Members of the Ashkenazi Jewish community are at an increased risk for inheritance of numerous genetic diseases such that carrier screening is medically recommended. This paper describes the development and evaluation of 30 TaqMan allelic discrimination qPCR assays for 29 mutations on 2 different high-throughput platforms. Four of these mutations are in the GBA gene and are successfully examined using short amplicons due to the qualitative nature of TaqMan allelic discrimination. Two systems were tested for their reliability (call rate) and consistency with previous diagnoses (diagnostic accuracy) indicating a call rate of 99.04% and a diagnostic accuracy of 100% (+/−0.00%) from one platform, and a call rate of 94.66% and a diagnostic accuracy of 93.35% (+/−0.29%) from a second for 9,216 genotypes. Results for mutations tested at the expected carrier frequency indicated a call rate of 97.87% and a diagnostic accuracy of 99.96% (+/−0.05%). This study demonstrated the ability of a high throughput qPCR methodology to accurately and reliably genotype 29 mutations in parallel. The universally applicable nature of this technology provides an opportunity to increase the number of mutations that can be screened simultaneously, and reduce the cost and turnaround time for accommodating newly identified and clinically relevant mutations.
Citation: Fedick A, Su J, Jalas C, Northrop L, Devkota B, Ekstein J, et al. (2013) High-Throughput Carrier Screening Using TaqMan Allelic Discrimination. PLoS ONE8(3): e59722. https://doi.org/10.1371/journal.pone.0059722
Editor: Giuseppe Novelli, Tor Vergata University of Rome, Italy
Received: November 12, 2012; Accepted: February 17, 2013; Published: March 26, 2013
Copyright: © 2013 Fedick et al. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Funding: Funding for this project was provided by Dor Yeshorim, The Committee for Prevention of Jewish Diseases, 429 Wythe Ave., Brooklyn, NY, 11211, USA. The funders had no role in study design, data collection and analysis, decision to publish, or preparation of the manuscript.
Competing interests: The authors have declared that no competing interests exist.
The high prevalence of carriers of recessive mutations in the Ashkenazi Jewish population  has made genotyping and carrier screening imperative. Carrier screening was first introduced for Tay-Sachs disease in the early 1970s, and because of the positive effect it had in reducing the number of children conceived with the disease, screening for Cystic Fibrosis and Gaucher Disease began in 1993 , . Since that time, the American College of Medical Genetics (ACMG) has also advised testing for six additional diseases including Bloom Syndrome, Canavan disease, Familial dysautonomia, Fanconi anemia group C, Mucolipidosis IV, and Niemann-Pick disease type A , . This has led to an increasing demand for quick and accurate diagnoses, and advances in detecting mutations and determining genotypes , . Methods such as Restriction Fragment Length Polymorphism (RFLP) testing  or the Amplification Refractory Mutation System (ARMS) , however, are hindered by time-consuming gel-based evaluation of PCR amplicons. Additionally, the platforms that have been developed for high-throughput genotyping purposes, such as a PCR and matrix-assisted, laser desorption/ionization time of flight (MALDI-TOF) mass spectrometry method  or arrayed primer extension (APEX) , are labor-intensive. Therefore, a more efficient, high-throughput, and automated method of screening for common mutations is needed.
Real time PCR with TaqMan allelic discrimination has frequently been used to characterize single nucleotide polymorphisms (SNPs) – and an insertion/deletion (indel) polymorphism 276 base pairs in length . It has also recently been shown to allow for the detection of large deletions  using the same methodology as applied to genotyping SNPs and small indels, and TaqMan Reverse transcription-PCR has been used to make clinical diagnoses for viral infections . The combined PCR and allelic discrimination procedure can be performed in a highly parallel manner using new technologies which allow nanoliter reactions to be performed. In addition, automated sample preparation, reaction setup, and data analysis, make this method ideal for genotyping a broad range of mutations in a high-throughput fashion.
In this paper, 30 individual TaqMan genotyping assays were designed for a subset of the ACMG approved mutations (ranging from point mutations to small and large indels) responsible for causing Bloom Syndrome, Canavan disease, Cystic fibrosis, Familial dysautonomia, Fanconi Anemia Type C, Gaucher disease, Mucolipidosis IV, Niemann-Pick disease, and Tay-Sachs disease. The four TaqMan assays designed to detect the Gaucher's mutations targeted short amplicons, consistent with TaqMan requirements, despite the similarities of the target GBA gene to the pseudogene, GBAP. All of the assays were validated on control samples and then evaluated on 2 platforms for high-throughput genotyping. While the mutations tested here represent those prevalent in the Ashkenazi Jewish population, the same methodology can be used to design TaqMan assays for other mutations.
