The gold standard for the diagnosis of schistosomiasis is the detection of the parasite's characteristic eggs in urine, stool, or rectal and bladder biopsy specimens. Direct detection of eggs is difficult and not always possible in patients with low egg-shedding rates. Confocal laser scanning microscopy (CLSM) permits non-invasive cell imaging in vivo and is an established way of obtaining high-resolution images and 3-dimensional reconstructions. Recently, CLSM was shown to be a suitable method to visualize Schistosoma mansoni eggs within the mucosa of dissected mouse gut. In this case, we evaluated the suitability of CLSM to detect eggs of Schistosoma haematobium in a patient with urinary schistosomiasis and low egg-shedding rates.
The confocal laser scanning microscope used in this study was based on a scanning laser system for imaging the retina of a living eye, the Heidelberg Retina Tomograph II, in combination with a lens system (image modality). Standard light cystoscopy was performed using a rigid cystoscope under general anaesthesia. The CLSM endoscope was then passed through the working channel of the rigid cystoscope. The mucosal tissue of the bladder was scanned using CLSM. Schistoma haematobium eggs appeared as bright structures, with the characteristic egg shape and typical terminal spine.
Citation: Fritzsche C, Stachs O, Holtfreter MC, Nohr-Łuczak C, Guthoff RF, Reisinger EC (2012) Confocal Laser Scanning Microscopy, a New In Vivo Diagnostic Tool for Schistosomiasis. PLoS ONE 7(4): e34869. https://doi.org/10.1371/journal.pone.0034869
Editor: Jason Mulvenna, James Cook University, Australia
Received: December 21, 2011; Accepted: March 6, 2012; Published: April 17, 2012
Copyright: © 2012 Fritzsche et al. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Funding: The authors have no support or funding to report.
Competing interests: The authors have declared that no competing interests exist.
Schistosomiasis, a major parasitic disease infecting over 200 million people worldwide , is associated with considerable morbidity and mortality in the developing world . The gold standard for the diagnosis of schistosomiasis is the detection of the parasite's characteristic eggs in urine, stool, or rectal and bladder biopsy specimens. Direct detection of eggs is difficult and not always possible in patients with low egg-shedding rates. Serological tests such as enzyme-linked immunosorbent assays, immunofluorescence assays and indirect haemagglutination assays are used widely to detect antibodies against worm or soluble egg antigens. However, these assays are unable to differentiate between persistent and inactive infection and fail to discriminate between parasite species . While polymerase chain reaction methods can detect schistosomal egg DNA in stool and urine, and parasite DNA in serum samples, none of the published PCR methods has so far been evaluated for use in routine diagnosis.
The detection of viable eggs indicates an active infection requiring drug treatment. The viability of eggs collected from stool, urine and biopsy specimens can be tested using the miracidium hatching procedure in which eggs are incubated in fresh water to see whether larvae hatch. However, this procedure is time-consuming and has a low sensitivity .
Confocal laser scanning microscopy (CLSM) permits non-invasive cell imaging in vivo and is an established way of obtaining high-resolution images and 3-dimensional reconstructions . CLSM has recently been introduced as a diagnostic tool in ophthalmology, urology, dermatology, gastroenterology and oncology , , , , , .
Recently, we were able to demonstrate that CLSM is a suitable method to visualize viable Schistosoma mansoni eggs within the mucosa of dissected mouse gut .
In the case in hand we used CLSM in combination with standard cystoscopy to detect Schistosoma haematobium (S. haematobuim) eggs in the urothelium of a patient with urinary schistosomiasis.
Materials and Methods
Imaging and Confocal Laser Scanning Microscopy
The CLSM device (wavelength 670 nm, image size 348×384 pixels, 30 frames per second) used in this study was based on a scanning laser system for imaging the retina of a living eye, the Heidelberg Retina Tomograph II (HRT II, Heidelberg Engineering GmbH, Germany), in combination with a lens system (image modality). The lens system has a field of view of 400×400 µm and was coupled to the mucosal tissue. The penetration depth of the optical system is about 100 µm, limited by signal-to-noise ratio and background intensity.
Image modality was based on the HRT II and a rigid endoscope (length 23 cm, diameter 5 mm) with an integrated rod lens system (Storz, Tuttlingen, Germany) as described by Farahati . Coupled to the HRT II, the endoscope attains a spatial resolution of 5 µm in the axial dimension and 1–2 µm in the lateral dimension.
Standard light cystoscopy was performed using a rigid cystoscope under general anaesthesia. The CLSM endoscope was then passed through the working channel of the rigid cystoscope. Endomicroscopy was recorded as a series of video sequences at a frame rate of 30 frames per second.
The patient in this manuscript has given written informed consent to publication of their case details. This study was specifically approved by the local Ethics Committee of the University of Rostock, Germany (registry number A 2011 127, “Ethikkommission an der Medizinischen Fakultät der Universität Rostock”).
