Citation: Klarkowski DB, Orozco JD (2010) Microscopy Quality Control in Médecins Sans Frontières Programs in Resource-Limited Settings. PLoS Med7(1): e1000206. https://doi.org/10.1371/journal.pmed.1000206
Published: January 26, 2010
Copyright: © 2010 Klarkowski, Orozco. This is an open-access article distributed under the terms of the Creative Commons Attribution License, which permits unrestricted use, distribution, and reproduction in any medium, provided the original author and source are credited.
Funding: The authors received no specific funding to write this paper.
Competing interests: The authors have declared that no competing interests exist.
Abbreviations: AFB, acid-fast bacilli; FN, false negative; FP, false positive; LQAS, lot quality assurance sampling; MSF, Médecins Sans Frontières/Doctors Without Borders; MSF-OCA, MSF Operational Centre Amsterdam; QC, quality control; TB, tuberculosis
Provenance: Commissioned; externally peer-reviewed.
The international humanitarian medical aid organization Médecins Sans Frontières/Doctors Without Borders (MSF) supports a wide network of medical laboratories in resource-constrained countries. Although MSF has always prioritized quality control (QC) for laboratory testing, prior to 2005 we were constrained by two significant limitations. First the QC workload was unsustainable in many programs, as MSF used the traditional protocol of reexamining 10% of negative slides and all positive slides. This is no longer considered practical –. Second MSF had no system for central data analysis as QC was performed independently at the individual laboratory level without standardized protocols.
In May 2005, MSF Operational Centre Amsterdam (MSF-OCA) developed and implemented a standardized, centrally reporting QC program to monitor the quality of microscopy for malaria, pulmonary tuberculosis (TB), and leishmaniasis. The malaria component of this protocol has been adapted by the World Health Organization (WHO) as the recommended international standard for malaria microscopy QC . Here we present a description of the QC protocol and an analysis over a 3-year period, the latter reflecting how the QC protocol has contributed to improved performance.
The QC protocol was designed to (1) have a small sample size to be feasible across all settings; (2) enable reliable analysis; (3) monitor both false-positive (FP) and false-negative (FN) results; and (4) be applicable to all microscopy testing.
Monthly QC Sample
The MSF-OCA protocol is based on a sample size of ten slides/month/site (for each test), as field experience has demonstrated that this QC workload is sustainable in most settings, and on the premise that it is better to perform less QC well than more QC poorly. A small sample size is also important to avoid overloading the limited capacity of the reference laboratory in many resource-constrained settings. Programs are encouraged to include more QC slides if this can be achieved without compromising the quality of the reexamination.
Sample selection and reexamination.
In summary, each month for each test: (1) Five weak positive slides are selected randomly from all weak positive slides; or if <5 weak positive slides, then all weak positive slides are selected. (2) Five negative slides are selected randomly from all negative slides; or if <5 negative slides, then all negative slides are selected. (3) If there are <5 weak positive (or negative) slides, then the number of negative (or weak positive) slides is increased to give a total minimum sample size of ten. (4) Strong positive slides are excluded from selection in the QC sample. (5) Laboratories unable to perform QC on a minimum of ten slides are assessed on an individual basis.
Blinded QC slides are reexamined within 4 weeks in the field by either a reference laboratory or an independent skilled laboratory technician.
Weak positive slides are defined as ≤9 trophozoites/acid-fast bacilli (AFB)/10 high power fields. These definitions were consistent across laboratory sites. Postimplementation experience now suggests that the criteria for a weak positive should be reduced to ≤9 trophozoites/AFB/100 high power fields.
While small sample QC has the important advantage of practicality, maintaining reliable analysis is also essential. To compensate for the small number of QC slides reexamined each month, our QC protocol uses analysis of cumulative data over 4-month periods (i.e., 4 months of data), referred to here as “cohort analysis”. These 4-month cohorts are used to increase the sample size analyzed, and as a compromise between the greater statistical stringency of analyzing a larger number of results over a longer duration (e.g., 12 months) versus the greater immediacy of detecting real-time laboratory performance by analyzing QC over a shorter period.
False-Positive and False-Negative Analysis
To enable FP analysis on small samples, our protocol uses biased sampling to increase the number of positive slides available for reexamination, and the targeting of weak positive slides to increase discriminatory power.