Materials and Methods
All research material was obtained through written patient consent. Institutional review board permission was not required due to the removal of all sample identifiers prior to receipt by our lab (45 CFR part 46.101(b)(4)).
This study was conducted in multiple phases. TaqMan assays were designed and validated on genomic (gDNA) with previously characterized genotypes for the targeted mutations. Three large blind validations were then done on different sets of gDNA. The first blind analysis was performed using conventional methods for qPCR (large volume reactions), the second blind analysis was done on two high-throughput platforms to assess applicability, accuracy (consistency), and reliability of genotyping in a manner consistent with routine application, and the third was done to test assay performance for several mutations at their expected carrier frequencies.
The gDNA used in this study was obtained from individuals of Ashkenazi Jewish decent. The gDNA was extracted from blood using the QIAamp DNA Blood Maxi Kit or from buccal swabs using the Gentra Puregene Buccal Cell Kit, (QIAGEN Inc, Germantown, MD, USA) and the concentrations were obtained via Nanodrop (Thermo Fisher Scientific Inc., Wilmington, DE, USA).
The assays were designed by using NCBI to search for the full sequence of the gene (FASTA). Once the gene was found, roughly two hundred base pairs upstream and downstream of the mutation site were selected and put into Repeat Masker (Institute for Systems Biology, Seattle, WA, USA) to mask for the repeats. The original selected sequence was then put into the NCBI Blast site (SNP Flanks) so that SNPs could be masked. The assays were then made in File Builder software (Life Technologies [LTI], Carlsbad, CA, USA) or designed with Primer Express Software (LTI). The assays targeted ten missense, four nonsense, three frameshift, eight splicing, three small deletions, and one large deletion (6,433 base pair) mutation. The assay design for the large deletion has been previously published  and required the use of two assays. The context sequences for all of the assays are available in Table S1 as in compliance with the minimum information for the publication of real-time quantitative PCR experiments (MIQE) guidelines . All of the assays were designed so that the wild type allele utilized the VIC probe and the minor allele utilized the FAM probe.
To validate the assays, known heterozygous carrier and/or homozygous affected control gDNA samples were normalized to 5 ng/uL. The number of samples tested for each assay varied based on the availability of the gDNA, but always included one wild type sample and a no template control (NTC). The gDNA was plated in duplicate (to ensure accurate genotyping) in 384 well plates along with TaqMan Master Mix (LTI) and the assay bringing the final volume to 5 uL. The plates were centrifuged for 1 minute, sealed, and then run in duplex real time PCR reactions using FAM and VIC as the detector probes for each assay on both the ABI PRISM® 7900 HT Sequence Detection System (LTI) and the Applied Biosystems ViiA™ 7 Real-Time PCR System (LTI). After the real time PCR was finished, allelic discrimination analysis was performed using SDS 2.3 software (LTI).
A blinded study was then performed on the ABI PRISM® 7900 HT Sequence Detection System. Every assay was tested on 382 blind samples that were normalized to 2 ng/uL and 2 NTCs. The samples were plated in duplicate in 384 well plates at a volume of 5 uL, dried, and then TaqMan Master Mix and assay (premixed) were added for a final volume of 5 uL. The plates were centrifuged for 1 minute, sealed, and then run in duplex real time PCR reactions using FAM and VIC as the detector probes for each assay. Allelic discrimination analysis was performed using SDS 2.3 software and the data was analyzed in TaqMan Genotyper v1.1 (LTI).
Assay performance was then tested on the OpenArray® Real-Time PCR Platform (LTI) and the Fluidigm BioMark™ HD System (Fluidigm Corporation [FC], San Francisco, CA, USA). 470 samples with a variable number of carriers for each mutation (Table 1) and 10 NTCs were normalized to 50 ng/uL. For the OpenArray® Real-Time PCR Platform, 2.5 uL of the TaqMan® OpenArray® Genotyping Master Mix (LTI) and 2.5 uL of each gDNA sample were premixed in a 384 well plate and transferred to the genotyping plates using the OpenArray AutoLoader. For the Fluidigm BioMark™ HD System, the samples were premixed with master mix for a final volume of 5 uL and both the samples and assays were pressure loaded into reaction chambers using the IFC Controller. 95% confidence intervals were included in the calculations for accuracy, diagnostic accuracy, sensitivity, and specificity for both platforms. To calculate reproducibility, the coefficient of variation was calculated for the VIC and FAM signals for the wild type samples of every assay on both platforms. Notches equivalent to 95% confidence intervals  were calculated for the boxplot based on the formula: +/−1.58 IQR/sqrt(n).
