.3 μA/cm2, RT 131.2±8.1 Ω. cm2, and VT 1.1±0.2 mV, lumen negative (N = 15). The colon showed ISC −14.7±1.2 μA/cm2, RT 126.6±6.9 Ω. cm2, and VT 0.59±0.1 mV, lumen negative (N = 15). In the ileo-ileal anastomosis the early stage, was associated with significantly decreased ISC (−74.1%, p<0.0001 vs control) and increased RT (+25.7%, p<0.05 vs control) (Fig. 5A,B). A similar trend (i.e. −35%) existed for VT (data not shown). In the late stage, ISC remained below the control value (−40.9%, p<0.05 vs control; Fig. 5A), RT increased significantly up to 53% (p<0.02 vs control; Fig. 5B), and VT decreased by 57.5% (p<0.02 vs control). The presence of colo-colic anastomosis in the early stage was associated with significantly decreased ISC (−36,4%, p<0.05 vs control; Fig. 5C), a tendency of increased RT (Fig. 5D) and reduced VT. In the late stage, ISC remained stable, while RT and VT tended to increase compared to the control. Affixing a pericardium bovine patch on ileal anastomosis prevented the electrophysiological changes seen with anastomosis alone (ISC virtually unchanged in the early and late stages, Fig. 5A) while RT (Fig. 5B) and VT did not increase. In the colon, likewise, the prosthesis prevented the changes of ISC, RT and VT (Fig. 5C–D).
Short circuit current (ISC) and tissue resistance (RT) in ileal and colic specimens. Bars represent mean ± SEM in C (before surgery, N = 15), A-ES (after surgery without patch, early stage, N = 15), P-ES (after surgery with patch, early stage, N = 15), A-LS (after surgery without patch, late stage, N = 10), P-LS (after surgery with patch, late stage, N = 10) in ileal and colic specimens, respectively. Significance was determined by Student's t test for unpaired data of follow-up stage vs control (*p≤0.05; **p≤ 0.002; ***p≤0.0001) and for anastomoses with patch vs anastomoses without patch control (#: p≤0.05).
Our experimental study is the first in literature to investigate whether a pericardium bovine patch, wrapping ileo-ileal and colo-colic hand-sewn anastomosis in pigs, seals the suture line and promotes processes of anastomotic healing. Dehiscence of anastomosis is a disturbing, common and severe complication after bowel resection, with an incidence ranging from 0.5 to 30% after large bowel resection and to about 6% after small bowel resection –.
Several studies have demonstrated that the early integrity of the anastomosis depends not only on the correct surgical performance, but also on the suture-holding property of the sub-mucosal layer and on formation of a fibrin seal on the serosa . Furthermore, the process of anastomotic healing is related to the structure and arrangement of the collagen matrix .
Despite the fact that many risk factors for postoperative leakage have been analyzed in literature , , , , few clinical and experimental studies have focused on the best prevention techniques of anastomotic dehiscence , . However, operative and post-operative treatments, allowing suitable oxygenation and blood perfusion of the anastomotic side and stimulating the angiogenesis, with infusion of grow factors or inhibitors of metalloproteinase, have been investigated , . Some studies investigated whether the reinforcement of anastomosis with biological or synthetic materials was able to prevent the anastomotic leakage. Reports regarding the use of synthetic materials are, however, uncommon and with discordant outcomes , . Hand in hand with the use of the biologic mesh in abdominal wall repair, many experimental studies on their application as reinforcement of anastomosis have spread, and the bovine pericardium –, small intestinal submucosa – and porcine dermis  are the biomaterials used. The biological materials are all basically composed of an extracellular matrix deprived of its cellular components. The extracellular matrix serves as scaffold for the remodeling process by host through the connective tissue ingrowth and cellular colonization and proliferation. Reports regarding the use of the bovine pericardium are experimental – or clinical studies  and the anastomoses have been performed using a circular stapler with the introduction of bovine pericardium as a buttressing material to reinforce staple lines.