QC protocols that use a small sample size with random sampling of all slides, such as lot quality assurance sampling (LQAS) , have the potential disadvantage of being unable to adequately monitor false positivity because of insufficient positive slides at low prevalence rates if QC results are analyzed over short periods of time. To address this, the MSF-OCA protocol uses a biased QC sample of an equal number (whenever possible) of weak positive and negative slides to enable both FP and FN analysis.
Targeting weak positives.
The protocol selects only weak positive slides because errors of false positivity are most likely to occur during routine microscopy through microscopists reporting negative findings as weakly positive (to be “on the safe side”) , or through the misidentification of artifacts as parasites . Using weak positive slides also has greater discriminatory power than reexamining strongly positive slides ,,.
However, because FP results are more likely to occur among weak positive slides, reexamining only weak positive slides (rather than all positive slides) may overestimate the FP frequency in routine microscopy. We correct for this by using the formula:A limitation of this correction is that it assumes a negligible FP frequency for strong positive slides.
Common Protocol for All Microscopy
A primary objective for MSF-OCA was to develop a protocol that could be used for all microscopy testing. Although LQAS is recommended by WHO and others for AFB direct-smear TB analysis , we found this methodology unsuitable for malaria microscopy because determining the LQAS sample size is problematic when there is seasonal variation in the positivity rate. Our protocol therefore uses a fixed rather than variable number of QC slides.
Laboratory Performance Analysis
All QC results were reported to the central office in Amsterdam, which enabled comparative monitoring of results across all programs and the identification of poorly performing laboratories. Summarized analysis was reported back to the field to enable individual laboratories to compare their performance to other laboratories in similar settings.
We use percentage agreement because it is simple, direct, and understandable at all levels . Laboratory performance was considered satisfactory if the percentage agreement between the laboratory results and the reexamined results was equal to or exceeded the internal standards set by MSF-OCA (simple cut-off analysis).
In contrast to stable programs, such as government health laboratory networks, MSF operates as an emergency humanitarian organization, and laboratory programs open and close according to changing priorities. Therefore the QC analysis presented here reflects the overall performance of MSF-OCA programs over 2005–2008 with a changing composition of laboratories.
Because only a limited number of laboratories performed leishmaniasis testing, these findings are not presented here.
To improve statistical reliability, we only analyzed percent agreement on cohort data that included at least three monthly reports in the 4-month period, and FP and FN on cohort data that included at least ten positive or ten negative slides, respectively (Table 1). Fifty-seven laboratories met these criteria for malaria microscopy QC, and 54 for TB.
During the reported period, the internal MSF-OCA standards were set at ≥95% agreement for all slides (percent agreement) and ≤5% FP and FN slides.
Tests of difference between two proportions were performed using the Pearson's Chi-squared test. Analysis was performed using Epi Info 6 (US Centers for Disease Control) and STATA version 8.2 (StataCorp).
Marked progressive improvement in the overall malaria microscopy QC performance was seen over the period (Figures 1 and 2; Table 2). At the commencement of the QC program for the period May–December 2005 (two cohorts), 32.3% (10/31), 17.4% (4/23), and 58.1% (18/31) of laboratories complied with the percent agreement, FP, and FN targets, respectively. By 2008, for the period January–August (two cohorts), there were significant improvements (p<0.001) in the proportion of laboratories meeting each QC target, with the results of 95.7% (45/47), 86.7% (13/15), and 91.3% (42/46), respectively.
Progressive improvement in the overall AFB microscopy QC performance was seen over the period (Figures 1 and 3; Table 2). At the commencement of the QC program for the period May–December 2005 (two cohorts), 59.1% (13/22) and 43.8% (7/16) of laboratories complied with the percent agreement and FP targets, respectively. By 2008, for the period January–August (two cohorts), 82.1% (46/56; p = 0.033) and 77.1% (27/35; p = 0.019) of laboratories, respectively, met these targets. In contrast, the FN frequency remained relatively constant throughout the period (Figure 3), with no significant difference between the May–August 2005 and May–August 2008 cohorts (90% and 96%, respectively, p = 0.527).