The additional mutations tested at their expected carrier frequencies were run on the QuantStudio™12K Flex Real-Time PCR System (LTI). 192 samples were normalized to 50 ng/uL. 2.5 uL of TaqMan® OpenArray® Genotyping Master Mix (LTI) and 2.5 uL of each gDNA sample were premixed in a 384 well plate and loaded onto the genotyping plates using the QuantStudio™ 12K Flex OpenArray® AccuFill™ System. The results were analyzed in TaqMan Genotyper v1.2 (LTI).
NGS Data Acquisition and Analysis
Nine gDNA samples were normalized to 5 ng/uL and amplified for 14 cycles of PCR using the four GBA TaqMan assays and PreAmp Master Mix as recommended by the supplier (LTI). The Ion Xpress™ Plus gDNA and Amplicon Library Preparation protocol was used for the barcoded, short amplicons procedure (LTI). Original concentrations were obtained using a Nanodrop-8000 spectrophotometer, and molar concentrations were obtained using a Bioanalyzer on the Agilent High Sensitivity DNA microfluidic chip (Agilent Technologies Inc, Santa Clara, CA). The Ion OneTouch Template Kit was used for template preparation and the Ion Sequencing Kit v2.0 was used for the Ion 316 Chip-based sequencing (LTI).
FASTQ files were then obtained from the Ion Torrent Server and aligned against the reference sequence, which consisted of the nucleotides in the four target GBA amplicons, using Bowtie 2. Local alignment was done with default parameters to output the alignment file in Sequence Alignment/Map (SAM) format. These files were then converted to BAM (binary version of SAM) format using SAMtools. The BAM files were loaded in the Integrative Genomic Viewer (IGV) from Broad Institute so sequence alignment could be observed. Aligned reads with the reference sequence were displayed in the IGV interface and the count of each nucleotide as it corresponded to the reference sequence for each position, based on the total number of reads at that particular position, was obtained.
The initial validation of the assays on a small scale yielded 100% genotyping accuracy. Figure 1 shows the successful genotyping of the 4 GBA mutations on both the ABI PRISM® 7900 HT Sequence Detection System and the Applied Biosystems ViiA™ 7 Real-Time PCR System (see Figures S1 and S2 for other assay results). Unique clusters indicative of different genotypes were formed based on the signal intensity ratio of the two probes being used (VIC and FAM), predicting the genotypes with an accuracy of 100%. Because TaqMan allelic discrimination is a qualitative calculation, the similarities of the four GBA target sequences to the pseudogene did not affect the clustering other than uniformly shifting all of the clusters either towards the VIC or FAM axis based upon which probe was naturally also present in the GBAP gene. As seen in Figure 2, based on the nucleotides in the primer sequences, three of the four GBA assays would amplify sequences in both the GBA and GBAP genes. Additionally, the mutant probes for two of the GBA mutations had the same sequence as the wild type probes for GBAP, and the same wild type probe was present for the c.84_85insG mutation in both genes.
Allelic discrimination plots of the four GBA mutations. Water was used as the no template control. Fig. 1A: Validation on the ABI PRISM® 7900 HT Sequence Detection System. Fig. 1B: Validation on the Applied Biosystems ViiA™ 7 Real-Time PCR System. Fig. 1C: Blind validation on the ABI PRISM® 7900 HT Sequence Detection System.
Nucleotide sequences of the target amplicons in both the GBA and GBAP genes highlighting both the similarities and differences between the sequences. Forward and reverse primers are indicated by arrows and the probe is underlined. Nucleotide differences are indicated by red, bold lettering.