In this study, the anastomosis was performed using a single layer of interrupted suture according to Gambee. In accordance with knowledge of the healing phases of anastomosis, the surgical procedure II was carried out in the early and late phase (at 48 h, 7, and 90 days, respectively after the first operation). However, the previous experimental studies analyzed the early effect (after a few hours) of the pericardium bovine buttress on the anastomosis ,  and only Hagerman et al  performed the evaluation in late phase of suture healing. In this study, we provide a complete overview of key events involved in anastomosis healing with and without bovine patch. By using highly integrated and translational methodologies, we describe a detailed intraoperative evaluation, histological and biochemical analyses, and tensiometric and electrophysiological studies on intestinal specimens after the surgical procedure II for each group of pigs. Overall, the whole intestinal wall from mucosa to muscle and serosa was fully assessed by morphological and functional methods. Macroscopic side effects of reinforcement with pericardium bovine patch were investigated in intraoperative setting II. In accordance with the uneventful convalescence of the pigs in the groups with the wrap of pericardium bovine on anastomosis, no leak or stenosis or abscess and peritonitis were found in early and late relaparotomy, nor were any cases of displacement and shrinkage of the mesh. Some Authors described that the reinforcement with collagen fleeces could cause intestinal obstruction, resulting from the phenomena of shrinkage and displacement of the prosthesis , . Moreover, we found decreased rate of intra-abdominal adhesions compared to standard anastomoses, while other studies showed a similar entity of adhesions , . As expected, when the iatrogenic perforation had been performed without any reinforcement, a diffuse and severe peritonitis was the intraoperative finding.
On the other hand, an original finding is that the absence of leak, stenosis, abscess and peritonitis was also found when one suture of the anastomosis was performed deliberately incomplete and wrapped by the pericardium bovine patch. Moreover, intra-abdominal adhesions in this group were less significant in comparison with the control again. The same biological features, for which the bovine pericardium has been employed widely for the abdominal wall hernia repair, could explain this important data. The bovine pericardium is an acellular membrane of pure collagen with native structure, not cross-linked, made up of multidirectional fibers with adequate tensile strength and representing the scaffold for replacement by new endogenous tissue. The remodeling process of the mesh by host determines complete degradation by fibroblasts with deposition of autologous tissue, without foreign body reaction, or fibrous capsule formation, or shrinking, or adhesion formation and fast peritoneal regeneration. We hypothesized that the biological wrap has obstructed the leak through the small iatrogenic perforation, firstly with mechanical action and definitively with the improvement of healing of suture. The mechanism of healing could be the result between the combination of immediate action as cap on the anastomotic dehiscence, and the known remodeling process of the mesh, through the gradual degradation of bovine pericardium via collagenase, and the rapid cellular ingrowth. Obviously, this effect is possible when a small leakage and incomplete anastomotic dehiscence occurs. Confirming the buffering effect of the pericardium bovine patch in order to avoid the anastomotic leakage in the cases of iatrogenic perforation and the improvement on the healing of the tissue, the microscopic findings included only a moderate granulocyte infiltration limited to the patch, but no signs of peritonitis. Statistically, the fibrosis was less induced than the control and on 90th day, the remodeling process of the mesh was complete. As reported previously , our histological examination showed a migration of fibroblasts into the pericardium bovine patch, a regeneration of the bowel layer at the anastomotic site, with the improvement of healing of the host tissue. This improvement of healing was shown first in the site of iatrogenic perforation.
There is currently a lack of biochemical, tensiometric and electrophysiological studies regarding the effect of the pericardium bovine patch on the intestinal anastomosis in literature. The biochemical results indicated that the colon relies differently on mitochondrial aerobic metabolism than the ileum, according to respiratory and enzymatic activities of the basal samples, which are about twofold higher in colon, as found from previous observations by our group , . Confirming these previous experimental results , , after surgical treatment, a temporarily ischemic condition, associated to some necrosis and release of inflammatory mediators, may generate a local anaerobic environment and consequently a switch to glycolytic metabolism for cellular ATP supply. Tissues depending on oxidative aerobic metabolism, like the colon, may suffer under these conditions with a lack of adaptation to glycolytic metabolism, whilst anaerobic metabolism can be better tolerated by the small bowel , . This may explain the marked depression of mitochondrial functions in the colo-colic anastomosis, whilst the ileum can tolerate the post-anastomosis stress. Consequently, repair processes after anastomosis may be more efficient in the ileum that is easily able to proceed from the initial critical phase to the healing process, as shown by the recovery of mitochondrial parameters in the subsequent days, which are fully restored in three months. It appears that the colon does not comply well with the post-anastomosis stress, due to the cellular component of the tissue exhibiting high metabolic demand from oxidative metabolism, causing an irreversible loss of noble elements, which will not be replaced in the subsequent phases of healing. The loss of mitochondrial functions, indeed, does not seem to be restored to the initial values, even after three months.