A 52-year-old male patient from Mozambique presented with dysuria due to urinary schistosomiasis which was first diagnosed two years ago by the histological detection of S. haematobium eggs in urinary bladder biopsies. Although treatment with praziquantel 40 mg/KG was administered twice within two years, histological examinations repeatedly revealed viable worm eggs in the urinary bladder biopsies.
When he presented to our department, urine and stool specimens were negative for worm eggs on multiple occasions and schistosoma antibodies were positive in the ELISA. Cystoscopy revealed the sandy patches and hyperaemic mucosa typical of schistosomiasis. The mucosal tissue of the bladder was scanned using CLSM. S. haematobium eggs appeared as bright structures (figure 1a), and after focus adjustment the characteristic egg shape and typical terminal spine became visible (figure 1b). However, miracidia were not observed within the egg shells.
After cystoscopy, repeated examination of the urine sediment revealed S. haematobium eggs which the miracidium hatching assay showed to be non-viable. Confocal laser scanning microscopy of the rectal mucosa and multiple stool examinations were negative for worm eggs.
This is the first case to document the direct detection of schistosomal eggs in vivo using CLSM. We found that CLSM is an appropriate method for visualising schistosomal eggs within the urothelium.
When stool and urine specimens are negative, biopsy specimens from the rectum or the urinary bladder can be turned into tissue crush preparations or stained paraffin sections and examined for the presence and viability of trapped eggs. However, biopsies are invasive, the examination area is limited and the specimens embedded in paraffin or used in miracidium hatching assays are evaluable only in specialised laboratories. Furthermore, the hatching assay is time-consuming and its sensitivity is low, as immature eggs not yet containing miracidia cannot be distinguished from dead eggs .
It was already demonstrated, that CLSM is not only a suitable method to detect schistosomal eggs in dissected mouse gut, but also to demonstrate their viability .
Our study aimed to test the suitability of CLSM as a method of detecting schistosomal eggs in vivo. CLSM allows a larger area to be scanned for eggs than does a biopsy, the method is non-invasive and the results are available immediately. Although the costs for the acquisition of the CLSM device might be high, further costs due to examination of the biopsies by pathologists, diagnostic viability assessments by specialised laboratories and costs of involved personnel could be saved.
However, the CLSM device in our study has several limitations. In order to allow a greater depth of penetration into the mucosal tissue, our CLSM device has a fixed wavelength of 670 nm. Therefore, the image is a reflectance image. Lower excitation wavelength would lead to shorter penetretation depths, but possibly lead to a better autofluorescens signal of the schistosomal eggs, improving the quality of the images. A possible useful technique for this case could be multiphoton endomicroscopy , . Because the field of view is so small, image acquisition is very sensitive to motion introduced by the investigator. Problems in manipulating the endoscope with the rigid cystoscope make it difficult to establish the required contact with the urothelium in some areas of the bladder, such as the anterior wall, and imaging the entire bladder is not practical using an endoscope with a 5 mm diameter and a 400 µm field of view. The theoretical focus distance from the last lens of the endoscope is about 300 µm, but in practice it depends from contact conditions, contact pressure, and refractive index of the tissue.
Despite these limitations, CLSM is a useful tool in the diagnosis of urinary schistosomiasis, especially as a way of avoiding biopsies of the bladder and the complications these can entail. However, smaller and more flexible endoscopes with integrated CLSM technology should be tested in order to improve CLSM as a diagnostic method. CLSM is already used as a diagnostic tool for gastroscopies and colonoscopies, especially for the detection of malignancies of the gut . These endoscopes should be further evaluated for the diagnostic of schistosomal eggs in rectal or intestinal tissues.
Conceived and designed the experiments: CF OS MH CNL RFG ECR. Performed the experiments: OS CNL. Analyzed the data: CF MH. Contributed reagents/materials/analysis tools: OS CNL RFG. Wrote the paper: CF.