We found the design of our QC protocol to be practical in field settings and easily understood and implemented by laboratory staff with limited training. We attribute this to a combination of a small QC sample size, a fixed number of slides independent of the workload, and use of simple percentage agreement for statistical analysis. The small sample size considerably decreased the QC workload while maintaining statistical reliability by using targeted sampling of only weakly positive slides and 4-month cohort analysis.
Our findings show a significant improvement in the accuracy of malaria and AFB microscopy comparing the periods May–December 2005 and January–August 2008. We attribute this improvement to the strengthening of our protocols, field support, and training over this period. However our QC protocol also played a central role by providing key information on a timely basis allowing us to prioritize those laboratory support activities. Also, and we believe critically, the reporting of compiled data back to the field provided the laboratories with clear performance indicators and enabled field laboratories to directly compare their performance against other laboratories working in similar circumstances. In our experience, this generated an environment of positive “competition” among laboratories that we believe has also contributed significantly to the improvement in laboratory quality performance.
For malaria microscopy, the number of FP and FN results decreased markedly. We attribute this to active follow-up of poorly performing laboratories identified by the QC protocol. In contrast, the frequency of FN results for AFB microscopy did not change significantly, and the improvement in percentage agreement reflects the decrease in the frequency of FP results. Laboratories for AFB also entered the analysis period at a higher level of performance compared with malaria microscopy (59.1% of AFB cohorts achieving ≥95% percentage agreement for May–December 2005 compared with 32.3% for malaria). This may be because AFB microscopy is relatively easier to perform than malaria microscopy as accurate malaria microscopy requires greater microscopy resolution and has a technically more demanding staining procedure.
However we also speculate that the random selection of negative AFB smears, which is the standard methodology for AFB QC protocols and is used in our protocol, may be problematic. Saliva smears are in general more likely to be negative or have an AFB density below the threshold of microscopy detection than sputum smears ,. Therefore there is less opportunity for QC to detect FN results by reexamining saliva slides as they have a higher prior probability of being truly microscopically negative than a sputum smear. With random selection, laboratories with a high proportion of saliva samples in routine practice will also have a high proportion of saliva slides in their QC sample, and therefore the QC FN frequency for such laboratories may be lower than their true FN frequency.
For the future, we are currently incorporating clerical error monitoring into our laboratory QC protocol, as this can also be a major source of error. With the increasing emphasis on disease eradication, we are also developing QC protocols to accommodate low positivity. Finally, we have implemented a pilot study to exclude saliva smears from the AFB QC sample.
From this recent field experience, our laboratory QC protocol was found to be well accepted and understood by all levels of field staff, practical in a wide variety of contexts, able to improve performance, and able to provide valuable program management information. As with all QC, implementation and sustainability requires commitment from field staff and project managers. Ongoing supervision and support are critical for central monitoring, ensuring compliance, and regular feedback reporting. The implementation of this centralized-reporting, standardized QC program has provided the catalyst for MSF-OCA to develop a laboratory “culture of quality” over the past 3 years, which in turn has strengthened the commitment and interest of laboratory field staff to ensure the success of health care programs.
We would like to thank the many MSF-OCA field teams for their consistent support, without whom this laboratory QC program would not have been possible. We thank Todd Swarthout for his valuable statistical analyses and editorial contributions to the manuscript, Sarah Moberley for her expert advice, and Pamela Hepple for her support on the implementation of the program. We also thank Oliver Yun for editing the manuscript.
ICMJE criteria for authorship read and met: DBK JDO. Wrote the first draft of the paper: DBK. Contributed to the writing of the paper: JDO. Developed the quality control model: DBK. Supported the field laboratories on the implementation and follow up of the methodologies: DBK JDO. Responsible for the protocol development, data collection, analysis and issuing benchmarking reports for the quality control program: JDO.
- 1. World Health Organization (2006) Informal consultation on quality control of malaria microscopy, March 2006. Accessed 4 December 2009. World Health Organization2006Informal consultation on quality control of malaria microscopy, March 2006. Accessed 4 December 2009.Available: http://www.who.int/malaria/docs/diagnosticsandtreatment/reportQua-mal-m.pdf. Available: http://www.who.int/malaria/docs/diagnosticsandtreatment/reportQua-mal-m.pdf.