The capacity of the ileum to tolerate the post-anastomosis stress may explain why the introduction of a patch support in the healing does not improve the whole process, as shown by the compared values of mitochondrial respiration and enzymatic activities. On the contrary, the affixing of the patch on the colic anastomosis produces a positive effect in the healing, as indicated by the data of mitochondrial functions, which perform better when compared to untreated samples and show an almost complete recovery in 3 months.
We have experimental evidence that the patch may play a role on the oxidative stress generated during the healing process of anastomosis, as shown by measurements of H2O2 production. In the ileo-ileal anastomosis, independently from the patch, a significant increase of ROS levels is observed from two to seven days after anastomosis, indicating that an inflammatory response and tissue regeneration take place contemporarily and obviously both are needed for the healing process (as also indicated by the presence of a heavy inflammatory infiltrate in the histological samples). In colo-colic anastomosis, without patch, the ROS levels have an early twofold increase at 48 hours, which declines in one week. This precocious oxidative stress could be an additional factor involved in the tissue damage that leads to mitochondrial dysfunction and lack of restoration of tissue function. On the contrary, when the patch is applied to the colo-colic anastomosis, the maximum increase of ROS is delayed at one week where it reaches a threefold level compared to basal production. After this peak, the ROS generation decreases and after one month their levels are normalized. The delay of oxidative stress in patch anastomosis could prevent damage to noble cells in the large bowel, like tissue stem cells, in the early stage of the repair processes, allowing a complete restoration of tissue functions and a decrease of fibrotic reaction in the subsequent stages. The protective effect of a patch is compatible with the histological observation of a moderate inflammatory infiltrate and the late increase of ROS can correlate with the appearance of a significant granulation tissue, which, at this final stage, is no more harmful for the repair process.
Regarding tensiometric evaluations, our results suggest that the use of the patch can preserve smooth muscle response to acetylcholine similar to the response of controls (samples without anastomosis) in colic specimens in the early postoperative time (48 h-7 days), while the colic preparations with traditional anastomosis showed contractility alterations compared to control. However, the use of pericardium bovine patch seems to impair the ileal contractile response at seven days after surgery. In late stages, the responses to acetylcholine were comparable to the specimens without anastomosis for both colon and ileum in which it is affixed to the mesh. It is conceivable that the decreased fibrotic process seen with the affixing of Tutomesh® might have preserved the smooth muscle function.
The results of electrophysiological parameters are able to describe fine changes occurring at the intestinal mucosal side, a very delicate monolayer structure that is highly sensitive to local and surrounding changes. Results are in good agreement with those reported for the pig by others –, when also including tissue-specific differences between ileum and colon. Overall, our results suggest that subtle but important changes are observed in both ileum and colon in the cases of unprotected anastomosis. A number of factors, including early inflammatory changes, oxidative stress and late cicatricial tissue formation around the anastomosis, might impair both permeability and transport properties in the mucosa. In the ileum, the presence of the pericardium bovine patch clearly prevents the alterations following the traumatic effect of surgery. The colon appears to behave slightly differently, in that the transport properties are significantly reduced only at an early stage, while the permeability is less affected. Tutomesh®, however, appears to modulate and counteract the traumatic effect of surgery. Overall, our results suggest that application of the patch also improves the intestinal mucosal function, restoring almost normal transport properties. This is a process in line with the histological finding of less fibrotic reaction within the intestinal wall upon affixing of Tutomesh®.
Our study demonstrated that the use of the pericardium bovine patch as reinforcement of the intestinal anastomosis is safe and effective. All performed analyses showed a significant improvement of the healing process of the anastomosis. Moreover, the histopathological data of the prevention of leakage into the abdominal cavity in cases of iatrogenic perforation of the ileal and colic anastomosis, covered with the wrap of bovine pericardium, resulting in prevention of acute peritonitis, is unpublished and surprising. The biochemical analysis confirmed that the post-anastomosis cellular stress, which probably involves inflammatory events, cytokines release, cell growth induction and anaerobic conditions, is better tolerated by the ileum, whilst the aerobic colon may suffer to a greater extent undergoing oxidative stress and mitochondrial failure, which are prevented by the use of a patch. The use of patch is advisable also for improving the healing process inasmuch as it is able to guarantee the efficient recovery of the functional transport properties of the intestinal epithelia and of the contractility. We suggest that our experimental results could be the basis for the multicenter controlled clinical trials in humans, comparing the outcomes of intestinal anastomosis performed with and without the bovine pericardium patch as reinforcement. The aim of these studies could be also to analyze the impact in terms of cost-benefit health, verifying a possible decrease of morbidity and mortality related to the healing of the anastomosis.