The non-specificity of the primers was confirmed through next generation sequencing (NGS) using the Ion Torrent Personal Genome Machine (PGM). NGS generates highly parallel data at base pair resolution, and allows multiple changes in a single amplicon to be observed in one viewing. Since each amplicon had at least one nucleotide that was different between the GBA and GBAP genes, the percentage that each gene was being targeted by the TaqMan assays could be determined by looking at these specific alleles. Results for two samples sequenced for this purpose are shown in Table 2, confirming the non-specificity of the primers for 3 of the 4 mutations. This non-specificity did not affect the genotyping calls, however, indicating that (q)PCR and allelic discrimination of short amplicons can successfully provide genotype results for GBA mutations.
To further investigate the accuracy of genotyping GBA mutations with short amplicons, 87 blind samples and an NTC were plated in duplicate and genotyped on the ABI PRISM® 7900 HT Sequence Detection System for the c.1448T>C mutation. Results were 100% concordant with those obtained using restriction enzymes highly specific to the normal copy of the gene, and indicated that 9 samples were wild type and 78 were carriers. To further confirm the genotypes, NGS was done for the wild type samples using the SNPs that differed between the GBA and GBAP sequences to determine which gene was being targeted and at what percentage. As indicated in Figure 3, using these informative SNPs in combination with parallel sequencing allowed for the successful detection of genotypes despite the assays having targeted both genes. When viewing the results, if there was a T allele present at the mutation site and a G allele present at the informative SNP site on the same strand, then the GBA gene had been targeted and the sequence was wild type. Similarly, if there was a C allele at the mutation site and a G allele at the informative SNP site, the GBA gene was still being targeted, but the sequence was the mutation. If there was a C allele at the mutation site and a C allele at the informative SNP site, then the GBAP gene had been targeted. By looking at the ratios of alleles at the mutation site for the GBA gene and ignoring the GBAP sequences, the genotypes of the samples could be obtained. For instance, the sample in Figure 3 was wild type because the frequency of the T and G allele pair was ∼50%, and the C and C allele pair was ∼50%. If the sample was a carrier, the T and G allele pair would be ∼25%, the C and G allele pair would be ∼25%, and the C and C allele pair would be ∼50%. NGS confirmed the TaqMan-assigned genotypes for all nine samples. Additionally, in no instance was a conversion of the pseudogene detected, indicating that this method can be used to genotype GBA mutations.
NGS integrative genomics viewer plot of a sample wild type for the c.1448T>C mutation. The plot includes a vertical bar graph (columns on top) which indicates the depth at each base, and in this situation, that both the GBA and GBAP genes are being targeted. Letter codes for each position are indicated at the bottom and represent the GBAP human genome reference sequence. Each plot also contains multiple horizontal bars representing individual sequence reads (grey bars), with a purple symbol indicating an insertion, a black dashed line indicating a deletion, and a letter indicating a variant relative to the reference sequence. For this mutation, a T allele at the mutation sight and a G allele at the informative SNP site indicates the wild type GBA gene, a C allele at the mutation site and a G allele at the informative SNP site indicates the mutated GBA gene, and a C allele at both the mutation and informative SNP site indicates the GBAP gene.
A large blind validation for all of the assays was then done, and indicated that 27 of the 30 assays had an autocalled accuracy of 100% (Figure 1c and Figure S3). The sample results were autocalled by TaqMan Genotyper v1.1 software and then manually reviewed. The genotyping clusters of two of the splice site mutations, (c.2204+6T>C in the IKBKAP gene and c.1421+1G>C in the HEXA gene), were clustered too closely for the software to make automatic calls. Manual calls were made with an accuracy of 100%, however, indicating that the two assays did not need to be redesigned. The assay designed to detect the indel in the BLM gene, however, had to be redesigned because the genotyping clusters were too close for automatic or manual calls to be made. Figure S4 shows both the original and revised assay results as tested on control samples. After redesigning the assay, the separation of the genotyping clusters increased significantly allowing automatic calls to be made.
The blind validation also included determining if the presence of other SNPs in the area of the target mutation would negatively affect genotypes obtained using TaqMan assays. The c.1521_1523delCTT (F508) mutation was specifically investigated because of the known benign SNPs located near this small deletion . 200 different blind DNA samples were genotyped for this mutation using the TaqMan assay in this panel. All of the samples were genotyped as being heterozygous carriers, which was concordant to their original genotypes. These results demonstrate that in this specific population, the performance of the F508 TaqMan assay was not negatively affected by theoretical nearby SNPs.