- 1. Loebermann M, Sombetzki M, Langner C, Fuchsbichler A, Gumhold J, et al. (2009) Imbalance of pro- and antifibrogenic genes and bile duct injury in murine Schistosoma mansoni infection-induced liver fibrosis. Trop Med Int Health 14: 1418–1425.M. LoebermannM. SombetzkiC. LangnerA. FuchsbichlerJ. Gumhold2009Imbalance of pro- and antifibrogenic genes and bile duct injury in murine Schistosoma mansoni infection-induced liver fibrosis.Trop Med Int Health1414181425
- 2. Edungbola LD, Asaolu SO, Omonisi MK, Aiyedun BA (1988) Schistosoma haematobium infection among schoolchildren in the Babana district, Kwara State, Nigeria. Afr J Med Med Sci 17: 187–193.LD EdungbolaSO AsaoluMK OmonisiBA Aiyedun1988Schistosoma haematobium infection among schoolchildren in the Babana district, Kwara State, Nigeria.Afr J Med Med Sci17187193
- 3. Doenhoff MJ, Chiodini PL, Hamilton JV (2004) Specific and sensitive diagnosis of schistosome infection: can it be done with antibodies? Trends Parasitol 20: 35–39.MJ DoenhoffPL ChiodiniJV Hamilton2004Specific and sensitive diagnosis of schistosome infection: can it be done with antibodies?Trends Parasitol203539
- 4. Giboda M, Zdarska Z (1994) Alkaline phosphatase as marker of Schistosoma mansoni egg viability. Folia Parasitol (Praha) 41: 55–58.M. GibodaZ. Zdarska1994Alkaline phosphatase as marker of Schistosoma mansoni egg viability.Folia Parasitol (Praha)415558
- 5. Dailey ME, Manders E, Soll DR, Terasaki M (1995) Confocal microscopy of the living cell. In: Pawley JB, editor. Handbook of Biological Confocal Microscopy. New York: Plenum Press. pp. 381–398.ME DaileyE. MandersDR SollM. Terasaki1995Confocal microscopy of the living cell.JB PawleyHandbook of Biological Confocal MicroscopyNew YorkPlenum Press381398
- 6. Sonn GA, Jones SN, Tarin TV, Du CB, Mach KE, et al. (2009) Optical biopsy of human bladder neoplasia with in vivo confocal laser endomicroscopy. J Urol 182: 1299–1305.GA SonnSN JonesTV TarinCB DuKE Mach2009Optical biopsy of human bladder neoplasia with in vivo confocal laser endomicroscopy.J Urol18212991305
- 7. Gerger A, Hofmann-Wellenhof R, Samonigg H, Smolle J (2009) In vivo confocal laser scanning microscopy in the diagnosis of melanocytic skin tumours. Br J Dermatol 160: 475–481.A. GergerR. Hofmann-WellenhofH. SamoniggJ. Smolle2009In vivo confocal laser scanning microscopy in the diagnosis of melanocytic skin tumours.Br J Dermatol160475481
- 8. Guthoff RF, Zhivov A, Stachs O (2009) In vivo confocal microscopy, an inner vision of the cornea – a major review. Clin Experiment Ophthalmol 37: 100–117.RF GuthoffA. ZhivovO. Stachs2009In vivo confocal microscopy, an inner vision of the cornea – a major review.Clin Experiment Ophthalmol37100117
- 9. Nohr-Westphal C, Stachs O, Kroger M, Kram W, Guthoff R, et al. (2009) [Confocal laser scanning microscopy of the urothelium]. Urologe A 48: 1025–1026, 1028, 1030–1021: C. Nohr-WestphalO. StachsM. KrogerW. KramR. Guthoff2009[Confocal laser scanning microscopy of the urothelium].Urologe A 48: 1025–1026, 1028,1030–1021
- 10. Farahati B, Stachs O, Prall F, Stave J, Guthoff R, et al. (2010) Rigid confocal endoscopy for in vivo imaging of experimental oral squamous intra-epithelial lesions. J Oral Pathol Med 39: 318–327.B. FarahatiO. StachsF. PrallJ. StaveR. Guthoff2010Rigid confocal endoscopy for in vivo imaging of experimental oral squamous intra-epithelial lesions.J Oral Pathol Med39318327
- 11. Zhang WH, Zhu SN, Lu SL, Huang YL, Zhao P (2000) Three-dimensional image of hepatocellular carcinoma under confocal laser scanning microscope. World J Gastroenterol 6: 344–347.WH ZhangSN ZhuSL LuYL HuangP. Zhao2000Three-dimensional image of hepatocellular carcinoma under confocal laser scanning microscope.World J Gastroenterol6344347
- 12. Holtfreter MC, Stachs O, Reinhard M, Loebermann M, Guthoff RF, et al. (2011) Confocal Laser Scanning Microscopy for Detection of Schistosoma mansoni in the Gut of Mice. PLoS ONE 6: e18799.MC HoltfreterO. StachsM. ReinhardM. LoebermannRF Guthoff2011Confocal Laser Scanning Microscopy for Detection of Schistosoma mansoni in the Gut of Mice.PLoS ONE6e18799
- 13. Jung JC, Schnitzer MJ (2003) Multiphoton endoscopy. Opt Lett 28: 902–904.JC JungMJ Schnitzer2003Multiphoton endoscopy.Opt Lett28902904
- 14. Bar-Joseph H, Ben-Aharon I, Tzabari M, Tsarfaty G, Stemmer SM, et al. (2011) In vivo bioimaging as a novel strategy to detect doxorubicin-induced damage to gonadal blood vessels. PLoS One 6: e23492.H. Bar-JosephI. Ben-AharonM. TzabariG. TsarfatySM Stemmer2011In vivo bioimaging as a novel strategy to detect doxorubicin-induced damage to gonadal blood vessels.PLoS One6e23492
- 15. Goetz M, Kiesslich R (2010) Advances of endomicroscopy for gastrointestinal physiology and diseases. Am J Physiol Gastrointest Liver Physiol 298: G797–806.M. GoetzR. Kiesslich2010Advances of endomicroscopy for gastrointestinal physiology and diseases.Am J Physiol Gastrointest Liver Physiol298G797806