We are indebted to P. De Benedictis, R. De Venuto, M. Persichella for their skillful technical support. The authors thank Prof. Malcolm Clark for assistance in the preparation of the English manuscript.
Conceived and designed the experiments: MT AG PP SS AM GP AC. Performed the experiments: MT AG PP SS AM GP GL LG MADS LB LD NS FS MRC AC. Analyzed the data: MT AG PP SS AM MADS LB LD NS MRC. Contributed reagents/materials/analysis tools: MT AG PP SS AM GP GL LG MADS LB LD NS FS MRC AC. Wrote the paper: AG PP SS AM FS.
- 1. Telem DA, Chin EH, Nguyen SQ, Divino CM (2010) Risk factors for anastomotic leak following colorectal surgery: a case-control study. Arch Surg 145: 371–375.
- 2. Hyman N, Manchester TL, Osler T, Burns B, Cataldo PA (2007) Anastomotic leaks after intestinal anastomosis: it's later than you think. Ann Surg 245: 254–258.
- 3. McArdle CS, McMillan DC, Hole DJ (2005) Impact of anastomotic leakage on long-tem survival of patients undergoing curative resection for colorectal cancer. Br J Surg 92: 1150–1154.
- 4. Testini M, Miniello S, Piccinni G, Di Venere B, Lissidini G, et al. (2003) Correlation between chronic obstructive bronchial disease and colic anastomosis dehiscence in the elderly. Ann Ital Chir 74: 247–250.
- 5. Docherty JG, McGregor JR, Akyol AM, Murray GD, Galloway DJ (1995) Comparison of manually constructed and stapled anastomoses in colorectal surgery. West of Scotland and Highland Anastomosis Study Group. Ann Surg 221: 176–184.
- 6. Kimberger O, Fleischmann E, Brandt S, Kugener A, Kabon B, et al. (2007) Supplemental oxygen, but not supplemental crystalloid fluid increases tissue oxygen tension in healthy and anastomotic colon in pigs. Anesth Analg 105: 773–779.
- 7. Zhou HY, Zhang J, Yan RL, Wang Q, Fan LY, et al. (2011) Improving the antibacterial property of porcine small intestinal submucosa by nano-silver supplementation: a promising biological material to address the need for contaminated defect repair. Ann Surg 253: 1033–1041.
- 8. Crawford FA Jr, Sade RM, Spinale F (1986) Bovine pericardium for correction of congenital heart defects. Ann Thorac Surg 41: 602–605.
- 9. Morell VO, Wearden PA (2007) Experience with bovine pericardium for the reconstruction of the aortic arch in patients undergoing a Norwood procedure. Ann Thorac Surg 84: 1312–1315.
- 10. Hutson JM, Azmy AF (1985) Preserved dura and pericardium for closur of large abdominal wall and diaphragmatic defects in children. Ann Roy Col Surg Engl 67: 107–108.
- 11. Van Tuil C, Saxena AK, Willital GH (2006) Experience with management of anterior abdominal wall defects using bovine pericard. Hernia 10: 41–47.
- 12. Schreinemacher MH, Bloeman JG, van der Heijden SJ, Gijbels MJ, Dejong CH, et al. (2011) Collagen fleeces do not improve colic anastomotic strength but increase bowel obstructions in an experimental rat model. Int J Colorectal Dis 26: 729–735.
- 13. Gaertner WB, Hagerman GF, Potter MJ, Karulf RE (2010) Experimental evaluation of a bovine pericardium-derived collagen matrix buttress in ileocolic and colon anastomoses. J Biomed Mater Res B Appl Biomater 92: 48–54.
- 14. Hagerman GF, Gaertner WB, Ruth GR, Potter ML, Karulf RE (2007) Bovine pericardium butress reinforces colorectal anastomoses in a canine model. Dis Colon Rectum 50: 1053–60.
- 15. Arnold W, Shikora SA (2005) A comparison of burst pressure between buttressed versus non-buttressed staple-lines in an animal model. Obes Surg 15: 164–171.