- 2. World Health Organization (2002) External quality assessment for AFB smear microscopy. PHL, CDC, IUATLD, KNCV, RIT, and WHO. Washington (D.C.) Accessed 4 December 2009. World Health Organization2002External quality assessment for AFB smear microscopy. PHL, CDC, IUATLD, KNCV, RIT, and WHO. Washington (D.C.) Accessed 4 December 2009.Available: http://wwwn.cdc.gov/dls/ILA/documents/eqa_afb.pdf. Available: http://wwwn.cdc.gov/dls/ILA/documents/eqa_afb.pdf.
- 3. Martinez A, Balandrano S, Parissi A, Zuniga A, Sanchez M, et al. (2005) Evaluation of new external quality assessment guidelines involving random blinded rechecking of acid-fast bacili smears in a pilot project setting in Mexico. Int J Tuberc Lung Dis 9: 301–305.A. MartinezS. BalandranoA. ParissiA. ZunigaM. Sanchez2005Evaluation of new external quality assessment guidelines involving random blinded rechecking of acid-fast bacili smears in a pilot project setting in Mexico.Int J Tuberc Lung Dis9301305
- 4. World Health Organization (2009) Malaria microscopy quality assurance manual version 1. Accessed 4 December 2009. World Health Organization2009Malaria microscopy quality assurance manual version 1. Accessed 4 December 2009.Available: http://www.wpro.who.int/NR/rdonlyres/D854E190-F1B2-41E4-93B4-EB3D1A0163E4/0/MalariaMicroscopyManualVer1EDITED0609.pdf. Available: http://www.wpro.who.int/NR/rdonlyres/D854E190-F1B2-41E4-93B4-EB3D1A0163E4/0/MalariaMicroscopyManualVer1EDITED0609.pdf.
- 5. Stow NW, Torrens JK, Walker J (1999) An assessment of the accuracy of clinical diagnosis, local microscopy and a rapid immunochromatographic card test in comparison with expert microscopy in the diagnosis of malaria in rural Kenya. Trans R Soc Trop Med Hyg 93: 519–520.NW StowJK TorrensJ. Walker1999An assessment of the accuracy of clinical diagnosis, local microscopy and a rapid immunochromatographic card test in comparison with expert microscopy in the diagnosis of malaria in rural Kenya.Trans R Soc Trop Med Hyg93519520
- 6. Kilian AHD, Metzger WG, Mutschelknauss EJ, Kabagambe G, Langi P, et al. (2000) Reliability of malaria microscopy in epidemiological studies: results of quality control. Trop Med Int Health 5: 3–8.AHD KilianWG MetzgerEJ MutschelknaussG. KabagambeP. Langi2000Reliability of malaria microscopy in epidemiological studies: results of quality control.Trop Med Int Health538
- 7. Maguire JD, Lederman ER, Mazie BJ, Prudhomme O'Meara WA, Jordon RG, et al. (2006) Production and validation of durable, high quality standardized malaria microscopy slides for teaching, testing and quality assurance during an era of declining diagnostic proficiency. Malar J 5: 92.JD MaguireER LedermanBJ MazieWA Prudhomme O'MearaRG Jordon2006Production and validation of durable, high quality standardized malaria microscopy slides for teaching, testing and quality assurance during an era of declining diagnostic proficiency.Malar J592
- 8. World Health Organization (1998) Laboratory services in tuberculosis control. Part II: microscopy. Geneva. Accessed 4 December 2009. World Health Organization1998Laboratory services in tuberculosis control. Part II: microscopy. Geneva. Accessed 4 December 2009.Available: http://wwwn.cdc.gov/dls/ILA/documents/lstc2.pdf. Available: http://wwwn.cdc.gov/dls/ILA/documents/lstc2.pdf.
- 9. International Union Against Tuberculosis and Lung Disease (2000) Technical guide sputum examination for tuberculosis by direct microscopy in low income countries. Paris. Accessed 4 December 2009. International Union Against Tuberculosis and Lung Disease2000Technical guide sputum examination for tuberculosis by direct microscopy in low income countries. Paris. Accessed 4 December 2009.Available: http://www.theunion.org/index.php?option=com_guide&task=OpenDownload&id_download=59&id_guide=32&what=Guide%20Complet%20En. Available: http://www.theunion.org/index.php?option=com_guide&task=OpenDownload&id_download=59&id_guide=32&what=Guide%20Complet%20En.