The assays were then tested for high-throughput genotyping by transitioning from 384-well plates to the OpenArray® Real-Time PCR Platform (LTI) and the Fluidigm BioMark™ HD System (FC). After the analysis was complete, the observed genotypes were compared to the expected genotypes of the control samples. The OpenArray® Real-Time PCR Platform had an individual data point call rate of 99.04%, a replicate call rate of 98.09%, an individual data point accuracy of 99.89% (+/−0.12%), and a diagnostic accuracy of 100% (+/−0.00%) (See Table 3). Samples that had different genotypes assigned between the two replicates were considered undetermined and rerun on the ABI PRISM® 7900 HT Sequence Detection System. The Fluidigm BioMark™ HD System had an individual data point call rate of 94.66%, a replicate call rate of 89.32% an individual data point accuracy of 92.96% (+/−0.39%), and a diagnostic accuracy of 93.35% (+/−0.29%). Additionally, the OpenArray® Real-Time PCR Platform had a sensitivity of 98.82% (+/−0.87%) and a specificity of 99.95% (+/−0.15%), while the Fluidigm BioMark™ HD System had a sensitivity of 94.89% (+/−0.02%) and a specificity of 92.91% (+/−0.12%).
The performance discrepancy between the two systems stemmed largely from the Fluidigm BioMark's inability to genotype two mutations occurring in the GBA gene (c.115+1G >A and c.1448T>C). For both mutations, only one genotyping cluster formed, which was autocalled as the carrier genotype. The probe sequence for the mutation in the GBA gene was the same as the wild type probe for the GBAP gene in both of these mutations, meaning that the clusters were shifted along the FAM axis. It is therefore possible that this shift caused a problem for genotyping on the Fluidigm BioMark. The assays performed accurately on the OpenArray system, however, suggesting that the designs are sufficient for high-throughput genotyping on some platforms. In order to perform better on the Fluidigm BioMark, the assays could potentially be redesigned with the primers covering one of the nucleotide bases that are variable between the two sequences.
The reproducibility of the platforms was evaluated by calculating the coefficient of variation (the standard deviation divided by the mean) for the VIC and FAM signals for the expected wild type samples of every assay on both platforms. The same samples were compared for each assay on each platform, and plate effects were avoided by using the same number of plates on each instrument. A two-sided, paired t- test with a 95% confidence interval showed that the Fluidigm BioMark had more variation in both the VIC and FAM signals when compared to the OpenArray. The p-value for the VIC signal was significant at 0.01814 and the p-value for the FAM signal was very significant at 0.000684. Figure 4 shows a notched box plot comparing the reproducibility between the two platforms, where the notches are equivalent to 95% confidence intervals. The notches for the two different FAM signals do not overlap, which indicates that the difference between the medians is statistically significant. The difference between the VIC signals is less pronounced, consistent with the t-test results. As previously stated, however, the Fluidigm Biomark's inability to genotype two mutations is influencing this result.
A notched box plot depicting the coefficient of variation for the FAM and VIC probes for the OpenArray® Real-Time PCR Platform and Fluidigm BioMark™ HD System. The coefficient of variation was calculated for the expected wild type samples, regardless of the final calls made by the instruments, for all 30 assays. The same samples were compared between instruments per assay, and the same numbers of plates were run on each instrument. The notches in the boxplot represent a 95% confidence interval.
Of the 29 mutations examined, 4 were originally tested at their expected carrier frequencies (ASPA c.914C>A, ASPA c.693C>A, CFTR c.3276C>A, and SMPD1 c.911T>C) , . These assays had an individual call rate of 99.28%, a replicate call rate of 99.04%, and an individual data point accuracy, a diagnostic accuracy, a sensitivity, and a specificity all of 100% on the OpenArray® Real-Time PCR Platform. Similarly, these assays had an individual call rate of 98.16%, a replicate call rate of 96.33%, and an individual data point accuracy, a diagnostic accuracy, a sensitivity, and a specificity all of 100% on the Fluidigm BioMark™ HD System. To further explore this, a second blind experiment was done on the OpenArray for four additional mutations (ASPA c.854A>C, CFTR c.3846G>A, IKBKAP c.2204+6T>C, and MCOLN1 c.406−2A>G) at their respective carrier frequencies as reported in literature . 192 samples were plated in duplicate and the combined results from the two studies yielded an overall individual call rate of 97.87%, a replicate call rate of 96.07%, an accuracy of 99.92% (+/−0.21%), a diagnostic accuracy of 99.96% (+/−0.05%), a sensitivity of 96.43% (+/−0.05%), and a specificity of 99.98% (+/−0.009%). These results therefore indicate that the assays are capable of successfully identifying heterozygous carrier samples at their expected frequencies in the population on a high-throughput platform.