- 16. Shikora Sa, Kim JJ, Tarnoff ME (2003) Reinforcing gastric staple-lines with bovine pericardial strips may decrease the likelihood of gastric leak after laparoscopic Roux-en-Y gastric bypass. Obes Surg 13: 37–44.
- 17. Hoeppner J, Crnogorac V, Marjanovic G, Jüttner E, Keck T, et al. (2009) Small intestinal submucosa for reinforcement of colic anastomosis. Int J Colorectal Dis 24: 543–550.
- 18. Hoeppner J, Wassmuth B, Marjanovic G, Timme S, Hopt UT, et al. (2010) Anastomotic sealing by extracellular matrices (ECM) improbe healing of colic anastomoses in critical early phase. J Gastrointest Surg 14: 977–986.
- 19. Downey DM, Harre JG, Dolan JP (2005) Increased burst pressure in gastrointestinal staple-lines using reinforcement with a bioprosthetic material. Obes Surg 15: 1379–1383.
- 20. Hoeppner J, Willa K, Timme S, Tittelbach-Helmrich D, Hopt UT, et al. (2010) Reinforcement of colic anastomoses with a collagenous double-layer matrix extracted from porcine dermis. Eur Surg Res 45: 68–76.
- 21. Nocca D, Aggarwal R, Deneve E, Picot MC, Sanders G, et al. (2009) Use of collagen wrap from bovine origin for the management of colic perforation. Preliminary study in a pig model. J Laparoendosc Adv Surg tech A 19: 79–83.
- 22. Bookelman H, Trijbels JM, Sengers RC, Janssen AJ, Veerkamp JH, et al. (1978) Pyruvate oxidation in rat and human skeletal muscle mitochondria. Biochem Med 20: 395–403.
- 23. Iuso A, Scacco S, Piccoli C, Bellomo F, Petruzzella V, et al. (2006) Dysfunctions of cellular oxidative metabolism in patients with mutations in the NDUFS1 and NDUFS4 genes of complex I. J Biol Chem. 281: 10374–10380.
- 24. Kosmidis C, Efthimiadis C, Anthinidis G, Basdanis G, Apostolidis S, et al. (2011) Myofibroblasts and colic anastomosis healing in Wistar rats. BMC Surgery 2 11: 6.
- 25. Enestvedt CK, Thompson SK, Chang EY, Jobe BA (2006) Clinical review: healing in gastrointestinal anastomoses, part II. Microsurgery 26: 137–143.
- 26. Zacharakis E, Demetriades H, Kanellos D, Sapidis N, Zacharakis E, et al. (2007) Contribution of insuline-like growth factor I to the healing of colic anastomoses in rats. J Invest Surg 20: 76–82.
- 27. Portillo G, Franklin ME Jr (2010) Clinical results using bioadsorbable staple-line rienforcement for circulare stapler in colorectal surgery: a multicenter study. J Laparoendosc Adv Tech A 20: 323–327.
- 28. Dilek ON, Bakir B, Dilek FH, Denirel H, Yiğit MF (1996) Protection of intestinal anastomoses in septic environment with peritoneal graft and polyglicolic acid mesh: an experimental study. Acta Chir Belg 96: 261–265.
- 29. Testini M, Scacco S, Loiotila L, Papa F, Vergari R, et al. (1998) Comparison of oxidative phosphorylation in the anastomotisis of the small and large bowel. An experimental study in the rabbit. Eur Surg Res 30: 1–7.
- 30. Testini M, Portincasa P, Scacco S, Piccinni G, Minerva F, et al. (2002) Contractility in vitro and mitochondrial response in small and large rabbit bowel after anastomosis. World J Surg 26: 493–498.
- 31. Debellis L, Diana A, Arcidiacono D, Fiorotto R, Portincasa P, et al. (2009) The Vibrio cholerae cytolysin promotes chloride secretion from intact human intestinal mucosa. PLoS ONE 4: e5074.
- 32. Moeser AJ, Nighot PK, Engelke KJ, Ueno R, Blikslager AT (2007) Recovery of mucosal barrier function in ischemic porcine ileum and colon is stimulated by a novel agonist of the ClC-2 chloride channel, lubiprostone. Am J Physiol Gastrointest Liver Physiol 292: G647–656.
- 33. Nejdfors P, Ekelund M, Jeppsson B, Weström BR (2000) Mucosal in vitro permeability in the intestinal tract of the pig, the rat, and man: species- and region-related differences. Scand J Gastroenterol 35: 501–507.