Mutations responsible for causing diseases present at higher frequencies in the Ashkenazi population were quickly and accurately screened for using TaqMan assays and high-throughput genotyping platforms. The diagnostic accuracy of the OpenArray was higher than the Dynamic Array  and other methods such as MALDI-TOF mass spectrometry . The overall throughput for the two platforms tested ranges from approximately 55,000–70,000 genotypes per day on one instrument, which is comparable if not higher than other platforms. Additionally, when considering that the large deletion would have previously needed to be genotyped separately through copy number analysis, or multiplex ligation-dependent probe amplification , these platforms have increased the efficiency of genotyping.
In order to reduce sample preparation time, the use of the TaqMan® Sample-to-SNP™ Kit (LTI) is currently being investigated. This procedure allows samples to be genotyped directly from blood as opposed to having to isolate the gDNA first. The potential benefits include avoiding purification, thereby reducing expense and giving a more rapid turnaround time, as well as better automation . The incorporation of quality control measures to ensure that no cross contamination occurs is also recommended. In this study the samples were always run in duplicate to ensure that every individual was genotyped correctly by essentially genotyping them twice. The addition of a gender determination assay and/or assays that target specific SNPs for DNA fingerprinting can also be used as patient/sample identification controls. In order to validate a gender assay that uses the single base differences in the amelogenin genes to determine the presence of the X and Y chromosomes , we tested the assay on 1,012 samples using the ABI PRISM® 7900 HT Sequence Detection System. Gender results were 100% concordant with expected results, indicating that this assay would be a good candidate to include in the array. The development of multiplexing assays for the deletion mutations is also being investigated so that the use of two assays and two result plots per deletion will not be necessary.
Included in the mutation panel were four assays used to genotype four mutations in the GBA gene, which is located on chromosome 1q21 in close proximity to pseudogene GBAP . Since the GBA gene is 96% homologous to GBAP, long amplicons are normally needed to ensure that only the GBA gene is being targeted when genotyping . TaqMan genotyping is done with short amplicons, however, and therefore three of the four primer pairs were not specific to the GBA gene, amplifying both the GBA and the GBAP target sequences. The only influence that the pseudogene had, however, was to uniformly shift the genotyping clusters towards either the VIC or FAM axis depending on which probe sequence was naturally present in it. Therefore, even though both genes were being targeted, the assays were still able to reliably genotype samples based on the consistency of the pseudogene sequence. While gene conversion, fusion events, and reciprocal crossing over between the GBA and GBAP genes are believed to be the cause of mutations in the GBA gene, – to our knowledge no mutations have been reported to occur in the GBAP gene at the four mutation sites focused on in this paper, and NGS on nine samples showed no evidence of this occurring either. The non-specificity of the primers would only be an issue if there was a mutation in the GBAP gene at the exact site where the mutation was in the GBA gene, and the allele change matched the alleles present in the GBA gene. In the event of this occurring, it would not be possible to reliably genotype the mutation using this method.
Further investigation into using TaqMan genotyping assays in combination with available high-throughput genotyping platforms will be carried out by our lab in order to improve the accuracy of the tests for clinical use. The use of these assays can be beneficial not only for carrier screening, but also for newborn screening using genotyping and for preimplantation genetic diagnosis (PGD) of embryos undergoing in vitro fertilization. Furthermore, these assays can be used on genotyping platforms for qPCR or allelic discrimination, or be used to perform targeted NGS. Targeted NGS has recently been shown to provide highly consistent genotyping results for blastocysts when compared to established methodologies , and the data's parallel nature can also be used to provide additional sequence information, as demonstrated here for the GBA mutations.
ABI PRISM® 7900 HT Sequence Detection System Validation. Allelic discrimination plots depicting the validation of the remaining assays on the ABI PRISM® 7900 HT Sequence Detection System. Water was used as the no template control.
Applied Biosystems ViiA™ 7 Real-Time PCR System Validation. Allelic discrimination plots depicting the validation of the remaining assays on the Applied Biosystems ViiA™ 7 Real-Time PCR System. Water was used as the no template control.
Blind Validation. Allelic discrimination plots depicting the blind results for the remaining assays on the ABI PRISM® 7900 HT Sequence Detection System. Water was used as the no template control.
Original vs. redesigned assay results for the c.2207_2212delinsTAGATTC mutation. Allelic discrimination plots showing the results for both the original and redesigned assay used to genotype the c.2207_2212delinsTAGATTC mutation. Separation between the wild type and heterozygous carrier cluster improved significantly for the redesigned assay.
The authors would like to thank Kelly Howard, Marcia Slater, and Drs. Dennis Fantin and Vanee Pho from Life Technologies for coordinating early access to the QuantStudio 12 K Flex Instrument and Dr. Andrew Brooks from the Rutgers University Cell and DNA Repository for evaluating samples on the Fluidigm BioMark Instrument.
Conceived and designed the experiments: AF JS CJ JE NT. Performed the experiments: AF JS LN. Analyzed the data: AF JS CJ NT BD. Contributed reagents/materials/analysis tools: CJ JE. Wrote the paper: AF NT.
- 1. Bray SM, Mulle JG, Dodd AF, Pulver AE, Wooding S, et al. (2010) Signatures of founder effects, admixture, and selection in the Ashkenazi Jewish population. PNAS 107 (37) 16222–16227.
- 2. Kronn D, Jansen V, Ostrer H (1998) Carrier Screening for Cystic Fibrosis, Gaucher Disease, and Tay-Sachs Disease in the Ashkenazi Jewish Population: the first 1000 cases at New York University Medical Center, New York, NY. Arch Intern Med 158: 777–781.
- 3. Zuckerman S, Lahad A, Shmueli A, Zimran A, Peleg L, et al. (2007) Carrier Screening for Gaucher Disease: Lessons for Low-Penetrance, Treatable Diseases. JAMA 298 (11) 1281–1290.
- 4. Watson MS, Cutting GR, Desnick RJ, Driscoll DA, Klinger K, et al. (2004) Cystic Fibrosis Population Carrier Screening: 2004 Revision of American College of Medical Genetics Mutation Panel. Genet Med 6 (5) 387–391.
- 5. Gross SJ, Pletcher BA, Monaghan KG (2008) Carrier Screening in Individuals of Ashkenazi Jewish Descent. Genet Med 10 (1) 54–56.
- 6. Shi MM (2001) Enabling Large-Scale Pharmacogenetic Studies by High-Throughput Mutation Detection and Genotyping Technologies. Clin Chem 47 (2) 164–172.
- 7. Tsuchihashi Z, Dracopoli NC (2002) Progress in high throughput SNP genotyping methods. Pharmacogenomics J 2: 103–110.
- 8. Scott SL, Edelmann L, Liu L, Luo M, Desnick RJ, et al. (2010) Experience with Carrier Screening and Prenatal Diagnosis for Sixteen Ashkenazi Jewish Genetic Diseases. Hum Mutat 31 (11) 1240–1250.
- 9. Monaghan KG, Feldman GL, Palomaki GE, Spector EB (2008) Ashkenazi Jewish Reproductive Screening Working Group, et al (2008) Technical Standards and Guidelines for Reproductive Screening in the Ashkenazi Jewish Population. Genet Med 10 (1) 57–72.
- 10. Farkas DH, Miltgen NE, Stoerker J, van den Boom D, Highsmith WE, et al. (2010) The Suitability of Matrix Assisted Laser Desorption/Ionization Time of Flight Mass Spectrometry in a Laboratory Developed Test Using Cystic Fibrosis Carrier Screening as a Model. J Mol Diagn 12 (5) 611–619.
- 11. Schrijverm I, Külm M (2007) Gardner PI, Pergament EP, Fiddler MB (2007) Comprehensive Arrayed Primer Extension Array for the Detection of 59 Sequence Variants in 15 Conditions Prevalent Among the (Ashkenazi) Jewish Population. J Mol Diagn 9 (2) 228–236.
- 12. Shen GQ, Abdullah KG, Wang QK (2009) The TaqMan Method for SNP Genotyping. Methods Mol Biol 578: 293–306.
- 13. Schleinitz D, DiStefano JK, Kovacs P (2011) Targeted SNP Genotyping Using the TaqMan® Assay. Methods Mol Biol 700: 77–87.
- 14. Hui L, DelMonte T, Ranade K (2008) Genotyping Using the TaqMan Assay. Curr Protoc Hum Genet Chapter 2: Unit 2.10.
- 15. Robledo R, Beggs W, Bender P (2003) A simple and cost effective method for rapid genotyping of insertion/deletion polymorphisms. Genomics 82: 580–582.
- 16. Fedick A, Su J, Treff NR (2012) Development of TaqMan allelic discrimination based genotyping of large DNA deletions. Genomics 99: 127–131.
- 17. Tan EL, Yong LLG, Quak SH, Yeo WCA, Chow VTK, et al. (2008) Rapid detection of Enterovirus 71 by real-time TaqMan RT-PCR. J Clin Virol 42: 203–206.
- 18. Bustin SA, Benes V, Garson JA, Hellemans J, Huggett J, et al. (2011) Primer sequence disclosure: a clarification of the MIQE guidelines. ClinChem 57 (6) 919–921.
- 19. Chambers JM, Cleveland WS, Kleiner B, Tukey PA (1983) Graphical Methods for Data Analysis. California:Wadsworth International Group. 62 p.
- 20. Rohlfs EM, Zhou Z, Heim RA, Nagan N, Rosenblum LS, et al. (2011) Cystic Fibrosis Carrier Testing in an Ethnically Diverse US Population. ClinChem 57 (6) 841–848.
- 21. Fares F, Badarneh K, Abosaleh M, Harari-Shaham A, Diukman R, et al. (2008) Carrier frequency of autosomal-recessive disorders in the Ashkenazi Jewish population: should the rationale for mutation choice for screening be reevaluated? Prenat Diagn 28 (3) 236–241.
- 22. Reish O, Borochowitz ZU, Adir V, Shohat M, Karpati M, et al. (2009) Dynamic modification strategy of the Israeli carrier screening protocol: inclusion of the Oriental Jewish Group to the cystic fibrosis panel. Genet Med 11 (2) 101–103.
- 23. Wang J, Lin M, Crenshaw A, Hutchinson A, Hicks B, et al. (2009) High-throughput single nucleotide polymorphism genotyping using nanofluidic Dynamic Arrays. BMC Genomics 10: 561.
- 24. Gómez-Fernández N, Castellví-Bel S, Fernández-Rozadilla C, Balaguer F, Muñoz J, et al. (2009) Molecular analysis of the APC and MUTYH genes in Galician and Catalonian FAP families: a different spectrum of mutations. BMC Med. Genet 10: 57.
- 25. Fedick A, Su J, Jalas C, Treff NR (2012) High-throughput real-time PCR-based genotyping without DNA purification. BMC Res Notes 5: 573.
- 26. Tzvetkov MV, Meineke I, Sehrt D, Vormfelde SV, Brockmöller J (2010) Amelogenin-based sex dentification as a strategy to control the identity of DNA samples in genetic association studies. Pharmacogenomics 11 (3) 449–457.
- 27. Winfield SL, Tayebi N, Martin BM, Ginns El, Sidransky E (1997) Identification of three additional genes contiguous to the glucocerebrosidase locus on chromosome 1q21: implications for Gaucher disease. Genome Res 7 (10) 1020–1026.
- 28. Brown JT, Lahey C, Laosinchai-Wolf W, Hadd AG (2006) Polymorphisms in the glucocerebrosidase gene and pseudogene urge caution in clinical analysis of Gaucher disease allele c.1448T>C (L444P). BMC Med. Genet 7: 69.
- 29. Horowitz M, Tzuri G, Eyal N, Berebi A, Kolodny EH, et al. (1993) Prevalence of Nine Mutations among Jewish and Non-Jewish Gaucher Disease Patients. Am J Hum Genet 53 (4) 921–930.
- 30. Jeong SY, Kim SJ, Yang JA, Hong JH, Lee SJ, et al. (2011) Identification of a novel recombinant mutation in Korean patients with Gaucher disease using a long-range PCR approach. J Hum Genet 56 (6) 469–471.
- 31. Koprivica V, Stone DL, Park JK, Callahan M, Frisch A, et al. (2000) Analysis and Classification of 304 Mutant Alleles in Patients with Type 1 and Type 3 Gaucher Disease. Am J Hum Genet 66 (6) 1777–1786.
- 32. Treff NR, Fedick A, Tao X, Devkota B, Taylor D, et al.. (2013) Evaluation of targeted next-generation sequencing-based preimplantation genetic diagnosis of monogenic disease. Fertil Steril doi: 10.1016/j.fertnstert.2012.12.018.